, risk of rust on painted parts. however, in order to avoid any degradation or deterioration, baxter recommends that beds are exposed to this process a maximum of 10 / year. the presence of chemical residues on the surface of the bed may impact the compatibility. remove the excess with a rag. for mattresses compatibility, please refer to the user manual of the concerned mattress. steam cleaning these beds can be steam cleaned. however, in order to avoid any damage or deterioration caused by high pressure or abnormal surface temperature, the following ons should be taken : avoid any excess water and use reduced steam pressure with microfiber support when cleaning electrical components ( control unit, actuators, lateral caregiver units, half - siderails with keypads, remote controls and control cluster arms ), • do not use accessories such as high pressure hoses ( a ). it is preferable to use soft non - metallic brushes ( b ) and microfiber support ( c ) in such a way as to reduce the table level, • a b c steam cleaning areas clean with a microfiber support only. clean with a soft non - metallic brush or a microfiber support. prevent water and steam from getting into connectors that are not in use, • do not brush and use reduced pressure on labels and markings, • carefully dry and test the bed before reuse. • only facility - authorized personnel should perform maintenance of the hill - rom 900 accella bed. before maintenance or servicing works : ensure that the bed has been immobilized ( if no movements are required ), lock out all electrical functions, • • disconnect the bed from the mains if no electrical operations are planned, • secure the sleep surface and take whatever steps are necessary to prevent any movement, • not to intervene on the device if it is busy. any device connected to the ( can ) connector exclusively reserved @topic:9


# # # issue mouse not working # # # impact unable to operate the medical device properly # # # possible causes - mouse is faulty - mouse is not properly connected - mouse driver is not installed or is outdated # # # recommendations - replace the mouse with a new one - check the mouse connection and make sure it is properly seated - update the mouse driver @topic:30


# # # issue * end - tidal co2 values are lower than expected # # # impact * inaccurate monitoring of patient's respiratory status * potential delays in diagnosis and treatment # # # possible causes * malfunctioning gas module * issues with the monitor's calibration * loose connections or damaged cables * software or hardware errors # # # recommendations * inspect the gas module and monitor for any visible damage or loose connections. * calibrate the monitor according to the manufacturer's instructions. * check the cables connecting the gas module to the monitor for any damage or disconnections. * troubleshoot any software or hardware errors by consulting the manufacturer's technical support. * if the problem persists, replace the gas module or the monitor as necessary. @topic:4


# # # issue : - bed locking spontaneously # # # impact : - inability to adjust bed position - potential safety hazard for patients and staff # # # possible causes : - faulty control board - damaged wiring or connections - software glitch - malfunctioning actuator # # # recommendations : - reset the bed using the manual override switch - inspect wiring and connections for visible damage - update software to the latest version - replace any faulty components as necessary - ensure proper training and usage instructions are provided to staff @topic:6


# # # * * issue * * : remote control for the huntleigh contoura 360 electric bed is broken. # # # * * impact * * : the bed cannot be operated without a functioning remote control. this can cause inconvenience and delays in patient care. # # # * * possible causes * * : - the remote control may have been dropped or damaged. - the remote control may have been exposed to liquids or other corrosive substances. - the remote control may have a faulty battery or internal component. # # # * * recommendations * * : - * * inspect the remote control * * for any visible signs of damage. - * * replace the battery * * if necessary. - * * clean the remote control * * with a disinfectant wipe if it has been exposed to liquids or other corrosive substances. - * * if the remote control is still not functioning * *, contact the manufacturer for further assistance. @topic:7


# # # issue - scanner not working # # # impact - inability to use the medical cart for patient care # # # possible causes - dirty circuit board on the scanner # # # recommendations - clean the circuit board on the scanner - test the scanner to ensure that it is working properly @topic:23


inm2.?. j 02 c 1us20 _? _ j 00 c graphic view of current alarm limits in relation to currently monitored measurement value 339 27 printing patient reports sample report printouts realtime report patient demographic information, time stamp active alarms and inops, followed by vital signs built rcultnic rs - ptirt 12wbs74 si apr cl3 13 : 51 : 2 : 3 phtiert cut, : adult pai ^ ml : ne cand » r ; main oatsot birth : 4f « 1b53 sd ’ tan b3a : -? ■ ii2 hclrfit ; -? - h * * * aeefea ■ * *? md azac * ■ * etcoa loh * - * tvcs / nia high dbg no ■li ^ nsdijc cherik settings no central urm lt. era ocartttatkn php remint stkh htp ttibnu - ctmrt rat. hf. 77 bpm e / vc 17 / min st - i d. o m sv - ll u, u nn sr i : i 0 * 1 imt st avr 0. ( nn sr atl 0 * 1 mu 3r - avf - o. i m ir - v l'jlc cbio ‘ ft - hit. n. r mm * 0 i ik - 37 1 - 1 pi p - uif 5 3 mbp zthp 117 / 79 i9j ; mnh - j pju? ie / lc ( lt > ttintg jolj < - v - > nmu : j stjju j3l mtke incmj 15 nrch * roch 17 rtm tt02 17 if inj2 h - cu xi'- n - n ~ r v via - ■ @topic:8


# # # issue - broken spo2 cable # # # impact - inability to monitor patient's oxygen saturation levels # # # possible causes - physical damage to the cable - loose connections # # # recommendations - replace the spo2 cable - inspect the cable for any visible damage - ensure all connections are secure @topic:4


##livue instrument telemetry infrastructure. wlan the built - in wlan interface allows wireless operation of the x2 / mp2 monitors with the intellivue 802. 11 bedside adapter srr the built - in srr interface allows wireless communication of the mp2 / x2 monitors with an intellivue instrument telemetry transceiver. msl all components of the monitoring system communicate using an ieee802. 3 / ethernet lan in the measurement link ( msl ). this network is used to distribute data between the components 25 testing and maintenance introduction this chapter provides a checklist of the testing and maintenance procedures to ensure the performance and safety of the monitor and the mms extensions. for testing of the host monitor and the flexible module rack ( fms ), see the service guide of the host monitor. these tests must be performed only by qualified personnel certified by the responsible organization. qualifications required are : training on the subject, knowledge, experience and acquaintance with the relevant technologies, standards and local regulations. the personnel assessing safety must be able to recognize possible consequences and risks arising from non - conforming equipment. all recurring safety and performance assurance tests must be performed under equal environmental conditions to be comparable. testing of the mp2 / x2 may be performed either on the mp2 / x2 ( with external power supply ) directly or ( for the x2 ) on the host monitor. preventive maintenance refers specifically to the series of tests required to make sure the measurement results are accurate. the accuracy and performance procedures are designed to be completed as specified in the following sections or when readings are in question. for detailed instructions on the maintenance and cleaning of the monitor and its accessories, see care and cleaning, using batteries and maintenance and troubleshooting in the monitor's instructions for use. 27 terminology and definitions the following terms and definitions are used throughout this chapter and taken from the @topic:10


# # # issue - the bed's wheels cannot be locked into place, even when all of the wheel locks are engaged. # # # impact - this could cause the bed to move unexpectedly, which could lead to injury to the patient. # # # possible causes - the wheel locks are worn or damaged. - the wheel assembly is not properly aligned. - the bed frame is bent or damaged. # # # recommendations - inspect the wheel locks for wear and damage. replace any worn or damaged wheel locks. - inspect the wheel assembly for proper alignment. adjust the wheel assembly if necessary. - inspect the bed frame for bends or damage. repair or replace the bed frame if necessary. - retrain staff on proper use of the bed and wheel locks. @topic:6


# # # issue ecg monitor not displaying graph # # # impact inaccurate or no patient monitoring data # # # possible causes - faulty ekg cable causing poor signal transmission - loose connections between the device and the cable - defunct sensor on the ecg electrodes - software glitch in the monitor # # # recommendations - replace the ekg cable - ensure secure connections throughout the system - inspect and clean the electrodes - update or reinstall the monitor software @topic:18


# # # issue - control board malfunction # # # impact - the bed cannot be adjusted. # # # possible causes - electrical failure - mechanical failure - software failure # # # recommendations - replace the control board. - inspect the electrical connections and wiring. - test the software and firmware. @topic:1


down firmly with a clean dry cloth, paying particular attention to the edges. fitting the gas cylinders when the calibration unit is delivered, no gas cylinder is fitted. before putting the unit into service, screw the cylinder into the opening in the rear panel ( see “ routine maintenance ” on page - 267, next section ). when new, the calibration unit will contain a small amount of normal air. to expel this air before use and thus prevent inaccurate calibration, turn the timer control fully clockwise after fitting the gas cylinders and allow it to run for the full period. the calibration unit is now ready for use. storage of gas cylinders new gas cylinders should be stored in a cool place and not exposed to direct sunlight. disposal of used gas cylinders do not crush or incinerate used gas cylinders. they may be disposed of as scrap metal. routine maintenance changing the gas cylinders before each calibration the gas pressure indicator on the philips 1521 ob front panel should be read. if the indicator is in the “ black ” zone, change the gas cylinder as follows : 1 from the rear of the unit turn the empty gas cylinder anti - clockwise until the cylinder is free ( 3 - 6 turns ). 2 withdraw the empty cylinder. 3 take a full gas cylinder and insert it squarely into the rear of the unit. turn clockwise until hand tight. 4 check that the pressure indicator is no longer in the “ black ” zone. care and cleaning keep the surfaces of the calibration unit clean and free of dust and dirt. clean regularly with a lint - free cloth or sponge dampened in soapy water. avoid using alcohol or ammonia based cleaners which may damage the calibration unit. other strong cleaners such as povidine rr, lysol r and mikroklene r are not recommended since they may stain the unit. do not pour any @topic:17


# # # issue - the physiologic monitor is not charging. # # # impact - the physiologic monitor cannot be used to monitor patients. # # # possible causes - the battery is damaged or old. - the charging port is damaged. - the power supply is not working. # # # recommendations - replace the battery. - inspect the charging port for damage. - check the power supply. @topic:11


# # # issue - equipment not operating as expected - unable to adjust head and foot of the bed # # # impact - patient discomfort and delayed care - potential safety concerns # # # possible causes - damaged or malfunctioning pull - lock pin - binding between the head and foot of the bed - loose or disconnected cables # # # recommendations - inspect and replace the pull - lock pin if necessary - lubricate the bed frame and ensure there is no obstruction - check and tighten all cables and connections - perform a comprehensive functional test to verify the equipment is operating properly @topic:1


# # # issue - the patient's bed is unable to be lowered. # # # impact - this could lead to a number of problems, including : - patient discomfort - difficulty in providing care - increased risk of falls # # # possible causes - the latches that lock the bed in place are broken or jammed. - the motor that lowers the bed is not working properly. - there is an obstruction in the path of the bed's descent. # # # recommendations - inspect the bed's latches for any damage or debris. if necessary, replace the latches. - test the bed's motor to ensure that it is functioning properly. - check the path of the bed's descent for any obstructions. if necessary, remove the obstructions. @topic:1


# # # issue - deteriorated cuff # # # impact - inaccurate blood pressure readings - patient discomfort # # # possible causes - cuff over - inflation - excessive use - improper storage # # # recommendations - replace the cuff with a new one - use the cuff properly according to the manufacturer's instructions - store the cuff in a cool, dry place @topic:3


# # # issue - * * sat probe failure * * # # # impact - monitoring of oxygen saturation is impossible. - clinical decisions may be delayed or inaccurate. # # # possible causes - damaged or disconnected sat probe cable - defective sat probe sensor - incompatible sat probe with the monitor - software issue in the monitor # # # recommendations - * * verify the sat probe cable connection. * * ensure that the cable is securely connected to both the sat probe and the monitor. - * * inspect the sat probe cable for damage. * * look for any cuts, crimps, or other signs of damage. replace the cable if necessary. - * * check the sat probe sensor. * * clean the sensor with an alcohol wipe and inspect it for any damage. replace the sensor if necessary. - * * confirm the compatibility of the sat probe with the monitor. * * consult the monitor's user manual to verify that the sat probe is compatible with the monitor model. - * * update the monitor's software. * * check the manufacturer's website for any software updates for the monitor. install the updates to potentially resolve any software issues. @topic:4


# # # issue - drawer not closing properly # # # impact - medications or supplies may fall out of the drawer - drawer may be difficult to open or close - medical cart may be unstable if drawer is not closed properly # # # possible causes - drawer alignment is off - drawer slides are damaged or obstructed - drawer latch is damaged or misaligned - drawer is overloaded - drawer is not being closed properly # # # recommendations - check the drawer alignment and adjust if necessary - inspect the drawer slides for damage or obstructions and repair or replace as needed - check the drawer latch for damage or misalignment and repair or replace as needed - ensure that the drawer is not overloaded - train staff on how to properly close the drawer @topic:21


# # # issue - bed electric unable to lock the side rail # # # impact - patient safety hazard. # # # possible causes - the latch mechanism is damaged or faulty. - the side rail is misaligned. - the bed frame is damaged. # # # recommendations - inspect the latch mechanism for any damage or wear. replace the latch if necessary. - check the alignment of the side rail and adjust as needed. - inspect the bed frame for any damage and repair or replace as necessary. @topic:6


# # # issue - lack of clear information on the cause of equipment failure # # # impact - difficulty in understanding the underlying issue and identifying appropriate solutions - delays in repair and maintenance processes - reduced efficiency and productivity in the biomedical engineering team # # # possible causes - insufficient documentation or record - keeping - incomplete or vague troubleshooting procedures - lack of proper training or knowledge among technicians - miscommunication or misunderstanding between technicians and other stakeholders # # # recommendations - implement a comprehensive documentation system to record all equipment repairs, failures, and troubleshooting steps in detail. - establish clear and standardized troubleshooting procedures to guide technicians through systematic problem identification and resolution. - provide adequate training and ongoing professional development opportunities for technicians to enhance their diagnostic skills and knowledge. - foster open and collaborative communication between technicians, engineers, and clinical staff to ensure accurate and complete information sharing. @topic:12


# # # issue - inability to adjust the bed's height ( up - down movement ) # # # impact - patient discomfort and immobility - difficulty repositioning patients for medical procedures - reduced efficiency in patient care # # # possible causes - * * faulty remote control cord : * * the disconnected cord, as identified in the description, could be the primary cause of the failure. - * * loose or damaged connections : * * internal or external wire connections may have become loose or damaged, disrupting the signal transmission between the remote control and the bed. - * * malfunctioning control panel : * * the control panel on either the remote or the bed itself could have developed an issue, hindering its ability to process and execute commands. - * * battery issues : * * if the remote control uses batteries, they may be weak or depleted, affecting its functionality. - * * software or firmware bugs : * * in some cases, software or firmware glitches can cause temporary or permanent malfunctions in the bed's control system. # # # recommendations - * * re - connect the remote control cord securely : * * verify that the cord is properly seated in the designated port on both the remote and the bed. if necessary, disconnect and reconnect the cord to ensure a stable connection. - * * inspect and tighten internal connections : * * open the bed's electrical enclosure and visually inspect all wires and connections for any signs of damage or looseness. tighten any loose connections and replace any damaged wires. - * * test the control panel : * * operate the buttons on the remote control and the bed's control panel to assess their functionality. if any buttons are unresponsive or erratic, further troubleshooting may be required. - * * replace batteries ( if applicable ) : * * if the remote control uses batteries, @topic:7


_ prs plast bezel press - square cables m1006 - 61689 453563225911 m _ prs cbl adptrm1006b # c01 1 / 4 " phone jk m1011a intravascular oxygen saturation module part number 12nc description 453564120301 453564120301 m _ s02 module exchange, eng 453564120311 453564120311 m _ s02 module exchange, symbols 6 parts ) part number 12nc description 453564105801 453564105801 m _ s02 front housing assy s02 english 453564105811 453564105811 m _ s02 front housing assy s02 symbol ml 011 - 42201 453564120341 m _ s02 bezel 989803151591 989803151591 s02 optical module 453564097201 453564097201 m _ s02 oem exchange s02 optical module m1012a cardiac output parameter module part number 12nc description non rohs, dom ( date of manufacture ) < 2014 - 05 or no dom is present modules m1012 - 69601 453563458801 m _ co module c. o., eng nrohs m1012 - 69604 453564463371 m _ co module c. o., symbol multilang nrohs modules with picco m1012 - 69651 453563463011 m _ co module cio picco, eng nrohs m1012 - 69654 453564463391 m _ co module picco symbol multilang nrohs rohs @topic:19


# # # issue the connect lock simulator cable is broken. # # # impact the fetal monitor is unable to connect to the lock simulator. # # # possible causes - the cable may have been damaged during use. - the cable may have been pulled or tugged on. - the cable may have been frayed or cut. # # # recommendations - replace the connect lock simulator cable. - inspect the cable for any damage before using it. - handle the cable with care to prevent damage. @topic:2


# # # issue - sensor malfunction in the bedside physiologic monitor ( philips intellivue mp70 ) # # # impact - inability to accurately measure and monitor patient's vital signs ( e. g., heart rate, blood pressure, oxygen saturation ) # # # possible causes - hardware failure of the sensor - loose connections - software malfunction - damage to the sensor during installation or handling # # # recommendations - * * verified cause : * * sensor malfunction - * * immediate action : * * - replace the faulty sensor with a new one. - ensure stable electrical connections and proper signal transmission. - * * preventive measures : * * - conduct regular preventive maintenance and calibration to ensure optimal sensor performance. - provide proper training to staff on correct handling and installation techniques to prevent damage. - monitor sensor performance and troubleshoot any issues promptly to avoid prolonged downtime. @topic:20


# # # issue - the electric bed's head could not be adjusted. # # # impact - the patient may not be able to get into a comfortable position. - the patient may be at risk of injury if they try to adjust the bed themselves. # # # possible causes - the head of the bed is locked. - the motor that adjusts the head of the bed is not working. - the control panel is not working. # # # recommendations - check if the head of the bed is locked. if it is, unlock it by pressing the key - shaped button on the control panel. - if the head of the bed is not locked, check if the motor is working. you can do this by trying to adjust the bed with the control panel. if the motor is not working, you will need to replace it. - if the motor is working, check if the control panel is working. you can do this by pressing the buttons on the control panel. if the control panel is not working, you will need to replace it. @topic:1


# # # issue - defective cuff on the patient transport monitor. # # # impact - the patient's blood pressure cannot be accurately monitored. - this could lead to incorrect treatment decisions and / or adverse events. # # # possible causes - the cuff may have been damaged during use or storage. - the cuff may be the wrong size for the patient. # # # recommendations - the cuff should be inspected for damage. - the cuff should be replaced with a new one of the correct size. - the staff should be trained on the proper use and storage of the cuff. @topic:3


# # # issue - cuff has an air leak. # # # impact - the patient may not receive accurate blood pressure readings. # # # possible causes - cuff is damaged. # # # recommendations - replace the cuff with a new one. @topic:3


4535 643 57421 6 - 1 2 lcd frame 4535 643 57491 6 - 1 3 lcd touchscreen controller board 4535 643 57801 6 - 1 6 main board 4535 643 57371 6 - 2 15 main frame 4535 640 24621 6 - 2 30 navigation wheel board with wireless interface 4535 643 57391 6 - 1 9 nbp filter replacement 4535 640 41171 6 - 2 20 nbp pump / valve assembly with filter 4535 642 08081 6 - 2 19 replacement parts and assembly drawings suresigns vs4 service guide 6 - 1 spare parts description part number figure item number power input module ( ac ) 4535 640 51201 6 - 2 29 power supply module 4535 640 20471 6 - 2 14 rear case with handle 4535 643 57431 6 - 2 26 rear cover 4535 642 09241 6 - 2 25 recorder assembly 4535 640 95691 6 - 2 dai rs - 232 serial adapter ( includes usb hub, insulator | 9898 031 59601 not shown sheath, and usb clamp ) small parts kit : 4535 640 31481 e battery compartment cover 6 - 2 23 e bed rail hook clip not shown e d - spring not shown e fuses, 1600 ma / 250v ( 2 ) not shown e ground strip 6 - 1 7 e ground strip ( for 10 - inch screen ) not shown e m3 x 0. 5 8mm screws ( 6 ) not shown e m4x 0. 7 12mm screws ( 6 ) not shown e m4x 0. 7 16mm screws ( 2 ) not shown e navigation wheel 6 - 1 11 e o - ring, handle 6 - 2 34 e power cord retainer 6 - 2 28 ¢ rubber feet ( 4 ) @topic:19


# # # issue - battery not holding charge during charging usage. # # # impact - may cause the device to turn off unexpectedly during use, leading to interruptions in patient monitoring. # # # possible causes - faulty battery - problem with the charging system within the device # # # recommendations - replace the battery with a new one. - if the problem persists with a new battery, the device should be further inspected by a qualified technician to identify and fix the underlying issue with the charging system. @topic:11


configuration mode. when the previous patient is discharged, this clears the protocol log, resets the timers and deletes manually entered measurement values. 319 25 protocolwatch ssc sepsis protocol 320 26 recording mp40 / mp50 / mp60 / mp70 / mp90 the m1116b plug - in recorder records numerics for all active measurements and up to three waveforms. you can use it for local recording mounted either in the monitor ’ s fms or in the integrated module slots ( mp40 / mp50, mp60 / mp70 ). continue led flashes if a continuous recording is ongoing run / cont key starts a delayed recording or makes the current recording continuous stop key stops the current recording m1116b plug - in recorder 321 26 recording starting and stopping recordings mp20 / mp30 the optional integrated recorder records numerics for all active measurements and up to three waveforms. mp20 / mp30 integrated recorder central for central recording from the bedside, your monitor must be connected via a network to an recording information center. you can use either the ml 116b 2 - channel recorder or the standalone m3160a 4 - channel recorder. recordings made on the m3160a may look slightly different to those described here. see the documentation supplied with the information center for information on the 4 - channel recorder. starting and stopping recordings the recordings pop - up keys let you start and stop recordings. select the recordings smartkey to call up the line of pop - up keys. scroll right or left to see any pop - up keys not displayed. the selection of pop - up keys available depends on the monitor you are using. delayed rt a rt b rt c select hires all ecg recordng recordng recordng recordng 1 waves recordng waves vital alarm st setup stop all signs limits segments recordng recordng quickstarting recordings to quickstart any type of @topic:8


# # # issue : monitor not displaying vital signs in room 9 # # # impact : inability to monitor patient's vital signs, potentially leading to adverse events. # # # possible causes : - loose or disconnected cables - faulty monitor or central station - incorrect network configuration - room 9 is not linked with the central monitor # # # recommendations : - inspect cables and connections for damage or loose connections. - test the monitor and central station using known - good equipment. - verify network configuration and connectivity between the monitor and central station. - confirm that room 9 is properly linked to the central monitor by checking the hospital's network management system. @topic:16


# # # issue - fabric cuff cannot measure # # # impact - the monitor may not be able to accurately measure the patient's vital signs, such as blood pressure, heart rate, and oxygen saturation. # # # possible causes - the fabric cuff may be damaged or punctured. - the fabric cuff may not be properly connected to the monitor. - the monitor may be malfunctioning. # # # recommendations - inspect the fabric cuff for any damage or punctures. - make sure that the fabric cuff is properly connected to the monitor. - test the monitor with a known - good fabric cuff. - if the problem persists, contact the manufacturer for service. @topic:5


” inop. if the value is within the tolerance limits, there are no leakages and the leakage check is completed ; proceed to the pump check. 4 if the value is not within the tolerance limits, there is a leakage between the filterline inlet and the pump inlet. 5 check the filterline connections and open the m3015a to check the tubing connections at the pump inlet and the m3015a gas inlet. if the connections are good, try replacing the filterline and repeating the leakage check. if the situation remains, there is a leakage in the tubing and the m3015a must be exchanged. pump check 1 connect the flowmeter inlet to the m3015a gas outlet. 2 connect the filterline to the m3015a inlet. 3 block the inlet of the filterline using your fingertip and observe the cell pressure on the m3046a display. the cell pressure ( x4 ) should be more than 120 mmhg below the ambient pressure shown. if the pressure difference is less than 120 mmhg, the pump is not strong enough and you should replace it, irrespective of the pump optime. flow rate check and calibration check the flow rate in the microstream c0 2 extension as follows : 1 connect the flowmeter to the c0 2 filterline. 2 check on the flowmeter the flow that the microstream c0 2 extension pump draws ( x5 ). it should be 30 ml / min ± 7. 5 ml / min. if the value is within tolerance, proceed to the c0 2 gas calibration check. if the value is not within tolerance, calibrate as follows. 3 adjust the flow in the instrument by selecting increase flow or decrease flow until it is as close as possible to 50 ml per minute as indicated on the flowmeter gauge. 4 when you are @topic:17


# # # issue - battery replacement due to end of lifespan # # # impact - interruption of patient monitoring and transport # # # possible causes - normal battery wear and tear # # # recommendations - replace batteries as per manufacturer's specifications - consider using extended - life batteries for extended usage - implement a regular maintenance schedule to monitor battery health - provide spare batteries for urgent replacements @topic:11


# # # issue - vital signs monitor not charging # # # impact - monitor cannot be used - patient monitoring compromised # # # possible causes - faulty power cord - loose connection between cord and monitor - damaged battery # # # recommendations - check power cord for damage and replace if necessary - ensure cord is securely connected to monitor - contact biomedical engineering for further assistance @topic:11


# # # issue - cuff of the monitor is deteriorated. # # # impact - the monitor may not be able to accurately measure blood pressure. # # # possible causes - the cuff may be old and worn out. - the cuff may have been damaged. - the cuff may not be the correct size for the patient. # # # recommendations - replace the cuff with a new one. - inspect the cuff for any damage. - ensure that the cuff is the correct size for the patient. @topic:3


##ibration once a year or after 4000 hours of continuous use and following any instrument repairs or the replacement of any instrument parts. performance assurance tests some of the following test procedures must be performed in service mode. to enter service mode select operating modes in the main menu. then select service mode and enter the password. if required, open the screen menu in the monitor info line at the top of the screen and select service to access the service screen. this is required particularly for anesthetic gas module testing procedures. basic performance assurance test this section describes the basic performance test procedure. please refer to the section for detailed information on when which test procedure is required. procedure : power on the monitoring system and go into demo mode. check that each connected parameter ( module, mms, gas analyzer, intellibridge connected device ) displays values. full performance assurance test the following sections describe the fuu performance testing procedures i. e. detailed testing of each parameter with a patient simulator or specified tools. please refer to the section for information on when which testing procedure is required. ecg / resp performance test this test checks the performance of the ecg and respiration measurements. tools required : patient simulator. ecg performance 1 connect the patient simulator to the ecg / resp connector on the mms / inteuivue x2. 2 configure the patient simulator as follows : — ecg sinus rhythm. - hr = 100 bpm or 120 bpm ( depending on your patient simulator ). 3 check the displayed ecg wave and hr value against the simulator configuration. 4 the value should be loobpm or 120 bpm + / - 2 bpm. 63 [ 3 testing and maintenance ] respiration performance 1 change the patient simulator configuration to : — base impedance line 1500 ohm. - delta impedance 0. 5 ohm. — respiration rate 40 rpm or 45 @topic:17


# # # issue - bed is not adjustable # # # impact - patients may be unable to be positioned comfortably or safely - healthcare providers may be unable to access patients for medical care - delays in patient care # # # possible causes - electrical issue - mechanical issue - software issue - user error # # # recommendations - * * electrical issue : * * - check the power cord and connections - ensure that the bed is plugged into a working outlet - reset the circuit breaker or fuse - * * mechanical issue : * * - inspect the bed for any loose or damaged parts - lubricate the bed's moving parts - adjust the bed's height and position settings - * * software issue : * * - update the bed's software - reset the bed's computer - * * user error : * * - ensure that the user is properly trained on how to operate the bed - provide clear instructions on how to adjust the bed - post signs reminding users to follow proper procedures @topic:1


# # # issue - spo2 sensor cable failure. # # # impact - the device is unable to monitor oxygen saturation levels, which can lead to inaccurate readings and potential patient harm. # # # possible causes - the cable may have been damaged due to wear and tear, improper handling, or kinking. - the cable may have become loose or disconnected from the sensor or monitor. - the sensor may be malfunctioning. # # # recommendations - replace the damaged cable with a new one. - ensure that the cable is properly connected to the sensor and monitor. - inspect the sensor for any damage or defects. - calibrate the device according to the manufacturer's instructions. - regularly inspect and maintain the device to prevent future failures. @topic:4


# # # issue the cause of failure in the given input is the loose nut on the bed guard. # # # impact a loose nut on the bed guard could cause the guard to malfunction, which could lead to a patient falling out of bed. # # # possible causes * the nut may have been improperly tightened during assembly. * the nut may have loosened over time due to wear and tear. * the nut may have been damaged by impact or other trauma. # # # recommendations * inspect the bed guard regularly for loose nuts and bolts. * tighten any loose nuts or bolts immediately. * replace any damaged nuts or bolts. * train staff on the proper assembly and maintenance of the bed guard. @topic:6


# # # issue - the electric bed's guardrails are not locking securely. # # # impact - this could create a safety hazard for the patient, as they could potentially fall out of the bed. # # # possible causes - the locking mechanism is broken or damaged. - the bed's frame is bent or warped. - the guardrails are not properly aligned. - the weight of the patient is too great for the locking mechanism to hold. # # # recommendations - inspect the locking mechanism for any damage or wear. - check the bed's frame for any bends or warps. - make sure that the guardrails are properly aligned. - if the patient is too heavy for the locking mechanism to hold, consider using a bariatric bed. @topic:6


# # # issue - blood pressure ( bp ) readings from the nibp monitor are significantly higher than expected. # # # impact - incorrect bp readings can lead to misdiagnosis and inappropriate treatment decisions. # # # possible causes - * * battery degradation : * * the original battery was malfunctioning, potentially affecting the accuracy of the readings. - * * equipment malfunction : * * the nibp monitor itself may have developed a fault, causing inaccurate readings. - * * incorrect calibration : * * the nibp monitor may not have been properly calibrated, resulting in inaccurate measurements. # # # recommendations - * * test battery : * * replace the battery again to ensure it is functioning properly. - * * inspect equipment : * * perform a thorough inspection of the nibp monitor to identify any visible damage or loose connections. - * * calibrate equipment : * * calibrate the nibp monitor according to the manufacturer's specifications using a certified calibration tool. - * * compare readings : * * use a different bp measuring device ( e. g., manual sphygmomanometer ) to compare readings and identify any discrepancies. - * * log incident : * * document the issue, actions taken, and any findings in the maintenance record. - * * contact manufacturer : * * if the problem persists after performing the recommended steps, contact the manufacturer for technical support. @topic:15


# # # issue - electric bed is malfunctioning and only adjusts sporadically. # # # impact - patient discomfort and potential safety hazard. # # # possible causes - faulty power cord or outlet - damaged or loose electrical connections - malfunctioning motor or control unit # # # recommendations - inspect the power cord and outlet for any damage or loose connections. - check the electrical connections within the bed for any loose or corroded wires. - test the motor and control unit to ensure proper functionality. @topic:9


# # # issue - bed end position unable to adjust # # # impact - patient discomfort - delays in patient care - increased workload for nursing staff # # # possible causes - motor failure - electrical fault - control system failure - broken or disconnected wires - faulty switch or button - damaged circuit board # # # recommendations - replace the motor - check the electrical connections and repair any loose or damaged wires - inspect the control system and replace any faulty components - confirm the motor is receiving power - consult the manufacturer's service manual for specific troubleshooting steps - test the bed's functionality to ensure it is working properly before returning it to service @topic:1


# # # issue - the bed is not able to adjust. # # # impact - the patient may not be able to get into or out of bed. - the patient may not be able to be positioned correctly for treatment. # # # possible causes - the power supply to the bed is not working. - the electrical connections to the bed are not working. - the control panel is not working. - the bed motor is not working. - the bed frame is damaged. # # # recommendations - check the power supply to the bed. - check the electrical connections to the bed. - check the control panel. - check the bed motor. - check the bed frame. - if the bed is still not working, contact the manufacturer. @topic:1


maintenance of the electric bed to prevent misuse or accidental damage. @topic:9


. no limit timers automaticallount up. 396 3 using ti1 select the required timer and display the setup window. 2 select direction to switch between up and down. notification when a timer expires ( except a no limit timer ), the color changes to red and a message appears in the status line on the main screen. be aware that if a timer has been configured to display in light red, the color will not visibly change when the timer expires. additional notification is also available and can be set to : alarm : an inop alarm indicates the expiry of the timer sound : a single tone indicates the expiry of the timer no sound : no additional notification to set the additional notification, in the timers window : 1 select the required timer and display the setup window. select notification and select the notification level required from the list. 2 timer volume for all timers with the notification setting sound, you can set the volume of the tone. ers window : select the required timer and display the setup window. 1 2 select timer volume and select a volume setting from the list. if you want to have a timer displayed on the main screen, you can substitute it for a numeric which is not directly associated with a wave. if limited space is available, some of the timer display elements may not be displayed. the minimum information displayed is the label and the elapsed or remaining time. timera 2 hrs 01 : 47 27 1 select the numeric you want to substitute. 2 select change numeric. 3 select any timer. 397 main screen timer pop - up keys when you select a timer displayed on the main screen, additional pop - up keys become available : selecting this pop - up key lets you.... pop - up keys select a different timer for display. timers in the same way that you can substitute a timer for a @topic:8


# # # issue - button nbp fails to start when pressed. # # # impact - inability to measure non - invasive blood pressure ( nbp ) for patients. # # # possible causes - faulty or unresponsive button switch - electrical connection issues within the button circuit - software or firmware errors in the device's control system - damage to the button housing or mechanism # # # recommendations - * * inspection and testing : * * visually inspect the button and surrounding area for any physical damage. test the button's functionality by pressing it multiple times. - * * electrical circuitry : * * verify the continuity of the electrical circuit associated with the button using a multimeter or diagnostic tools. check for loose connections or damaged wires. - * * software / firmware update : * * ensure that the device is running the latest software or firmware version. updates may address known issues related to button functionality. - * * replacement / repair : * * if the button is physically damaged or the electrical circuit is faulty, it may need to be replaced or repaired by a qualified technician. - * * user training : * * provide training to users on proper button operation and maintenance to prevent damage or incorrect handling. @topic:15


welch [UNK] connex® spot monitor back - bottom - right view 5 6 8 no. feature description usb ports ( 2 ) connect optional accessories 1 2 usb cable connects apm work surface to monitor pm power cable 3 connects apm work surface to monitor provides an external ac power connection power connection 4 ground lug ( equipotential terminal ) provided for electrical safety testing and for connecting a otential - equalization conductor 5 secures the apm work surface when it is mounted on the apm stand ( with 4 screws ) recess for mounting 6 secures apm work surface battery cover 7 battery cover screw pm lighilluminates accessory bins and path for apm stand 8 23 dvanced settingadvanced settings provide access to device settings and service menu information with an access code. this section covers the following primary tabs : general. settings for system language as well as date and time, plus open source softwardisclosure. parameters. settings for physiological parameters, averaging, and intervals. network. settings for the internal ethernet and security, as well as wifi radio and bluetooth service. settings and features that support installation and on - device troubleshooting. note changes made in advanced settings take effect immediately, but they dnot change the configuration file and cannot be used to clone settings from one device to another. follow the instructions in " device configuration and setup " to create, edit, download, and install any settings you want available on multiple devices. access advanced settingnote you cannot access advanced settings if sensors or physiological alarms are active or if vital sign measurements are displayed. note create or change your advanced settings passcode using the configuration tool per the device configuration and setup instructions in this manual. 1. from the home tab, touch the settings tab. 2. touch the advanced vertical tab. enter your advanced settings password as the access code and touch ok. 3. note @topic:10


# # # issue - the guardrail of the bed fell off. # # # impact - the patient could fall out of bed and get injured. # # # possible causes - the guardrail was not properly attached to the bed. - the guardrail was damaged. # # # recommendations - the affected bed should be taken out of service immediately. - the guardrail should be inspected and repaired or replaced as soon as possible. - all other beds in the hospital should be inspected to ensure that their guardrails are properly attached and in good condition. @topic:6


* * issue : * * spo2 sensor not reading * * impact : * * inability to accurately monitor patient's oxygen saturation levels * * possible causes : * * - faulty spo2 sensor cable : the replaced cable may be defective or damaged. - loose connection between the sensor and the monitor : check if the sensor is securely connected to the monitor. - damaged sensor probe : the probe on the sensor may be damaged or obstructed. - insufficient blood flow to the sensor site : ensure that the sensor is applied to an area with good blood flow. - interference from other devices : electrical or magnetic interference from nearby devices can affect the sensor's accuracy. - software error in the monitor : the monitor's software may have a bug that affects the spo2 readings. * * recommendations : * * - replace the spo2 sensor cable again with a new, undamaged one. - ensure that the sensor is securely connected to the monitor. - clean the sensor probe and apply it to a new area with good blood flow. - check for and eliminate any potential sources of interference from nearby devices. - contact the manufacturer or a qualified technician to investigate and fix any software errors in the monitor. @topic:4


” on page 5 - 2 ) and restart the monitor. if the problem persists, replace the nbp module. see “ removing the nbp module ” on page 5 - 36. if the problem persists, replace the main board. see “ removing the main board ” on page 5 - 31. 833 ( 1 ) spo2 equip malfunc ( 1 ) spo rom checksum error. replace the spo > board. see “ removing the spo, board ” on page 5 - 28. 833 ( 2 ) spo2 equip malfunc ( 2 ) spo, ram test error. replace the spo > board. see “ removing the spo, board ” on page 5 - 28. 833 ( 4 ) spo2 equip malfunc ( 4 ) spo > analog signal path self test failed. replace the spo > board. see “ removing the spo, board ” on page 5 - 28. troubleshooting suresigns vs4 service guide 4 - 17 error codes code description possible cause action 833 ( 6 ) spo2 equip malfunc ( 6 ) spo > self test timeout. reseat the spo > board. if the problem persists, replace the spo, board. see “ removing the spo, board ” on page 5 - 28. 833 ( 8 ) spo2 equip malfunc ( 8 ) spo > internal dsp communication error. replace the spo, board. see “ removing the spo, board ” on page 5 - 28. 833 ( 10 ) | spo2 equip malfunc ( 10 ) spo, host communication error. replace the spo, board. see “ removing the spo, board ” on page 5 - 28. if @topic:22


# # # issue the scanner on the medical cart is not working. # # # impact the medical cart cannot be used for scanning patients. # # # possible causes - the scanner's scanner is damaged. - the scanner's cable is damaged. - the scanner's software is not working properly. # # # recommendations - replace the scanner's scanner. - replace the scanner's cable. - reinstall the scanner's software. @topic:23


# # # issue - equipment not charging # # # impact - equipment cannot be used for patient care - delays in patient care - increased risk of infection # # # possible causes - battery is not making good contact with the charging terminals - charging cable is faulty - power outlet is not working - battery is dead - equipment is not turned on # # # recommendations - check the connection between the battery and the charging terminals - replace the charging cable - try a different power outlet - replace the battery - turn on the equipment @topic:11


# # # issue cuff of the philips suresigns vs3 vital signs monitor is damaged. # # # impact the monitor may not be able to accurately measure blood pressure. # # # possible causes - the cuff may have been damaged during use. - the cuff may have been damaged during cleaning or disinfection. - the cuff may have been damaged due to age or wear and tear. # # # recommendations - replace the cuff with a new one. - inspect the cuff regularly for damage and replace it if necessary. - follow the manufacturer's instructions for cleaning and disinfecting the cuff. @topic:15


usb flash drive with the software upgrade into the usb port on the back of the monitor. note — the software upgrade folder must be located in the top directory of the usb flash drive. 3 open the system admin menu. see “ accessing the system admin menu ” on page 3 - 5. performance verification testing suresigns vs4 service guide upgrading the software step 4 select the upgrade software button. the monitor searches for a valid software image on the usb flash drive, and then displays the software image information in the upgrade software window. upgrade software current version : new version : if the monitor cannot find a valid software image, the screen returns to the system admin menu. 5 select the upgrade software button to start the upgrade. the upgrade in progress indicator increments while the upgrade is in progress. when the upgrade is complete, the checking memory crc and upgrade successful messages appear. after the software upgrade, the monitor automatically shuts down and restarts. 6 remove the usb flash drive. 7 before placing the monitor back in service, perform the following : e power - on self test. see “ power - on self test ” on page 3 - 12. e verify that your system settings are as expected. performance verification testing suresigns vs4 service guide 3 - 9 performing verification tests performing verification tests some of the verification tests require using the system diagnostics menu or the maintenance options. when you open the system diagnostics menu, monitoring is suspended and measurements stop. notes e the charging led does not show battery status while the system diagnostics menu is open. e the system diagnostics menu is not available in demo mode. accessing the system diagnostics menu use the system diagnostics menu to track component usage hours and perform the system diagnostic and verification tests. note — you cannot access the system diagnostics menu when the monitor is in demo mode. to access the system diagnostics menu @topic:22


# # # issue : - electric bed is not adjustable # # # impact : - bed cannot be adjusted to suitable positions for patient care and comfort - may cause discomfort and inconvenience to patients - delay in patient care if bed cannot be adjusted properly # # # possible causes : - power supply failure - electrical connection issue - mechanical [UNK] [UNK] - software error # # # recommendations : - * * check power supply : * * ensure that the bed is properly plugged into a power outlet and that the power supply is functioning. - * * inspect electrical connections : * * inspect all electrical connections, including wires, cables, and plugs. look for signs of damage, loose connections, or corrosion. - * * examine mechanical components : * * check for any mechanical obstruction or damage that may prevent the bed from adjusting. - * * troubleshoot software : * * if applicable, check for any software errors or updates that may be causing the issue. - * * contact manufacturer or qualified technician : * * if the technician is unable to resolve the issue, contact the manufacturer or a qualified biomedical engineer for further assistance. @topic:1


# # # issue - loose nut leading to detached wheel from the cart # # # impact - impaired mobility and stability of the vital signs monitor, posing risks to patient safety and healthcare operations. # # # possible causes - insufficient tightening of the nut during assembly or maintenance - wear and tear over time, causing the nut to loosen - improper handling or accidental impact # # # recommendations - ensure proper tightening of all nuts and bolts during assembly and maintenance - regular inspection and maintenance to detect and address any loose hardware - train staff on proper handling and storage of medical equipment to minimize accidental damage @topic:24


# # # issue - remote control failure # # # impact - inability to operate bed functions remotely # # # possible causes - physical damage to the remote control, such as cracking or breakage - electrical malfunction within the remote control - interference with the signal between the remote control and the bed # # # recommendations - replace the damaged remote control with a new one - inspect the electrical connections within the remote control and repair or replace any faulty components - ensure there are no sources of interference, such as other wireless devices or metal objects, near the bed or remote control @topic:7


# # # issue - battery degradation resulting in blinking status lights # # # impact - impaired functionality of the electric bed # # # possible causes - aged or worn - out battery # # # recommendations - replace the battery with a compatible spare part as per the attached quotation from the supplier. - ensure the replacement battery is covered by a warranty ( 12 months in this case ). @topic:32


hillrom. com welch [UNK] connex® spot monitor 10 overview technical support services welch allyn offers the following technical support services : telephone support. oaner equipmen • ervice agreementervice traininreplacement service parts product service • for information on any of these services, go to this site : service loaners for warranty or non - warranty repairs not covered under a support agreement, loaners are available for a nominal charge, subject to availability. payment is required prior to shipment for all loaners not covered under a support agreement. the loaner fee can be found on the welch allyn loaner welch allyn service centers that provide repair service for this product can, on request, loan a device for use while the device is being repaired. loaned devices are provided free of charge for products repaired while under a support agreement that includes a free loaner provision. ervice optionsmartcare tm services for maintenance and repair while product warranties provide basic assurance of welch allyn hardware quality, they may not include the full range of services and support you need. welch allyn offers premium service and support through our smartcare program. whether you service your own devices and require a minimum of support or rely on us to service your device, welch allyn provides a program that will meet your needs. a list of available service and support agreements is presented in the " field replaceable units " section of this manual. for more information, call your sales representative or visit our website : https : / / www. hillrom. com / en / services /. smartcare remote management smartcare remote management is a secure cloud - based portal that offers biomeds access to manage their hillrom connected devices remotely in a centralized portal. for more information and a complete list of compatibility information, refer to the smartcare remote management instructions for use. all repairs on products @topic:10


# # # issue - probe failure, leading to inaccurate readings and sensor malfunction. # # # impact - inability to accurately monitor patient vital signs. - potential for delayed diagnosis and treatment. - patient discomfort or harm due to inaccurate or missing data. # # # possible causes - damage to the probe during handling or storage. - excessive use or exposure to harsh conditions. - electrical or mechanical malfunction within the probe. - compatibility issues between the probe and the monitoring device. # # # recommendations - replace the faulty probe with a new one. - inspect the monitoring device to ensure compatibility with the new probe. - calibrate the monitoring device and probe according to manufacturer's instructions. - handle and store the probe carefully to avoid damage. - train personnel on proper usage and maintenance of the monitoring device and probe. - consider investing in a backup probe or monitoring device to minimize downtime in case of equipment failure. @topic:20


# # # issue - equipment : beds, electric - band : hill - rom - model : care assist es220 - description : beds, electric was not weighing due to the power button of weight control being off. biomedical engineer ( bme ) reset the weight button by holding down the 0. 0 button until the words " hold " and " cal " appeared. bme calibrated the beds, electric and tested it, and it worked normally. # # # impact - patients in room 18 of the icu2 may not be able to be weighed accurately, which could affect their treatment. # # # possible causes - the power button of the weight control may have been accidentally turned off. - there may be a loose connection in the wiring of the weight control. - the weight control may be faulty. # # # recommendations - check the power button of the weight control to make sure that it is turned on. - check the wiring of the weight control for any loose connections. - if the power button and wiring are both secure, then the weight control may need to be replaced. @topic:25


# # # issue : monitors, physiologic, central station not sending signals to central monitor. # # # impact : central monitor is unable to display patient data. # # # possible causes : * * - central monitor is not turned on or is not receiving power. * * * * - central monitor is not connected to the network. * * * * - central monitor is not configured correctly. * * * * - there is a loose connection between the central monitor and the monitors, physiologic, central station. * * * * - the central monitor is faulty. * * # # # recommendations : * * - check if the central monitor is turned on and is receiving power. * * * * - check if the central monitor is connected to the network. * * * * - check if the central monitor is configured correctly. * * * * - check for loose connections between the central monitor and the central station. * * * * - if the problem persists, contact the manufacturer for technical support. * * @topic:2


# # # issue - cuff leak # # # impact - potential inaccurate blood pressure measurements # # # possible causes - age and wear of the cuff - accidental damage to the cuff - improper cleaning or maintenance of the cuff # # # recommendations - replace the cuff with a new one - regularly inspect and clean the cuff - follow the manufacturer's guidelines for proper usage and maintenance of the cuff @topic:3


# # # issue spo2 sensor malfunction # # # impact inability to accurately measure patient's oxygen saturation levels, potentially leading to delayed or inaccurate diagnosis and treatment. # # # possible causes - faulty spo2 sensor - damaged cable connecting the sensor to the monitor - malfunction of the monitor's internal spo2 processing circuitry # # # recommendations - replace the spo2 sensor with a new one. - inspect the cable connecting the sensor to the monitor for any damage or loose connections. - if replacing the sensor and inspecting the cable does not resolve the issue, the monitor may need to be serviced by a qualified technician. @topic:4


# # # issue - the philips suresigns vs2 vital signs monitor is not working properly. # # # impact - the monitor cannot be used to accurately measure patient vital signs. # # # possible causes - the cuff is not attached securely. # # # recommendations - ensure that the cuff is attached securely to the patient. @topic:15


# # # issue - broken spo2 sensor cable # # # impact - inability to accurately monitor patient's oxygen saturation levels # # # possible causes - physical damage to the cable - manufacturing defect - improper handling or storage # # # recommendations - replace the damaged cable with a new one - inspect the remaining cables for any signs of damage or wear - train staff on proper handling and storage of medical devices - consider purchasing a spare set of cables to have on hand in case of emergencies @topic:4


##longs the lifespan of the equipment. * * * contact manufacturer : * * if the issue persists despite these measures, contact the device manufacturer for technical support and guidance on further troubleshooting or repair. @topic:12


# # # issue equipment failure of physiologic monitor ( bedside ) # # # impact * inability to monitor patient's vital signs accurately * potential delay or interruption in patient care * increased risk of patient complications # # # possible causes * malfunctioning components within the monitor * electrical power surge * software glitch * battery failure * improper use or handling # # # recommendations * send the equipment to the manufacturer or an authorized repair center for further diagnostics and repair * contact the manufacturer's technical support for troubleshooting and guidance * regularly inspect and maintain the equipment according to the manufacturer's instructions * train staff on proper use and handling techniques * ensure a stable power supply for the equipment @topic:2


# # # issue - oxygen saturation ( spo2 ) monitoring is unavailable. # # # impact - accurate monitoring of the patient's oxygen levels is compromised. - clinical decisions may be delayed or inaccurate due to lack of spo2 data. - potential for adverse patient outcomes if spo2 levels drop undetected. # # # possible causes - defective or damaged oxygen saturation ( spo2 ) cable. - loose or disconnected spo2 cable. - issues with the spo2 sensor or monitor. # # # recommendations - replace the damaged spo2 cable with a new one. - ensure all connections between the spo2 cable, sensor, and monitor are secure. - inspect and test the spo2 sensor and monitor to rule out any other underlying issues. - calibrate the spo2 monitor as per manufacturer's instructions to ensure accurate readings. - regularly inspect and maintain the spo2 equipment to prevent future failures. @topic:4


# # # issue the patient transport monitor is showing stuck vital signs readings, specifically the pulse rate is not updating. # # # impact this issue can affect the accuracy of patient monitoring and timely intervention in critical situations. # # # possible causes - sensor malfunction or improper placement - loose connections or damaged cables - software glitch or system error - electrical interference # # # recommendations - * * check sensor : * * ensure the sensor is properly placed and making good contact with the patient's skin. clean the sensor and the connection port if necessary. - * * inspect connections : * * tighten any loose cables and connectors. replace damaged cables as needed. - * * reset the device : * * power cycle the monitor to clear any software glitches or temporary errors. - * * contact manufacturer : * * if the issue persists after following the above steps, contact the manufacturer for technical support or a potential device repair. - * * monitor readings closely : * * pay attention to other vital signs being displayed and manually monitor the patient's pulse if possible. @topic:2


# # # issue - the headrest of the electric bed is not sturdy. # # # impact - patients may be uncomfortable or unsafe if the headrest is not sturdy. # # # possible causes - the headrest may have been damaged during shipping or installation. - the headrest may not have been properly attached to the bed frame. - the bed frame may be damaged or unstable. # # # recommendations - the bme should inspect the headrest and bed frame for damage. - the bme should tighten any loose screws or bolts. - the bme should replace the headrest or bed frame if necessary. @topic:1


# # # issue the bedside physiologic monitor ( philips intellivue mp70 ( m8007a ) ) is unable to power on due to a lack of electricity supply at the electrical outlet. # # # impact this issue can have a significant impact on patient care, as the monitor is used to monitor vital patient parameters such as heart rate, blood pressure, and oxygen saturation. without a functioning monitor, healthcare providers will be unable to accurately assess the patient's condition and provide appropriate care. # # # possible causes as the bme ( biomedical engineer ) has already verified that the monitor is functioning properly, the most likely cause of the failure is a problem with the electrical outlet or the building's electrical system. # # # recommendations the following recommendations are made to resolve the issue : 1. * * inspect the electrical outlet. * * verify that the outlet is properly wired and that there is power. if the outlet is not functioning properly, it should be repaired or replaced by a qualified electrician. 2. * * check the building's electrical system. * * if the outlet is functioning properly, the issue may be with the building's electrical system. contact the building's maintenance department to inspect the system and repair any faults. 3. * * use a surge protector. * * once the electrical outlet and building's electrical system have been verified to be functioning properly, using a surge protector can help to protect the monitor from future power surges. @topic:2


# # # issue - excessive noise produced during bed adjustments. # # # impact - discomfort and disturbance for patients - potential interference with patient care # # # possible causes - insufficient lubrication - worn or damaged components - improper assembly or installation # # # recommendations - perform regular maintenance and lubrication of the bed - inspect and replace any worn or damaged components - ensure proper assembly and installation of the bed according to the manufacturer's instructions @topic:9


# # # issue - electric bed does not power on # # # impact - unable to use the bed for patient care # # # possible causes - problem with power supply - problem with electrical wiring in the bed - problem with the bed's control panel # # # recommendations - check the power supply to the bed to ensure it is working properly - check the electrical wiring in the bed for any loose or damaged connections - check the bed's control panel for any damaged buttons or switches @topic:9


# # # issue - the medical cart cannot establish a wifi connection. # # # impact - the cart cannot be used for patient care. # # # possible causes - network connectivity issues - incorrect network configuration - hardware or software malfunction # # # recommendations - check the network connectivity and ensure that the cart is within range of the wifi network. - verify the network configuration on the cart to ensure that it is correct. - restart the cart and try to connect to the wifi network again. - if the problem persists, contact the manufacturer for further troubleshooting and repair. @topic:13


clinical unit, that encloses / encompasses the unit. what is its radius? • the number of the highest floor on which a transmitting device will operate. • how many transmitting devices will be used, i. e. the total number of m8001 / 2a # j45 and m2638a intellivue instrument telemetry adapter devices combined. • the effective radiating power : 6. 3 mw. • the equipment manufacturer : philips medical systems. • the model numbers : m8001 / 2a # j45 ( mp20 / 30 ) and m2638a intellivue instrument telemetry adapter used with m8004 / 5a ( mp40 / 50 ) 240 installing remote devices 7 installation instructions • the frequency range to be used : two separate coordinations are required : for the first one, click on the range of 1395. 0 through 1400. 0 mhz. for the second one, click on all the frequency ranges listed in the range of 1427. 0 through 1432. 0 mhz. when both registration and frequency coordination have been successfully completed, the intellivue instrument telemetry system can be activated. note that this process is the responsibility of the customer, as the final “ operator ” of the transmitting equipment. connections the cable specifications and part numbers for through wall solutions of the m3001a are described in the site preparation section of this manual. msl cable termination the following installation procedure describes how to install the wall installation cable kit when the patient monitor and the measurement server are not located at the same site. the kit consists of two connector boxes and a cable ( 15m or 25m ). for this procedure you need the insertion tool ( m3086 - 43801 ) and a small screwdriver. 1 draw the msl cable through the wall from the site of the monitor to the site of the measurement @topic:10


# # # issue - equipment not functioning properly # # # impact - patient care may be delayed or compromised - inconvenience to staff and patients # # # possible causes - faulty control unit - improper calibration - damaged electrical components - software issues # # # recommendations - replace the control unit - calibrate the equipment according to manufacturer's specifications - inspect electrical components for damage and replace as necessary - update software if needed @topic:12


# # # issue - equipment : philips suresigns vs2 monitor - description : spo2 cable is damaged. # # # impact - the spo2 cable is not functioning properly and cannot provide accurate spo2 readings. # # # possible causes - the spo2 cable may have been damaged due to : - physical damage, such as cuts or tears - excessive bending or twisting - exposure to extreme temperatures or humidity - normal wear and tear # # # recommendations - replace the damaged spo2 cable with a new one. - inspect the new spo2 cable regularly for signs of damage. - avoid bending or twisting the spo2 cable excessively. - protect the spo2 cable from extreme temperatures and humidity. @topic:4


configured, alarms generated from another device can be silenced at the bedside. either the monitor silence key is configured to silence both monitor and other device alarms ( must be configured at the information center ) or the silence bed pop - up key will be available : 1 2 select the silence bed pop - up keydepending on your information center configuration, the silence bed key may silence both other device alarms and bedside alarms. warning even when the other device data is not visible on the screen, you may be silencing monitor alarms and other device alarms, if the information center and monitor are so configured. switching off or pausing alarms from other devices when you select pause alarms or alarms off at the monitor, the alarms are off or paused for the bedside measurements. when you switch alarms off or pause alarms at the information center, both monitor alarms and alarms from other assigned devices are affected. when refer to the information center instructions for use for more details on the suspend / pause alarms behavior of the information center. using standby when you 1 select standby mode at the monitor, the bedside goes into standby mode but other assigned devices may continue monitoring. refer to the information center instructions for use for details on how selecting standby at the information center affects the monitor and other devices. ecg source tracking at the information center the information center continuously checks whether a valid ecg signal is coming from the monitor or from another assigned device. if you unplug the ecg patient cable from the monitor and plug it into another device, the information center will automatically switch to monitoring the ecg from the other device. at the monitor, its own ecg measurement will be deactivated and the setup ecg menu ill no longer be accessible. when you unplug the patient cable from the other device and plug it back into the monitor again, the information center will switch back to monitoring the ecg from the monitor. 124 @topic:10


# # # issue - remote control malfunction # # # impact - inability to adjust the bed's height # # # possible causes - defective remote control unit - loose or damaged wiring - malfunction in the bed's control system # # # recommendations - replace the remote control unit - inspect and repair any loose or damaged wiring - test the bed's control system for proper functionality @topic:7


# # # issue - battery not charging # # # impact - inability to use the device without a power source # # # possible causes - faulty battery - faulty charging port - faulty wiring - software issue # # # recommendations - replace the battery - inspect and clean the charging port - inspect and repair any faulty wiring - update the software @topic:11


# # # issue : drawer cannot be opened # # # impact : restricted access to medical supplies stored within the drawer # # # possible causes : - * * solenoid malfunction : * * the solenoid responsible for unlocking and opening the drawer may have become jammed or stuck due to friction or wear and tear. - * * mechanical obstruction : * * the drawer may be obstructed by foreign objects or misalignment of components, preventing its proper opening. - * * electrical fault : * * an electrical fault in the wiring or circuitry connected to the solenoid may have caused it to malfunction. - * * software glitch : * * a software glitch in the control system managing the drawer's operation could have prevented the solenoid from receiving the correct commands. - * * insufficient lubrication : * * the solenoid or other mechanical components may lack proper lubrication, leading to increased friction and hindering the drawer's opening. # # # recommendations : - * * inspect and clean the solenoid : * * check the solenoid for any visible damage or obstructions. clean the solenoid and its surrounding area to remove any debris or contamination that may be causing friction. - * * check for mechanical obstructions : * * examine the drawer and its guides for any foreign objects or misalignments that may be preventing its movement. ensure all components are securely fastened and properly positioned. - * * test electrical connections : * * verify that the electrical connections to the solenoid are secure and free of any damage or corrosion. use a multimeter to measure voltage and continuity in the circuit. - * * update software : * * if applicable, check for any available software updates for the control system managing the drawer's operation. install the updates to address any potential software glitches. @topic:21


# # # issue - the electric bed cannot be adjusted up or down. # # # impact - the patient may not be able to get in or out of bed. - the patient may not be able to be positioned comfortably. - the patient may be at risk for pressure sores. # # # possible causes - the bed is locked. - the bed's motor is not working. - the bed's remote control is not working. - the bed's power cord is not plugged in. - the bed's battery is dead. # # # recommendations - check if the bed is locked. - check if the bed's motor is working. if not, check the motor's fuses or circuit breakers. - check if the bed's remote control is working. if not, replace the batteries or try a different remote control. - check if the bed's power cord is plugged in. if not, plug it in. - check if the bed's battery is dead. if it is, charge the battery or replace it. @topic:1


# # # issue - oxygen saturation ( o2 sat ) measured by the propaq lt monitor is inaccurate or not measurable. # # # impact - incorrect o2 data may lead to improper patient monitoring and treatment decisions. # # # possible causes - * * defective or disconnected finger probe : * * the finger probe may be faulty or not properly connected to the monitor, resulting in inaccurate or absent o2 readings. - * * oxygen sensor malfunction : * * the o2 sensor within the finger probe may be malfunctioning or damaged, leading to incorrect o2 measurements. - * * electrical interference : * * external electrical interference, such as from nearby medical devices or electrical equipment, can disrupt the o2 sensor's signal. - * * software glitches : * * bugs or errors in the monitor's software can cause inaccurate o2 readings. - * * improper calibration : * * the monitor may not be properly calibrated, leading to incorrect o2 measurements. # # # recommendations - * * inspect finger probe : * * check the finger probe for any visible damage or loose connections. replace the probe if necessary. - * * clean o2 sensor : * * gently clean the o2 sensor on the finger probe with a soft cloth to remove any dirt or debris that may interfere with the signal. - * * check electrical connections : * * ensure that all electrical connections between the finger probe, monitor, and power source are secure and not damaged. - * * restart monitor : * * power cycle the monitor to reset any software glitches that may be causing the issue. - * * calibrate monitor : * * calibrate the monitor according to the manufacturer's instructions to ensure accurate o2 readings. - * * contact technical support : * * if the issue persists, contact the equipment manufacturer for further troubleshooting and potential @topic:4


# # # issue - the physiologic monitor is experiencing intermittent power cycling. # # # impact - the monitor is unable to provide reliable patient monitoring data. - this could lead to delayed or missed diagnoses, which could have serious consequences for the patient's health. # # # possible causes - * * power supply issue : * * the monitor's power supply may be faulty, causing the monitor to shut off unexpectedly. - * * software issue : * * a software bug could be causing the monitor to crash, resulting in power cycling. - * * hardware issue : * * a hardware component, such as the motherboard or the power button, could be faulty, causing the monitor to turn off. # # # recommendations - * * check the power supply : * * ensure that the power supply is providing the correct voltage and current to the monitor. if the power supply is faulty, replace it. - * * update the software : * * check for any software updates that could resolve the issue. if an update is available, install it. - * * inspect the hardware : * * inspect the monitor's hardware for any signs of damage or wear and tear. if any damaged components are found, replace them. - * * contact the manufacturer : * * if the issue cannot be resolved by following the above steps, contact the manufacturer for assistance. @topic:2


# # # issue the bed cannot be adjusted. # # # cause of failure the given input does not provide any information about the cause of failure. # # # recommendations the cause of failure could be one of the following : - a mechanical issue, such as a broken gear or motor. - an electrical issue, such as a loose wire or faulty switch. - a software issue, such as a corrupted program or incorrect settings. - a user error, such as improper operation or overloading the bed. # # # next steps to identify the cause of failure, the technician should perform the following steps : 1. inspect the bed for any visible damage or loose parts. 2. test the bed's electrical system using a multimeter or other diagnostic tool. 3. check the bed's software settings using a laptop or other programming device. 4. observe the user operating the bed to identify any potential errors. 5. consult with the manufacturer or other experts for additional troubleshooting guidance. # # # conclusion the given input does not provide enough information to determine the cause of failure. the technician should perform the recommended troubleshooting steps to identify the underlying issue and make necessary repairs or adjustments. @topic:1


# # # issue - rusted side rails # # # impact - unsafe for patient use # # # possible causes - lack of maintenance - exposure to moisture or corrosive substances # # # recommendations - replace rusted side rails - implement regular maintenance schedule - protect equipment from moisture and corrosive substances @topic:6


# # # issue - the bedside physiologic monitor is not staying in the desired position. # # # impact - the monitor may not be able to be used by medical staff, which could delay or prevent patient care. # # # possible causes - * * loose mounting screws : * * the screws that hold the monitor in place may have come loose. - * * damaged mounting bracket : * * the bracket that holds the monitor in place may be damaged. - * * faulty monitor : * * the monitor itself may be faulty and not able to stay in the desired position. # # # recommendations - * * tighten the mounting screws : * * if the mounting screws are loose, tighten them using a screwdriver. - * * replace the mounting bracket : * * if the mounting bracket is damaged, replace it with a new one. - * * contact the manufacturer : * * if the monitor is faulty, contact the manufacturer for repair or replacement. @topic:2


- 23 board, wireless, removing, 5 - 32 communications board, 5 - 31 installing battery, 5 - 5 lcd, removing, 5 - 28 nbp module, removing, 5 - 16 overview, 5 - 1 power supply, removing, 5 - 17 removing battery, 5 - 3 spo > board, removing, 5 - 22 temperature module, 5 - 6 tools required, 5 - 2 replacement parts, 6 - 1 replacing battery, 2 - 4 s safety routine checks, 2 - 1 specification, vi standards, vi verification test, 3 - 17 self test, running, 4 - 30 sequence, start - up and power, 4 - 3 serial number location, 5 - 25 resetting, 5 - 25 symbol, iv viewing, 4 - 2 service codes, description, 4 - 12 service option, system admin menu, 3 - 5 shutdown, hard, performing, 5 - 2 software upgrades procedure, 3 - 7 recommended flash drive, 3 - 7 software version, viewing, 4 - 2 spare parts, 6 - 1 speaker removing, 5 - 15 testing, 4 - 31 theory of operation, a - 4 spo, board, removing, 5 - 22 processing, theory of operation, a - 6 readings, troubleshooting, 4 - 9 verification test, 3 - 13 start - up and power sequence, description, 4 - 3 symbols, on monitor and packaging, 1v system admin menu accessing, 3 - 4 options, 3 - 5 system configuration, setting, 5 - 27 system diagnostics menu, accessing, 4 - 29 system diagnostics, running, 4 - 29 system information, viewing, 4 - 2 t temperature calibration key, 3 - 16 measurement problems, troubleshooting, 4 - 9 verification test, 3 - 17 temperature module, removing, 5 - 6 test equipment required, 3 - 3 testing audio, 4 - 31 keys and wheel @topic:19


# # # issue the battery of the philips suresigns vs2 vital signs monitor has failed. # # # impact the monitor cannot be used without a functional battery. # # # possible causes * the battery is old and has reached the end of its lifespan. * the battery has been damaged due to physical abuse or improper storage. * the battery is not compatible with the monitor. # # # recommendations * replace the battery with a new one that is compatible with the monitor. * if the battery is old, consider replacing it even if it is still functioning. * store the battery in a cool, dry place when not in use. * avoid exposing the battery to extreme temperatures or physical abuse. @topic:11


# # # issue battery not charging # # # impact device cannot be used without a charged battery. # # # possible causes - faulty battery - faulty charging port - faulty power supply - software issue # # # recommendations - replace the battery. - inspect the charging port for damage. - check the power supply. - update the software. @topic:11


possible, check if there are any software updates available for the bed or remote control. install the updates to address any potential software - related issues. 6. * * replace the remote control : * * if all other troubleshooting measures fail, the remote control may need to be replaced. use an oem ( original equipment manufacturer ) or authorized service provider to ensure compatibility and proper functionality. @topic:7


# # # issue - loose handrail mounting bolts # # # impact - potential patient fall hazard # # # possible causes - normal wear and tear - insufficient tightening of bolts during assembly - damaged threads on bolts or mounting holes # # # recommendations - regularly inspect handrails for loose bolts and tighten as necessary - use thread - locking compound on bolts to prevent loosening - inspect the bed frame for any damage to the mounting holes. - ensure that any repairs are performed by qualified personnel - train staff on the importance of proper bed assembly and maintenance @topic:6


# # # issue - broken handle # # # impact - inability to properly grip and maneuver the device # # # possible causes - wear and tear - accidental damage - improper storage or handling # # # recommendations - inspect the device regularly for signs of wear or damage - handle the device with care - store the device in a safe and secure location - replace worn or damaged parts promptly @topic:24


the manually entered address is used. 3. select subnet mask. the subnet mask must be provided for manual ip addresses. the subnet mask must consist of a single consecutive series of " 1 " bits ; e. g. 255. 255. 248. 0. the configured value is ignored when the ip address is provided by a bootp server. 4. select default gateway. the ip address of the default gateway can be optionally configured. the configured value is ignored if ip address and gateway are provided by a bootp server. the configured value must be within the range of the subnet mask. configuration settings for cscn routed bedside monitors ( rbm ) the following settings are used for customer supplied clinical network ( cscn ) routed bedside monitors. to access these settings, select the bed label screen element to call up the bed info menu. igmp : shows status of igmp support ( on or off ). igmp ( internet group multicast protocol ) is used by many switch manufacturers to limit the number of destinations targeted by a multicast packet. ci mode : the mode in which ci messages ( connect indication messages ) are send ( broadcast, multicast, manual ). ci address : ip address for connect indication messages only being used if ci mode is set to manual. if ci mode is broadcast the ci address 1s implicitly the subnet broadcast address. if ct mode is multicast the ci address is implicitly 224. 0. 23. 63. ci ttl : sets the ttl ( time to live ) of the ci message. defaults to 1. 214 configuring routed bedside monitors support an intellivue mp2 / x2, mp5 / mps5t or mp20 - 90 monitor must be running software revision level g. o or higher to be used as a routed bedside monitor ( rbm ). caution a @topic:16


# # # issue the cause of the equipment failure is a defective cuff. # # # impact due to defect, the monitor may not be able to accurately measure patient data, which could lead to incorrect treatment decisions. # # # possible causes - cuff may get damaged due to improper handling or storage. - cuffs may deteriorate over time due to normal wear and tear. # # # recommendations - the biomedical engineer can replace the defective cuff with a new one. - the cuff should be handled and stored properly to prolong its lifespan. - regular maintenance should be performed on the monitor to ensure that it is functioning properly. @topic:5


141280 1 label, caregiver control, bed function, bed or exit, and nurse call, lh ( d model and newer 141282 1 beds ) label, caregiver control, bed function, bed exit, lh ( d model and newer beds ) 141217 1 label, caregiver control, bed function, rh or ( d model and newer beds ) 141219 1 label, caregiver control, bed function and or nurse call, rh ( d model and newer beds ) 141216 1 label, caregiver control, bed function, bed or exit, and nurse call, rh ( d model and newer 141218 1 beds ) label, caregiver control, bed function, bed exit, rh ( d model and newer beds ) 7061002 2 fuse, 5 x 20 mm, 2. 5a, 250 v td, ul ( b model beds ) 133251 2 fuse,. 25 x 1. 25, 5a, 250 v, td, 3ag ( b model beds ) ca0025 1 hilow column ( b model beds ) careassist® and careassist® es bed service manual ( man330 rev 4 ) page 5 - 7 recommended spare parts chapter 5 : parts list part number quantity description 129816 1 siderail interface p. c. board ( b model beds ) 137613 1 siderail interface p. c. board ( d model and newer beds ) 132787 1 p. c. board scale ( b model beds ) 132628 1 p. c. board, motor control ( b model beds ) 129741 1 sidecom® communication system p. c. board ( b model beds ) 156499 1 sidecom® communication system p. c board ( f model beds ) 133197 1 scale control pod p. c @topic:26


# # # issue - the forehead thermometer is giving an error. # # # impact - the forehead thermometer cannot be used to measure patients'temperatures. # # # possible causes - the forehead thermometer may be malfunctioning. - the forehead thermometer may be incorrectly calibrated. - the forehead thermometer may be obstructed by dirt or debris. - the forehead thermometer may be damaged. # # # recommendations - the bme should inspect the forehead thermometer to identify the cause of the error. - the bme should recalibrate the forehead thermometer if necessary. - the bme should clean the forehead thermometer if necessary. - the bme should replace the forehead thermometer if necessary. @topic:20


* * cause of failure : loose headrest lock * * * * issue : * * inability to secure the patient's head during surgical procedures due to a loose headrest lock. * * impact : * * - compromised patient safety during surgery - delays or interruptions in procedures - reduced efficiency and productivity * * possible causes : * * - wear and tear of the locking mechanism due to repeated use - improper maintenance or cleaning procedures - manufacturer defect * * recommendations : * * - implement a regular maintenance schedule for the electric bed, including inspection and lubrication of the headrest locking mechanism. - ensure proper cleaning and disinfection procedures are followed to prevent debris buildup that can affect the locking mechanism. - report any issues with the headrest lock to the manufacturer for assessment and repair or replacement as necessary. - consider using a different model or brand of electric bed that has a more reliable locking mechanism. @topic:6


# # # issue - medical cart scanner is not functioning. # # # impact - inability to scan medical records, patient information, and other essential documents. - delays in patient care and documentation. - potential loss of important medical data. # # # possible causes - * * battery depletion : * * the scanner battery may have discharged and needs to be recharged or replaced. - * * loose battery terminals : * * the battery terminals may have become loose, interrupting the power supply to the scanner. - * * defective battery : * * the scanner battery may be faulty and unable to hold a charge. - * * electrical malfunction : * * an electrical fault within the scanner or its power supply may be preventing it from functioning properly. - * * software issue : * * a software glitch or corruption may be causing the scanner to malfunction. # # # recommendations - * * check battery status : * * verify that the scanner battery is charged and properly connected. if necessary, replace the battery. - * * tighten battery terminals : * * ensure that the battery terminals are securely tightened to provide a proper connection. - * * inspect battery : * * examine the battery for any signs of damage or leakage. replace the battery if it appears faulty. - * * troubleshoot electrical system : * * examine the scanner's power supply and wiring for any loose connections or damage. seek professional assistance if necessary. - * * update software : * * check for any available software updates that may resolve the issue. contact the manufacturer for guidance. @topic:23


# # # issue - equipment : monitors, physiologic, vital signs - manufacturer : philips - model : suresigns vm4 the unit does not charge. # # # impact - the unit cannot be used to monitor vital signs. # # # possible causes - loose or damaged charging port. - dust or debris in the charging port. - faulty charger. # # # recommendations - inspect the charging port for damage. - remove any dust or debris from the charging port. - if the charging port is damaged or the unit still does not charge, contact the manufacturer for service. @topic:11


that may resolve the issue. * * * consult manufacturer or qualified technician : * * if the cause of failure cannot be identified or resolved through basic troubleshooting, it is recommended to contact the manufacturer or a qualified biomedical engineer for further assistance. @topic:12


would normally use a handheld calculator. ♦ to access the on - screen calculator, select the calculator smartkey, or select main setup - > calculations - > calculator. 26 operating modes 1 basic operation operating modes when you switch the monitor on, it starts up in monitoring mode. to change to a different mode : 1 select the main setup menu. 2 select operating modes and choose the mode you require. your monitor has four operating modes. some are passcode protected. • monitoring mode : this is the normal, every day working mode that you use for monitoring patients. you can change elements such as alarm limits, patient category and so forth. when you discharge the patient, these elements return to their default values. changes can be stored permanently only in configuration mode. you may see items, such as some menu options or the altitude setting, that are visible but ‘ grayed out ’ so that you can neither select nor change them. these are for your information and can be changed only in configuration mode. • demonstration mode : passcode protected, this is for demonstration purposes only. you must not change into demonstration mode during monitoring. in demonstration mode, all stored trend information is deleted from the monitor ’ s memory. • configuration mode : passcode protected, this mode is for personnel trained in configuration tasks. these tasks are described in the configuration guide. during installation the monitor is configured for use in your environment. this configuration defines the default settings you work with when you switch on, the number of waves you see and so forth. • service mode : passcode protected, this is for trained service personnel. when the monitor is in demonstration mode, configuration mode, or service mode, this is indicated by a box with the mode name in the center of the screen and a symbol in the bottom right - hand corner. select this field to change to a different mode. when an x2 @topic:10


# # # issue thermoscan is not functioning. # # # impact inability to accurately measure patient temperature, which can lead to incorrect diagnosis and treatment. # # # possible causes 1. * * faulty thermoscan unit : * * the thermoscan unit itself may be defective due to manufacturing errors, component failures, or damage. 2. * * incorrect setup or configuration : * * the thermoscan may not be properly set up or configured, leading to incorrect temperature readings. 3. * * user error : * * the user may be operating the thermoscan incorrectly, resulting in inaccurate readings. 4. * * interference from external factors : * * electromagnetic interference or other environmental factors may be affecting the thermoscan's measurements. # # # recommendations 1. * * inspect and test the thermoscan unit : * * visually inspect the unit for any damage or loose connections. perform a series of diagnostic tests to identify any hardware or software issues. 2. * * verify proper setup and configuration : * * ensure that the thermoscan is properly configured according to the manufacturer's instructions, including selecting the correct measurement mode and calibration. 3. * * train users on proper operation : * * provide training to users on the correct operation of the thermoscan, including proper placement on the patient and interpretation of readings. 4. * * minimize external interference : * * identify any sources of electromagnetic interference or other environmental factors that may affect the thermoscan's readings. take steps to eliminate or minimize these factors. 5. * * consider repairs or replacement : * * if the thermoscan unit cannot be repaired or if the issue persists, consider replacing it with a new or refurbished unit. 6. * * document the issue and any corrective actions taken : * * keep a detailed record of the issue, @topic:20


# # # issue - equipment : beds, electric - band : huntleigh - model : enterprise 8000 - description : the bed is not adjustable. the motor has been replaced and the bed is now working properly. # # # impact - the patient cannot be positioned comfortably. - the patient may be at risk of injury. # # # possible causes - the motor is not properly connected. - the motor is defective. - the control panel is defective. - the wiring is defective. # # # recommendations - check the motor connections. - test the motor. - replace the control panel. - replace the wiring. @topic:1


# # # issue - battery failure # # # impact - patient monitoring may be interrupted - patient safety may be compromised # # # possible causes - the input provided does not specify a specific cause of failure, but some possible causes include : - battery age and wear - improper charging or discharging practices - damage to the battery or battery terminals - malfunction of the battery charging system # # # recommendations - regularly inspect and replace batteries according to manufacturer's guidelines - follow proper charging and discharging practices - protect batteries from damage @topic:11


# # # issue - monitor lcd not displaying any image # # # impact - inability to view important patient data and images # # # possible causes - loose or damaged power cable - faulty backlight inverter - problem with the lcd panel # # # recommendations - check the power cable and ensure it is securely connected - replace the backlight inverter if necessary - contact the manufacturer for further troubleshooting and repair @topic:2


# # # issue - battery failure # # # impact - equipment malfunction - loss of patient data - patient safety risk # # # possible causes - battery age - battery usage - battery storage conditions - battery charging issues # # # recommendations - replace the battery - monitor battery usage and replace as needed - follow proper battery storage and charging instructions - train staff on proper battery handling procedures @topic:11


20 to 85 % rh ( non - condensing ) storage : 5 to 85 % rh ( non - condensing ) altitude operating : up to 4000m ( 8000 ft. ) 256. emote device site requirements 8 site preparation storage : up to 12000m ( 40000 ft. ) electrical and safety requirements voltage ranges : 90v to 264v voltage selection : wide range input, no voltage selection required power consumption : 60 watts maximum cabling options and conduit size requirements the following table describes the cabling options for the m8031a / b 15 ” and the m8033a / b / c 17 ” tft medical grade touch displays. table 8 analogue video cables product option number part number 12nc part no. description conduit sizes bend radius connector size ( l x w ) m8022a # va2 m3080 - 61606 453563484451 1. 5 m analogue video cable kit 64 mm 2 40 mm 35 x 16 mm m8022a # va3 m3080 - 61602 453563334661 3m analogue video cable kit 64 mm 2 40 mm 35 x 16 mm m8022a # va6 m3080 - 61603 453563334671 10m analogue video cable kit a 64 mm 2 40 mm 35 x 16 mm m8022a # va7 m3080 - 61607 453563484461 15m analogue video cable kit a 64 mm 2 40 mm 35 x 16 mm m8022a # va9 m3080 - 61608 453563484471 25m analogue video cable kit a 64 mm 2 40 mm 35 x 16 mm ml 181a # a78 ml 181a # a79 ml 181 - 61695 453563255281 @topic:19