made to eeg settings apply to both eeg channels. switching eeg numerics on and off each eeg numeric can be individually switched on or off in the setup eeg menu. 1 in the setup eeg menu, select the numeric label. 2 select on / off to toggle between the settings. changing the scale of the eeg waves for display this only changes the visual appearance of the wave. it does not affect the signal analyzed by the monitor or printed in reports or recordings. 1 in the setup eeg menu, select wave scale to call up a list of wave scales. 2 select the required scale from this list. scaling information is displayed with each eeg wave. • if show gridlines is set to on in configuration mode, gridlines and the current wave scale values are shown with the eeg wave. • if show gridlines is set to off in configuration mode, the current wave scale is indicated by a size bar beside the eeg wave. 254 eeg reports 18 monitoring eeg changing filter frequencies the low and high pass filters screen out undesirable interference from the raw eeg wave display. the current eeg filter frequency settings are shown in the header of the csa. changing filter settings affects the eeg wave and all the eeg numerics. whenever the filter setting is changed, a filter change marker appears next to the spectral lines. to change the filter settings : 1 in the setup eeg menu, select low filter or high filter to call up a list of available frequencies. 2 select the required frequency from this list. changing the speed of the eeg wave the eeg measurement has its own speed control and is not affected by the wave speed settings of the other measurements. ♦ in the setup eeg menu, select eeg speed. choose the required speed from the pop - up list. this defines the speed at which the wave @topic:8


you do not acknowledge them. when alarms are set to non - latching, their indicators end when the alarm condition ends. switching alarm latching on means that visual and / or audible alarm indications are still displayed or announced by the monitor after the alarm condition ends. the indication lasts until you ge the alarm. viewing the alarm latching settings to see the alarm latching setting for your monitor in the monitor's main setup menu, select alarms. 1 2 select alarm settings, and see the visual latching and audible latching settings. this setting can only be changed in configuration mode. you should be aware of the settings chosen for your unit. there are three possible choices each for visual and audible latching, red only, red and yellow, and off. these choices can be combined to give the following settings : audible latching visual latching ed & yelred & yellow red only red & yellow ed & yellow off red only ed onled onloff off off alarm latching behavior visual and audible latching visual latching, audible red & yellow measurement alarms non - latching alarm tone on. alarm lamp onalarm message. flashing numerics. alarm has not alarm tone on. alarm lamp on. alarm message. flashing numerics. alarm tone on. alarm lamp on. alarm message. flashing numerics. alarm condition till been present. cknowledged. alarm tone on. larm lamp on. alarm message. flashing numerics. alarm condition no longer present. all audible and visual alarm indicators automatically stop. alarm message. flashing numerics. audible alarm indicatorautomatically stop. alarm tone off. alarm lamp off. alarm message. flashinnumerics. audible alarm reminder ( if configured ). alarm has been acknowledged. alarm tone off. alarm lamp off. alarm tone off. alarm lamp alarm condition off. alarm message. flashinnume @topic:8


# # # issue - bed alarm box connection broken and damaged # # # impact - bed alarm system may not function properly, potentially compromising patient safety. # # # possible causes - wear and tear on the connection - improper handling or storage - loose or damaged wiring # # # recommendations - inspect the connection for signs of damage or wear. - ensure proper storage and handling procedures are followed. - tighten or replace loose or damaged wiring. - consider using a more durable connection type or protective casing. @topic:26


# # # issue - the bed is not adjusting as expected. # # # impact - patients may be unable to get in or out of bed. - patients may be uncomfortable or in pain. - nurses and other caregivers may be unable to provide proper care. # # # possible causes - the electrical system is not functioning properly. - the mechanical system is not functioning properly. - the control panel is not functioning properly. - the bed is not properly calibrated. - the bed is overloaded. # # # recommendations - check the electrical system for any loose connections or damaged wires. - check the mechanical system for any broken or jammed parts. - check the control panel for any loose buttons or switches. - calibrate the bed according to the manufacturer's instructions. - ensure that the bed is not overloaded. @topic:1


# # # issue - monitor is not displaying oxygen values # # # impact - clinicians cannot accurately monitor patient's oxygen levels, which can lead to delayed diagnosis and treatment. # # # possible causes - * * loose signal cable : * * the cable connecting the monitor to the patient's finger probe may have become loose, causing the monitor to not receive oxygen level data. - * * faulty finger probe : * * the finger probe that measures the patient's oxygen levels may be faulty, leading to inaccurate readings. - * * software error : * * a software error in the monitor may be preventing it from displaying oxygen values correctly. # # # recommendations - * * inspect and tighten the signal cable : * * check the cable connecting the monitor to the finger probe and ensure that it is securely connected at both ends. - * * replace the finger probe : * * try using a different finger probe to see if the issue persists. - * * contact the manufacturer : * * if the issue persists after checking the cable and finger probe, contact the manufacturer of the monitor for support. @topic:4


* * issue : * * knob is loose. * * impact : * * the knob may not function properly, which could lead to inaccurate readings or difficulty in operating the device. * * possible causes : * * - the knob was not properly tightened during assembly. - the knob has become loose over time due to wear and tear. - the knob has been subjected to excessive force. * * recommendations : * * - tighten the knob securely using the appropriate tools. - inspect the knob regularly for signs of wear and tear. - avoid applying excessive force to the knob. @topic:26


# # # issue - cuff expired. # # # impact - incorrect blood pressure readings. - patient harm. # # # possible causes - cuff not replaced according to manufacturer's recommendations. - cuff damaged or punctured. # # # recommendations - replace cuff according to manufacturer's recommendations. - inspect cuff for damage or punctures before each use. - train staff on proper cuff maintenance and replacement procedures. @topic:3


# # # issue - module replacement received from the company was defective. # # # impact - device is not functioning properly, compromising patient care. # # # possible causes - module was damaged during transit. - module was not properly tested before shipment. - company sent the wrong module. # # # recommendations * * immediate actions : * * - verify that the module received matches the order. - contact the company for a replacement module. * * preventive actions : * * - establish clear communication protocols with the company to ensure correct module delivery. - request the company to implement more rigorous testing procedures before shipping modules. - consider stocking spare modules to minimize downtime in case of module failures. * * root cause analysis : * * - conduct a thorough investigation to determine the root cause of the failure : - check shipping records for evidence of damage. - review the company's testing procedures. - verify that the order was processed correctly. - implement corrective measures based on the findings of the investigation. @topic:12


# # # issue - adapter co2 is faulty. # # # impact - cal co2 failed. # # # possible causes - the adapter may be damaged physically, such as a loose connection or broken wire. - the adapter may be malfunctioning due to electrical issues, such as a short circuit or power surge. - the adapter may be incompatible with the monitor, either due to wrong specifications or a lack of proper configuration. # # # recommendations - inspect the adapter for any physical damage and ensure it is properly connected. - check the electrical connections of the adapter and ensure they are secure and provide a stable power supply. - verify that the adapter is compatible with the monitor, both in terms of specifications and configuration. - replace the faulty adapter with a new one from the manufacturer or an authorized distributor. - consider using a surge protector to prevent damage to the adapter due to power surges. - contact the manufacturer or authorized service provider for further troubleshooting and repair if the issue persists. @topic:2


# # # issue barcode scanner not working. # # # impact inability to scan barcodes, which can lead to errors in patient identification and medication administration. # # # possible causes * loose battery connection * defective barcode scanner * software issue # # # recommendations * check the battery connection and ensure it is secure. * if the battery connection is secure, replace the barcode scanner. * if the scanner and battery are both fine, check for any software updates that may resolve the issue. @topic:23


# # # issue - cuff deterioration # # # impact - impaired blood pressure measurement accuracy # # # possible causes - prolonged use and wear - inadequate cleaning and maintenance - exposure to extreme temperatures or moisture # # # recommendations - regularly inspect cuff for signs of wear and tear - clean and disinfect cuff as per manufacturer's instructions - store cuff in a dry, temperature - controlled environment - replace cuff when necessary ( typically every 6 - 12 months ) @topic:3


the “ troubleshooting ” section for remedies. battery has no power left : if the monitor is not running on ac power : battery will switch off power delivery at any moment - in this case recharge the battery immediately - or, if the monitor is running on ac power, the battery is in deep discharge and requires pre - charging to restore communication. to avoid this condition charge batteries to 50 % for storage. note that the battery malfunction inop will eventually be issued if the pre - charging does not restore battery communication within about 4 minutes. battery temperature too high : this symbol is displayed if the battery temperature goes above 65°c. in addition the inop message check batt temp is displayed. if the battery temperature increases further above 70°c the batteries will switch off for safety reasons. allow the battery to cool down to avoid the monitor switching off. 102 viewing battery details ^ to view detailed information for the battery, select the pop - up key batt. chewistr > cycles 5 qe printing battery status to print all battery information in the battery status window, 8. select the battery status information on the screen or select main setup - > battery to open the battery status window 9. select the print status pop - up key to print the information on a connected printer. checking battery charge ^ to check the charge status of a battery in a monitor, see the battery power gauge on the screen or select main setup - > battery to enter the battery status window. ^ tocheck the charge status of a battery that is not connected to a monitor or battery charger, press the black dot marked “ push ” on the labeled side of the battery. the remaining charge is indicated by four leds on the electronic fuel gauge directly above the dot. each led represents 2540 of charge. if all leds are lit, the battery is fully charged, if only one @topic:11


# # # issue - bed not adjusting # # # impact - patient discomfort - difficulty in providing patient care # # # possible causes - faulty remote control connector - damaged remote control - loose or disconnected wires - malfunctioning control unit - software glitch # # # recommendations - inspect the remote control connector for damage or loose connections. - test the remote control to ensure it is functioning properly. - check the wiring harness for any loose or disconnected wires. - contact the manufacturer for troubleshooting and repair instructions if the control unit is suspected to be malfunctioning. - check for software updates that may address the issue. @topic:7


# # # issue - remote is not working # # # impact - bed cannot be adjusted # # # possible cause - * * defective remote : * * remote may be damaged or malfunctioning, preventing it from sending signals to the bed. # # # recommendations - repair or replace the remote. - ensure proper communication between the remote and the bed. @topic:7


# # # issue - the bed's side rails are not moving up and down. # # # impact - the patient may fall out of bed if the side rails are not in place. - the patient may be uncomfortable if the side rails are not in the correct position. # # # possible causes - the locking mechanism is not engaged. - the motor is not working. - the control panel is not functioning. # # # recommendations - check the locking mechanism to ensure that it is engaged. - check the motor to ensure that it is working. - contact the manufacturer for further troubleshooting. @topic:6


# # # issue - cuff degradation # # # impact - accurate blood pressure readings cannot be obtained. # # # possible causes - cuff is old and has been used multiple times. - cuff is not properly inflated or deflated. - cuff is not the correct size for the patient. # # # recommendations - replace the cuff with a new one. - ensure that the cuff is properly inflated and deflated. - use the correct size cuff for the patient. @topic:3


.. event setushow episodopen the event episoshow review review group choose a different event group for reviewing in the event review window ( advanced event surveillance only ). show summarmanual event start car start a manually - triggered event capture. open the start car window to start a car seat assessment record ( car ) running car. open car histogrm open the window showing the car spo2 histogram move the cursor left or right to the next event to navigate through the events database. placing the cursor over an event highlights it and shows the event values for the selected event. jump to the first or last event in the event database. k 335 26 event surveillance selecting this pop - up key lets you.... pop - up keys graphic iew toggle between a tabular and graphic version of the event episode window vitals view toggle between a tabular and graphic version of the event review window table graphic car review review review delete event delete the currently - selected event from the database. the monitor asks you to confirm this deletion. you cannot retrieve deleted events. select annotatn access the list of available annotations to add a nursing note for the current vent episode ( advanced event surveillance only ). access the printing and recording pop - up keys to document events. vent triggeryou can trigger event capture manually, for example, if you want to record a patient's condition before a procedure. you can also set events to trigger automatically, when the patient's values cross a predefined threshold value, or when a particular measurement or procedure is carried out. if more than one trigger is available for the measurements in the active event group, the trigger condition may be at least one param., at least two param., at least three par., or all four parameter. if @topic:8


7 kpa ) neonate : 80 mmhg ( 10. 7 kpa ) ap lower limit enableno adult : 50 mmhg ( 6. 7 kpa ) map : lower limit pediatric : 50 mmhg ( 6. 7 kpa ) neonate : 35 mmhg ( 4. 7 kpa ) advanced default view sys _ dia _ map ube typtubes unit of measure mmhg surebp 146 appendix welch [UNK] connex® spot monitor default value settings cuff inflation target ( step algorithm ) dul160 mmhg ( 21. 3 kpa ) pediatric 40 mmhg ( 18. 7 kpa ) 0 mmhg ( 12. 0 kpaneonate aximum cycle pressuradult automatic pediatric automatic intervals default value settings time between each automatic interval 15 minutes intervals profile default intervals mode automatic office profile intervals mode selected interval parametenibp automatic mode time interval 15 minutes. podefault value settings yes adult : 100 % pediatric : 100 % neonate : 100 % po2 lower limit elower limit adult : 90 % pediatric : 90 % spodefault view service manual default value settings advanced spo2 alarm condition delay 0 seconds satseconds tm ( nellcor only ) 25 seconds rr rrp alarm condition delay - the quickest detection of a change in respiratory rate would take at least two breaths. depending on respiratory rate, anything less than 6 breaths per minute must be ettings default value yes adult : 30 pediatric : 30 eonate : 80 rr lower limit enabled yes lower limit adult : 6 pediatric : 6 neonate : 12 default view bpm temperature default value settings temperatno 101 °f ( 38. 3 °c ) no 94 °f ( 34. 4 °c ) lower limit advanced unit of measure °f ( fahrenheit ) yes 148 appendix welch [UNK] connex® spot monitor default value settings default suretemp plus site @topic:17


and 40 mmhg : ±2. 2 mmhg ( ±0. 29 kpa ) for values between 40 and 76 mmhg : ±5. 5 % of reading the specifications are valid for 45 % 0 2 and n 2 or n 2 0 balance. outside these conditions the accuracy reaches at a minimum the requirements of en864 / is09918. resolution numeric : 1. 0 mmhg ( 0. 1 kpa ) wave : 0. 1 mmhg ( 0. 01 kpa ) stability ±1. 0 mmhg over a 7 day period awrr range 0 to 150 rpm accuracy ±2 rpm warm - up time 20 minutes with c0 2 transducer attached for full accuracy specification response time less than 125 ms ( for step from 10 % to 90 % ) mainstream c0 2 humidity correction factor 441 38 installation and specifications measurement specifications either btps or atpd can be selected as the humidity correction factor for the mainstream c0 2 readings. the formula for the correction calculation is : abs patpd pbtps'p p r abs r h20 where p = partial pressure, p a (, s = absolute pressure, and ph 20 = 47 mmhg @ 37°c and 100 % rh. co 2 alarm specifications range adjustment delay etc0 2 high 20 to 95 mmhg ( 2 to 13 kpa ) 1 mmhg ( 0. 1 kpa ) m3002a / m3014a / m3016a : less than 14 seconds m3015a : less than 18 seconds. etc0 2 low 10 to 90 mmhg ( 1 to 12 kpa ) imc02 high 2 to 20 mmhg ( 0. 3 to 3. 0 kpa ) steps of 1 mmhg ( 0. 1 kpa ) m3002a / m3014a / m3016a : less than 14 seconds m301 @topic:17


# # # issue * * equipment not functional due to being unavailable * * # # # impact * inability to provide care to patients * delays in treatment * potential safety hazards # # # possible causes * * * equipment loaned out to another department or ward * * * equipment stolen or misplaced * lack of documentation or communication regarding equipment usage # # # recommendations * * * establish clear policies and procedures for equipment checkout and return. * * * * * use a tracking system to monitor equipment location and usage. * * * * * train staff on proper equipment handling and storage. * * * * * conduct regular audits to ensure compliance with equipment policies. * * * * * consider using a shared equipment pool to reduce the likelihood of equipment being unavailable due to loan outs. * * @topic:12


# # # issue the spo2 sensor cable is intermittently connecting and disconnecting. # # # impact this issue can lead to inaccurate or missing spo2 readings, which can delay or hinder patient care. # # # possible causes * damaged or loose cable connections * faulty cable or sensor * interference from other nearby devices # # # recommendations * inspect the cable connections and ensure they are tight and secure. * replace the spo2 cable or sensor if necessary. * check for potential sources of interference in the vicinity of the device and move them away if possible. @topic:4


# # # issue the provided input indicates that a physiologic monitor ( bedside ) manufactured by welch allyn, specifically model propaq lt, has experienced a failure. the cause of the failure is reported as " monitor propaq bed3. " # # # impact the impact of this failure is not explicitly stated in the given input. however, a physiologic monitor is a critical medical device used to monitor vital parameters such as heart rate, blood pressure, and oxygen saturation. a failure in this device could potentially affect the timely detection and management of any medical emergencies. # # # possible causes the exact cause of the failure is not specified in the provided input. however, there are several possible causes that could have contributed to the malfunction of the device, such as : - * * hardware failure : * * this could include issues with the electrical components, sensors, or display of the monitor. - * * software glitches : * * the monitor may be running on outdated or malfunctioning software, leading to errors or system crashes. - * * environmental factors : * * the monitor may have been exposed to extreme temperatures, humidity, or other environmental conditions that have compromised its performance. - * * operator error : * * the monitor may have been used improperly or mishandled, leading to damage or malfunction. # # # recommendations to address this failure and prevent similar issues in the future, the following recommendations could be considered : - * * proper maintenance and calibration : * * regularly scheduled maintenance and calibration of the monitor by qualified technicians can help identify and fix potential issues early on, preventing major failures. - * * software updates : * * ensure that the monitor is running on the latest software version, as updates may include bug fixes and performance improvements. - * * environmental monitoring : * * monitor the operating environment to ensure @topic:2


# # # issue - cuff bp issue # # # impact - blood pressure readings may be inaccurate # # # possible causes - cuff bp may be damaged or not properly inflated # # # recommendations - replace cuff bp - ensure cuff bp is properly inflated @topic:3


# # # issue - cuff of the vital signs monitor is damaged. # # # impact - cannot accurately measure blood pressure. # # # possible causes - cuff is old and worn out. - cuff is not properly inflated. - cuff is too small or too large for the patient. # # # recommendations - replace the cuff with a new one. - make sure the cuff is properly inflated. - choose the correct size cuff for the patient. @topic:5


# # # issue - sat failure # # # impact - nibp monitor is not able to take blood pressure measurements. # # # possible causes - the sat sensor may be faulty. - the sat sensor may be disconnected. - the sat sensor may be miscalibrated. # # # recommendations - replace the sat sensor. - reconnect the sat sensor. - calibrate the sat sensor. @topic:15


a numeric, you can also substitute a larger analog or 1 select the numeric you want to substitute. select change numeric. 2 select clock. 3 when you and time. select the clock on the main screen you can select date, time in the menu to set the date 98 34 respiratory loops using a spirometry module or a vuelink / intellibridge module connected to a ventilator, you can measure and store graphic representations of realtime respiratory loops. respiratory loops can help you recognize changes in your patient's lung function, and they can also indicate a fault in the airway tubing ( for example, if the respiratory loop does not close ). you can measurepressure - volume loops • pressure - flow loop • olume - flow loops • note that you cannot store loops from different patients and different source devices in the same lisas they are patient and device specific. this prevents you from inadvertently comparing information from different patients. viewing loops in the loops display, the current loop is drawn in white, and up to six stored loops are color - coded to match their timestamps. the currently - used source device is shown in the window title. to view respiratory loops permanently on your screen, select the current screen name to open the change screen menu, then select a screen configured t.. select the loops screen element to access the loops pop - up keys. x awp mode1 1. 0 awv mode1 1. 0 w 0. 5 0. 0 pip peep 8. 5 60. 80 0. 0 minvol tv 25 1. 90 awp 16 apr 16 : 50 16 apr 16 : 50 16 apr 16 : 52 loop capture in progress 399 34 respiratory loops the loops window, and its associated pop - up keys, to open select main setup then loops, or select the loops smartkey, if configured. • in the loops @topic:8


# # # issue the suresigns vs4 monitor is reading low body temperature values. # # # impact this could lead to incorrect diagnoses and treatment decisions. # # # possible causes - faulty temperature probe - incorrect calibration of the monitor - electrical interference from other devices - environmental factors ( e. g., cold ambient temperature ) # # # recommendations - check the temperature probe for damage or contamination. - calibrate the monitor according to the manufacturer's instructions. - check for electrical interference from other devices. - ensure that the monitor is being used in an appropriate environment. @topic:20


see “ x2 controls and indicators ” on page 14 ) 3 3. 5 - inch tft lcd touchscreen qvga display 4 alarm lamps ( see “ x2 controls and indicators ” on page 14 ) 5 battery eject button 6 hard keys ( see “ x2 controls and indicators ” on page 14 ) 7 measurement connectors ( see “ x2 patient connectors, right side ” on page 14 ) 8 battery compartment 13 1 basic operation devices for acquiring measurements x2 controls and indicators 6 7 1 on / standby switch 2 on / standby led. green when monitor is on. red indicates an error. 3 battery status led. yellow when charging. flashing red when battery is empty. 4 external power led. green when monitor is powered from an external power source. 5 alarms off indicator. when alarms are suspended, the lamp is red, and the alarms off symbol is shown. 6 active inop alarm lamp in light blue. blinks until active inop is acknowledged. 7 active alarm lamp. red or yellow, depending on alarm level. blinks until active alarm is acknowledged. 8 silence key 9 alarms key : turns alarms on / off, or pauses them 10 smartkeys key : brings up smartkeys on the screen 11 main screen key : closes all open menus / windows and returns to the main screen. x2 patient connectors, right side showing symbols version ( international ) and text version ( english only ) pressure ( option ) temperature ( option ) noninvasive blood pressure s p o 2 ecg sync pulse output ecg / respiration c0 2 ( option in place of pressure and temperature ) 14 devices for acquiring measurements 1 basic operation x2 left side 1 2 1 loudspeaker 2 msl connector. connects to the external power supply or a host monitor via the msl cable for ac mains operation, battery charging, and communication with a @topic:10


a known volume and temperature is injected into the right atrium through a cvp catheter. the injectate bolus mixes with the blood in the heart and the change in blood temperature is measured with a thermistor at the distal end of an arterial catheter placed in one of the bigger systemic arteries, for example, the femoral or the axillary artery. the picco method requires a pressure measurement made using either the m1006b pressure module, or a m3001a / m3002a mms, or a measurement extension module m3015a, m3014a, or m3012a. ( a pressure from an external device may not be used. ) you will also need a conventional central venous ( cvp ) line and an arterial catheter from pulsion medical systems. you must use the pproved catheters and puncture locations. measuring continuous cardiac output every time c. o. is measured with the picco method, the monitor uses this c. o. value and the result of the pulse contour analysis to calculate a patient - specific calibration factor. the monitor uses this value to compute cco and the other continuous hemodynamic parameters. cco values are calculated on a beat - to - beat basis and then averaged over a 12 - second time frame. the calculated values are displayed as numerics on the monitor screen. the monitor uses cco, an arterial pressure, and cvp to calculate a continuous svr value. if a continuously measured cvp value is not available, the monitor uses a preset, static cvp value to calculate the svr ( you will see the inop message svr set cvp used ). if the bsa is available, the monitor uses the cci to calculate the svri. a cco or cci from an external device can also @topic:17


# # # issue remote is not functioning. # # # impact inability to adjust the bed. # # # possible causes - remote is damaged. - remote is not paired with the bed. - batteries in the remote are dead. # # # recommendations - inspect the remote for any damage. - pair the remote with the bed according to the manufacturer's instructions. - replace the batteries in the remote. @topic:7


# # # issue - loss of blood pressure ( bp ) monitoring capability. # # # impact - inability to monitor patient's blood pressure, leading to potential delays in detecting critical changes and providing appropriate treatment. # # # possible causes - missing bp line. # # # recommendations - replace the missing bp line with a new one to restore monitoring capability. - investigate the reason for the bp line going missing to prevent future occurrences, such as : - theft or misplacement by staff - damage during transport or storage - lack of proper inventory management @topic:14


# # # issue - patient leg positioning device is stuck in place. # # # impact - inability to properly position the patient for the procedure. - delay in the procedure. - potential discomfort or pain for the patient. # # # possible causes - loose screws or bolts. - corrosion or rust. - mechanical wear and tear. - misalignment or damage to the device. # # # recommendations - inspect the device for any loose screws or bolts. tighten if necessary. - clean and lubricate the device to address any corrosion or rust. - check for any mechanical wear and tear or damage. replace or repair the device if necessary. - ensure that the device is properly aligned and installed. @topic:1


# # # issue - lcd monitor split into two screens on both the primary and secondary displays # # # impact - compromised image quality and visual clarity, leading to potential inaccuracies in patient monitoring and diagnostic interpretation # # # possible causes - incorrect or incomplete calibration settings - software or firmware glitches - loose connections or damaged cables - hardware malfunction within the monitor or its power supply - compatibility issues with other connected devices ( e. g., bmc11952, bmc12008 _ 2 ) # # # recommendations - * * verify calibration settings : * * ensure that the monitor is calibrated according to the manufacturer's specifications. - * * update software / firmware : * * check for available software or firmware updates and install them to address potential glitches. - * * inspect connections : * * tighten loose connections and replace damaged cables. - * * troubleshoot hardware : * * perform diagnostic tests to identify potential hardware faults. if necessary, contact a qualified biomedical engineer for further assistance. - * * review compatibility : * * confirm that the monitor is compatible with the other connected devices, especially bmc11952 and bmc12008 _ 2. check for any known compatibility issues or consult the manufacturer's documentation. @topic:2


# # # issue : - bed, electric ( uq2000b ) is not adjustable in height. # # # impact : - inability to adjust patient positioning for comfort, medical procedures, or emergencies. # # # possible causes : - malfunctioning motor - damaged gear or drive system - electrical connection issues - software or control system failures # # # recommendations : - * * inspect motor : * * verify if the motor is receiving power and if it is operating correctly. - * * examine gear and drive system : * * check for any broken or worn gears, chains, or belts. - * * troubleshoot electrical connections : * * ensure proper voltage and continuity to the motor and control system. - * * review software and control system : * * update or reprogram if necessary to resolve any functional issues. - * * mechanical misalignment : * * check if the bed frame or other components are properly aligned and not obstructing movement. - * * overload : * * ensure the bed is not overloaded beyond its specified weight capacity. @topic:1


# # # issue : mainboard failure in philips intellivue mp5 ( m8105a ) physiologic monitor, leading to device malfunction ( opening and immediately turning off ). # # # impact : - inability to monitor vital patient parameters, such as heart rate, blood pressure, and oxygen saturation. - delay in diagnosis and treatment due to lack of accurate patient data. - potential risk to patient safety if the monitor is not functional during critical procedures or emergencies. # # # possible causes : - power supply issues ( e. g., voltage fluctuations, surges ) - overheating due to poor ventilation or blocked air vents - component failure ( e. g., capacitors, resistors ) - software malfunction or corrupted firmware - physical damage to the mainboard ( e. g., from drops, impacts ) - ageing and wear of electronic components # # # recommendations : - ensure a stable and reliable power supply for the device. - regularly check and clean air vents to prevent overheating. - conduct preventive maintenance checks to identify and replace any failing components. - update firmware as recommended by the manufacturer. - handle the device with care and avoid physical damage. - monitor the device's performance and report any unusual behavior or error messages. - consider using a redundant or backup monitoring system to ensure continuity of patient monitoring. @topic:2


# # # issue - cuff on vital signs monitor is deteriorated. # # # impact - inaccurate vital signs measurements. - potential for patient harm. # # # possible causes - cuff deterioration due to age or excessive use. - cuff not properly stored or maintained. - cuff not compatible with the monitor. # # # recommendations - replace the deteriorated cuff with a new one. - ensure the cuff is compatible with the monitor. - properly store and maintain the cuff to prevent further deterioration. - regularly inspect the cuff for signs of wear and tear. @topic:5


# # # issue - monitor is not displaying readings # # # impact - inability to monitor patient's vital signs # # # possible causes - loose connector - faulty cuff # # # recommendations - check and tighten connector - test cuff functionality @topic:5


into a monitor connected to mains power. the battery led will light yellow to indicate that charging is in process ( it can take up to three minutes before charging begins ). 2 charge the battery until it is full, the battery led is green, and the battery power gauge is filled. in certain situations, where many measurements are in use plus the recorder, the load on the monitor may be so high that the batteries will not charge. in this case you must use the 865432 smart battery conditioner ( previously m8043a smart battery conditioner ) to charge the battery. conditioning a battery you must condition a battery when its " battery requires maintenance " symbol shows on the screen. do not interrupt the charge or discharge cycle during conditioning. you can condition a battery in a caution do not use a monitor that is being used to condition batteries to monitor patients. the monitor switches off automatically when there is no battery power left. it is, however, preferable to use the 865432 smart battery conditioner ( previously m8043a ) for external battery conditioning. the smart battery conditioner automatically performs the correct charge or conditioning process and evaluates the capacity when fully charged. for details please see the instructions for use for the smart battery conditioner. do not use any other battery chargers or conditioners. to condition a battery using a monitor, insert the battery into a monitor connected to mains power. 1 switch the monitor power off. 2 charge the battery until it is completely full. open the battery status window and check that the batt fully charged message is displayed. 3 disconnect the monitor from mains power, and let the monitor run until there is no battery power left and the monitor switches itself off. 4 reconnect the monitor to mains power and charge the battery until it is full for use or charge to 50 % for storage. 5 408 storing a battery warning remove @topic:11


* * issue : * * remote being stuck * * impact : * * unable to adjust the headrest of the bed * * possible causes : * * - mechanical issues with the remote control - electrical issues with the bed or its components - software glitches or firmware malfunctions * * recommendations : * * 1. * * test the remote control : * * verify if the remote control is functioning correctly by testing it on another bed or using a multimeter to check for continuity and voltage. 2. * * inspect the bed : * * visually inspect the bed for any loose connections, damaged wires, or malfunctioning components. 3. * * reset the bed : * * if the issue persists, try resetting the bed by unplugging it from the power outlet and waiting a few minutes before plugging it back in. 4. * * update firmware : * * if available, check if there is a firmware update for the bed and install it if necessary. 5. * * replace the remote or bed components : * * if the above steps do not resolve the issue, consider replacing the remote control or the affected components of the bed. 6. * * consult with a qualified technician : * * if all else fails, contact a qualified biomedical engineer or the manufacturer for further assistance and repair. @topic:7


# # # issue - monitor not displaying ppv ( peak positive pressure ) value. # # # impact - clinicians are unable to accurately monitor patient's respiratory function. - potential for delayed or inappropriate interventions. # # # possible causes - * * improper setup or configuration : * * the monitor may not have been correctly configured to display ppv. - * * sensor malfunction : * * the sensor responsible for measuring ppv may be damaged or malfunctioning. - * * software glitch : * * a software error may be preventing the monitor from displaying the ppv value. - * * electrical interference : * * external electrical noise may be interfering with the monitor's ability to receive the ppv signal. - * * loose connections : * * loose or damaged cables or connectors may be interrupting the connection between the sensor and the monitor. # # # recommendations - * * verify setup and configuration : * * ensure that the monitor is properly configured to display ppv. consult the manufacturer's instructions for guidance. - * * check sensor integrity : * * inspect the ppv sensor for any damage or loose connections. replace the sensor if necessary. - * * update software : * * check for and install any available software updates from the manufacturer. - * * eliminate electrical interference : * * identify and remove any potential sources of electrical interference near the monitor. - * * tighten connections : * * ensure that all cables and connectors are securely tightened. @topic:20


# # # issue the basket on the philips suresigns vm4 patient transport monitor has come loose. # # # impact the loose basket could pose a safety hazard to patients and staff. it could also interfere with the operation of the monitor. # # # possible causes * the locking nuts that hold the basket in place may have become loose. * the basket may have been damaged, causing it to come loose. * the monitor may have been dropped or otherwise mishandled, causing the basket to come loose. # # # recommendations * tighten the locking nuts that hold the basket in place. * inspect the basket for any damage. if the basket is damaged, it should be replaced. * inspect the monitor for any damage. if the monitor is damaged, it should be repaired or replaced. @topic:24


# # # issue - bed does not adjust # # # impact - inability to position the patient safely and comfortably - delays in patient care # # # possible causes - remote control lock is engaged - electrical or mechanical issue with the bed # # # recommendations - ensure the remote control lock is disengaged - inspect the electrical connections and mechanical components - if the problem persists, contact the manufacturer for professional assistance @topic:1


* * issue : * * cuff failure in vital signs patient monitor * * impact : * * inability to accurately monitor patient's vital signs, such as blood pressure, heart rate, and respiratory rate. * * possible causes : * * * * * natural wear and tear : * * over time, the rubber or plastic material used in the cuff can degrade and become damaged. * * * excessive use : * * frequent or prolonged use of the cuff can put excessive strain on the material, leading to premature failure. * * * improper storage : * * exposure to extreme temperatures or humidity can weaken the cuff material. * * * accidental damage : * * the cuff can be accidentally punctured or torn, causing a loss of pressure. * * * defective materials : * * in rare cases, the cuff may have been manufactured with defective materials that are prone to failure. * * recommendations : * * * * * regular inspection and maintenance : * * inspect the cuff regularly for signs of wear or damage. replace the cuff if any damage is found. * * * proper storage : * * store the cuff in a cool, dry place away from extreme temperatures and humidity. * * * adequate training : * * train staff on the proper use and storage of the vital signs patient monitor to minimize the risk of accidental damage. * * * consider high - quality brands : * * choose manufacturers with a reputation for producing durable and reliable cuffs. * * * establish a preventive maintenance program : * * regularly schedule preventive maintenance checks to identify and address potential issues before they cause failure. @topic:5


##and away from the articulating frame. the flexibility of the line manager lets you bend it in any direction. make sure the lines are not pinched or kinked and there is sufficient slack in the lines for bed articulations and patient movement. failure to do so could cause injury or equipment damage. do not wrap • 1. the safe working load specifications allow for a substantial safety margin. x - ray - transparent adjustable head section ( ad242a ) * * the x - ray - transparent adjustable head section accessory allows a cassette for 35 x 43 cm x - ray films ( as per the standard en iso 4090 ) to be installed in order to take chest x - rays. it is installed in place of the hard surface of the head section. [UNK] the type ( foam or air ), the materials, the density and the thickness of the mattress, and the weight and morphology of the patient can affect the quality of the x - ray images. the best way to produce x - rays of an optimal quality is to get as close to the patient as possible. the radiologist is responsible for deciding on the best solution to take the x - ray according to the medical target and the hospital's protocol adapted to the patient's illness. note : for patients weighing more than 100kg, the user must adjust the angle of the head section and the position of the patient to produce quality images. installing the accessory 1. remove the mattress to gain access to the hard surface of the head section. unclip and remove the hard surface of the head section. 2. 3. install and clip the accessory in its place. remove the headboard to install the x - ray cassette in the top of the head section. 1. raise the sleep surface or raise the head section in order to insert the cassette. 2. unhook the buckle of the right strap from its storage hook. @topic:9


# # # issue - faulty power cord # # # impact - interruption of medical device operation - patient safety hazard # # # possible causes - physical damage to the power cord - loose connections - faulty manufacturing # # # recommendations - replace the faulty power cord - inspect the power cord regularly for any damage - ensure that the power cord is securely connected to the medical device and the power outlet @topic:12


# # # issue - battery degradation and failure to hold charge # # # impact - inability to use the device in portable mode during emergencies or power outages # # # possible causes - battery life cycle exceeded - excessive charging or discharging cycles - exposure to extreme temperatures - physical damage to the battery # # # recommendations - regularly monitor battery health and replace when recommended by the manufacturer - avoid overcharging or completely discharging the battery - store the battery in a cool, dry place - handle the battery with care to prevent physical damage @topic:11


# # # issue - air hose connection failure # # # impact - interruption of patient monitoring or transport # # # possible causes - deterioration of air hose material - loose or damaged connection joints # # # recommendations - regularly inspect and replace air hoses as needed - ensure secure connections by tightening joints and using appropriate connectors - train staff on proper handling and maintenance of equipment @topic:12


# # # issue knob is difficult to press # # # impact unable to operate the device properly # # # possible causes * knob is loose * knob is worn * knob is misaligned # # # recommendations * check if the knob is loose and tighten it if necessary * replace the knob if it is worn or damaged * realign the knob if it is misaligned @topic:12


# # # issue * the laparoscopic monitor is not displaying images that are suitable for use in surgical procedures. * the signal from the camera is distorted or otherwise compromised. # # # impact * the surgical team is unable to perform the procedure safely and effectively. * patient outcomes may be compromised. # # # possible causes * faulty camera * damaged cables * faulty monitor * software issues # # # recommendations * inspect the camera and cables for any damage. * replace the camera or cables if necessary. * inspect the monitor for any damage. * reinstall the software if necessary. * contact the manufacturer for further assistance. @topic:2


* * issue : * * bp cuff failure * * impact : * * inaccurate blood pressure measurements, delayed care for patients * * possible causes : * * * cuff damage due to improper use or storage * internal component failure within the cuff * * recommendations : * * * replace damaged cuff with a new one * ensure proper storage and handling of cuffs * inspect cuffs regularly for signs of damage * perform regular maintenance and calibration of the monitor to identify any underlying issues @topic:3


##sy battery door assembly 38 219 [ 6 parts ] part number 12nc description number in diagram 453564408251 453564408251 iv2 - flex mechasy quick mount 39 453564409001 453564409001 iv2 - flex small parts ht 1 453564409011 453564409011 iv2 - flex small parts ht 2 453564409031 453564409031 iv2 - flex cable kit basic 453564415331 453564415331 iv2 - flex cable kit mx400 453564415341 453564415341 iv2 - flex cable ht mx450 / 500 453564415351 453564415351 iv2 - flex cable ht mx550 453564434051 453564434051 iv2 - flex screw kit m4605a 989803135861 battery 10. 8v 6ah lilon 453564239731 453564239731 iv2 mechasy table mount kit iv2 - flex smau parts kit 1 ( 453564409001 ) contents : item quantity no. plastic pin 2 1 tapped - guide 1 2 lightguide power - status 1 3 plastic holder iit 1 4 ingress protection clamp 1 5 plastic holder rack board 1 6 220 6 parts ) iv2 - flex smau parts kit 2 ( 453564409011 ) contents : item quantity no. blank cover recorder 1 1 blank cover quickmount 1 2 blank cover side usb 1 3 blank cover i / o boards 2 4 blank cover advanced system interface board 1 5 221 iv2 - flex cable kit basic ( 453564409031 ) contents : item quantity no. antenna cable 802. 11 wlan 2 1 plastic holder antenna cable 2 @topic:19


in the monitor. 44 1 introduction??? measurement selection key with question marks indicating a label conflict. resolve the conflict automatically, by assigning a new, generic label to the most recently connected conflicting label ( e. g. a second fap label could be changed to abp ) • display the measurement selection window automatically for you to resolve the conflict • • take no action, you must enter the measurement selection window and resolve the all the currently available measurement devices, for example mmss, mms extensions, measurement modules, gas analyzers, devices connected to a host monitor and manually entered measurements • 123 ), are depicted in the measurement selection window. measurement selection x de - activated devices are grayed - out ( here the temp measurement on the left ) abp temp ® icp cvp bis svoz pap tskin tcore ec10 nbp ao co₂ spo2 cgrsp cany measurement labels causing a label conflict are shown in red. if a measurement device is connected but currently unavailable, for example, because it was deactivated due to a label conflict, the device is shown " grayed - out ". when an x2 is connected to a host monitor, the measurement selection window looks like this : me abp temp ap ao nbp o48 spo2 x2 connected to a host monitor warning when an x2 with an active measurement, say spo2, is connected to a host monitor with the same measurement already active, the spo2 measurement on the x2 is deactivated and the meas. deactivated inop is displayed. the measurement can only be reactivated if the x2 is disconnected from the host monitor. the label conflict can be resolved on the host monitor like any other label conflict. 159 5 45 1 introduction to resolve a label conflict : select the measurement selection key or select main setup, then @topic:8


# # # issue - the monitors, physiologic, vital signs device, model suresigns vs4, by philips, is not yet connected to the vsm system. # # # impact - the device cannot be used to provide patient monitoring data. # # # possible causes - the vsm system is not configured correctly. - the device is not properly connected to the vsm system. - the device is not turned on. - the device is not compatible with the vsm system. # # # recommendations - recheck the configuration of the vsm system. - verify that the device is properly connected to the vsm system. - ensure that the device is turned on. - contact the manufacturer of the device to verify compatibility with the vsm system. @topic:22


# # # issue - technician was not provided with enough information to determine the cause of failure. # # # impact - the technician may not be able to properly repair the equipment, which could lead to further problems or even patient injury. # # # possible causes - the person who reported the issue did not provide enough information. - the technician did not ask enough questions to gather more information. # # # recommendations - the person reporting the issue should provide as much information as possible, including the following : - the specific symptoms of the problem - the brand and model of the equipment - the date and time the problem occurred - any recent changes to the equipment or its environment - the technician should ask questions to gather more information, such as : - what were you doing when the problem occurred? - have you ever seen this problem before? - have you made any changes to the equipment or its environment recently? @topic:12


# # # issue - cuff deterioration # # # impact - affects accuracy of vital sign monitoring # # # possible causes - prolonged use - improper storage - physical damage # # # recommendations - regular inspection and maintenance of cuffs - proper storage and handling of cuffs - replacement of cuffs when necessary @topic:5


# # # issue - cuff of the vital signs monitor is damaged and deteriorated. # # # impact - inaccurate or unreliable vital signs measurements. - potential for patient harm if the monitor fails to provide accurate information to healthcare providers. # # # possible causes - normal wear and tear over time. - exposure to excessive moisture or heat. - improper storage or handling. # # # recommendations - replace the damaged cuff with a new one. - regularly inspect and maintain the monitor and its components, including the cuff. - store and handle the monitor and its components according to the manufacturer's instructions. - train staff on the proper use and maintenance of the monitor. @topic:5


* * issue : * * equipment malfunction - electric bed unable to adjust headrest. * * impact : * * impaired patient care due to inability to adjust bed for optimal comfort and positioning. * * possible causes : * * * dislodged bed base support plate obstructing motor operation. * * recommendations : * * * realign and secure the bed base support plate to prevent further obstruction. * test the bed to ensure proper operation and adjustability. * document the repair and notify the appropriate staff of the resolved issue. @topic:1


# # # issue shelf damage # # # impact the damaged shelf may affect the stability of the medical cart and make it unsafe for use. this could lead to injuries to patients or staff. # # # possible causes * overloading the shelf * using the shelf for purposes other than intended * exposure to harsh chemicals or other corrosive substances * poor maintenance # # # recommendations * inspect the shelves regularly for damage or wear and tear. * do not overload the shelves or use them for purposes other than intended. * keep the shelves clean and free of harsh chemicals. * follow the manufacturer's guidelines for maintenance. @topic:31


# # # issue the electric bed is not locking the wheels properly. # # # impact this can be a safety hazard for patients and staff, as the bed could move unexpectedly. # # # possible causes - loose or damaged lock nut - worn or damaged wheel lock mechanism - bent or damaged bed frame # # # recommendations - inspect the bed for any loose or damaged parts. - tighten the lock nut to the specified torque. - replace any worn or damaged parts. - ensure that the bed frame is not bent or damaged. - test the bed to make sure that the wheels are locking properly. @topic:6


# # # issue * degraded bp cuff # # # impact * inaccurate blood pressure measurements * patient discomfort or harm * delayed diagnosis or treatment # # # possible causes * cuff material deterioration due to age or use * hook and loop fastener failure * improper storage or handling * excessive cuff inflation pressure * manufacturer defect # # # recommendations * replace the bp cuff with a new one that meets the manufacturer's specifications. * inspect the hook and loop fastener and replace it if damaged. * ensure proper storage and handling of the equipment. * verify the cuff inflation pressure is within the recommended range. * contact the manufacturer if the issue persists after following these recommendations. @topic:3


# # # issue - the monitor cannot display the ekg waveforms. # # # impact - the clinician cannot monitor the patient's heart rhythm. # # # possible causes - the ekg cable is damaged. - the ekg cable is not properly connected to the monitor. - the ekg cable is not properly connected to the patient. - the monitor is not turned on. - the monitor is not in ekg mode. - the patient's heart rate is too slow or too fast for the monitor to detect. # # # recommendations - inspect the ekg cable for damage. - ensure that the ekg cable is properly connected to the monitor. - ensure that the ekg cable is properly connected to the patient. - turn on the monitor. - put the monitor in ekg mode. - check the patient's heart rate. @topic:18


# # # issue : cuff deterioration # # # impact : failure to obtain accurate blood pressure readings # # # possible causes : - * * arrhythmia * * can cause irregular heartbeats, which can interfere with the cuff's ability to accurately measure blood pressure. - * * other factors * * that may contribute to cuff deterioration include : - prolonged use of the cuff - improper storage of the cuff - exposure to extreme temperatures or moisture # # # recommendations : - replace the cuff with a new one. - inspect the cuff regularly for signs of wear and tear. - store the cuff in a cool, dry place. - avoid exposing the cuff to extreme temperatures or moisture. @topic:3


automatically determines the supramaximal stimulation current and takes the reference twitch ( 100 % ) at this current. if the nmt module is unable to establish a reference twitch, it uses an internal reference value for single - mode measurements. clearing the reference twitch resets the supramaximal stimulation current and reference twitch to default values. to clear the stored references : select clear reference. 1 select the confirm key to reset the stored supramaximal stimulation current and reference twitch. 2 ote this function is not available if a calibration is in progress or no calibration data is stored. aution changing the stimulation current or pulse width also clears stored references. to start nmt measurements, in the setup nmt window select the start key for the mode used, ( for if you need to change the nmt settings after startup, stop the measurements, change the settings, then restart the measurements. to stop nmt measurements, in the setup nmt window select stop, the measurement is immediately interrupted. note to stop nmt stimulation in an emergency,, disconnect the nmt patient cable from the nmt module. during an automatic measurement cycle the yellow start / stop led on the nmt module is lit. if the automatic cycle is stopped, or the measurement mode is set to manual the led is off. the led flashes ( in both modes ) when a stimulation pulse is given. caution to avoid unwanted electrical shocks do not touch the electrodes when the yellow start / stop led note take care when removing the transducer from the patient. do not pull on the cable. 297 changing the nmt measurement settings select stimulation to enter the stimulation window. 1 choose the stimulation mode required : 2 tof ptc witch ( if configureddbs ( if configured ) measurement mode allows the choice between manual and automatic nmt measurements. n the setup nmt windo1 select @topic:17


# # # issue - bent plug # # # impact - inability to connect the bed to power, rendering it inoperable # # # possible causes - the plug may have been stepped on or otherwise damaged # # # recommendations - replace the plug - inspect the power cord for any other damage and replace if necessary - train staff on proper handling of medical equipment to prevent future damage @topic:9


# # # issue - none specified # # # impact - none specified # # # possible causes - alarm volume setting was inadvertently set to 0 - equipment malfunction - user error # # # recommendations - verify that the alarm volume setting is set to an appropriate level - test the equipment to verify that the alarm is functioning properly - contact the manufacturer for further troubleshooting assistance if necessary @topic:26


# # # issue - remote control cable is damaged # # # impact - potential safety hazard to patients and staff due to exposed wires # # # possible causes - cable may have been damaged by : - physical wear and tear - cuts or punctures - water damage - electrical surges # # # recommendations - replace the damaged cable immediately - inspect the cable regularly for any signs of damage - keep the cable away from potential hazards - educate staff on the importance of proper cable handling @topic:7


# # # issue - drawer does not eject # # # impact - inconvenient for medical staff to access medical supplies and equipment stored in the drawer # # # possible causes - solenoid malfunction # # # recommendations - replace solenoid @topic:21


# # # issue - equipment : monitor, patient transport - band : welch allyn - model : propaq lt - description : faulty charging dock # # # impact - inability to charge the device, leading to potential downtime and disruption of patient care. # # # possible causes - physical damage to the charging dock, such as cracks or breakage. - electrical malfunctions within the charging circuitry. - misalignment or improper connection between the device and the charging dock. # # # recommendations - inspect the charging dock for any visible damage or loose connections. - check the electrical connections to ensure they are secure and free of corrosion. - perform a diagnostic test on the charging dock to identify any electrical faults. - replace the charging dock if necessary to ensure proper functionality. - implement a regular maintenance schedule to inspect and clean the charging dock to prevent future failures. @topic:11


# # # issue - the nibp monitor is not charging its battery. # # # impact - the nibp monitor cannot be used wirelessly. - the nibp monitor may not be able to be used at all if the battery dies. # # # possible causes - the adapter is faulty. - the battery is faulty. - the charging port is faulty. # # # recommendations - replace the adapter. - replace the battery. - inspect the charging port for any damage. @topic:11


##mi 60 kg cm tunisia 5kg iec cm turkey kg 50 iec cm turkmenistan lb aami 60 in aami turks and caicos islands 60 kg cm tuvalu aami lb 60 in 507 41 default settings appendix ine frequenccountry - description nits units ecg cable color weight height 0 / 60 [ hzn, clb ganda aami 60 in aami ukraine 60 lb in united arab emirates aami 50 kg cm united kingdo50 kg cm united states lb 60 aami in united states minor outlying 60 lb aami in islands ruguaaami 50 kg cm uzbekistan lb aami 60 in vanuatu 60 lb aami in aami venezuela 60 lb in viet nam iec 50 kg cm aami 50 kg cm lb 60 aami in lb wallis and futuna islands aami 60 in western sahara 50 kg iec cm yemen kg aami 50 cm aami lb zambia 60 in zimbabwe lb aami 60 in alarm and measurement default settings the default wave speed for all measurements except respiration is 25 mm / sec. for respiration the default wave speed is 6. 25 mm / sec. in the following tables, settings may be entered only once per table row if they are the same for all patient categories. larm default settingh30 ( deviations from h10 / h20 / h40 ) alarm settings actory default h10 / larm volume 5 alarms off 2 min pause al. 5min enabled enabled pause al. 10min uto alarms off off alarmoffreminder off visual latching ed onlred & yellow 508 41 default settings appendix h30 ( deviations from h10 / h20 / h40 ) alarm settingfactory default h10 / h20 / h40 audible latching alarm reminder red & yellow off on eminder time 3 min larm sounds raditional redalarminterval 0 seel. al. @topic:8


##ution c. o. interface cables 2. 4 m + 2. 4 m cable m1643a pulsion pressure transducer kits ( pressure line available in various lengths ) contact pulsion for information pressure interface cable for disposable pressure transducer pulsion pmk 206 mainstream c0 2 accessories description part no. c0 2 sensor m2501a adult / pediatric airway adapter ( reusable ) m2513a infant airway adapter ( reusable ) m2516a adult airway adapter ( single - patient use ) m2533a infant airway adapter ( single - patient use ) m2536a sidestream c0 2 accessories description part no. c0 2 sensor m2741a nasal and oral - nasal cannulas c0 2 nasal cannula, adult m2744a c0 2 nasal cannula, pediatric m2745a c0 2 nasal cannula, infant m2746a c0 2 / 0 2 nasal cannula, adult m2750a c0 2 / 0 2 nasal cannula, pediatric m2751a c0 2 / 0 2 nasal cannula, infant 989803144471 co 2 oral - nasal cannula, adult m2756a c0 2 oral - nasal cannula, pediatric m2757a c0 2 / 0 2 oral - nasal cannula, adult m2760a c0 2 / 0 2 oral - nasal cannula, pediatric m2761a airway adapters airway adapter set, et > 4. 0 mm m2768a airway adapter set, et = < 4. 0 mm 989803144531 airway adapter set h, et > 4. 0 mm m2772a 394 mainstream co 2 @topic:8


# # # issue * remote control cable is broken. # # # impact * the bed cannot be adjusted using the remote control. # # # possible causes * the cable may have been damaged during use. * the cable may have been pinched or cut. * the cable may have become frayed over time. # # # recommendations * replace the remote control cable. * inspect the cable for any damage and repair or replace as necessary. * avoid pinching or cutting the cable. * regularly check the cable for fraying and replace it if necessary. * train staff on proper use and handling of medical equipment to prevent damage to cables. @topic:7


# # # issue - blood pressure ( bp ) pump failure # # # impact - inability to accurately measure and monitor patient's blood pressure # # # possible causes - worn or damaged cuff bp # # # recommendations - replaced worn or damaged cuff bp with a new one and tested the device, which is now functioning normally. @topic:3


# # # issue the numeric keypad on a medical cart is malfunctioning, specifically the " 0 " key is not responding. # # # impact the inability to press the " 0 " key on the numeric keypad can hinder the input of numerical data into the medical cart, potentially leading to errors or delays in patient care. # # # possible causes # # # # hardware issues : - faulty numeric keypad or loose connections within the keypad assembly - malfunctioning motherboard or processor that handles keypad input # # # # software issues : - corrupted or outdated firmware or drivers related to the numeric keypad - interference from other software or applications that are running simultaneously # # # # user error : - incorrect settings or configurations related to the numeric keypad - accidental disabling of the numeric keypad functionality # # # recommendations 1. * * hardware troubleshooting : * * - inspect the numeric keypad for any visible damage or loose connections - clean the keypad contacts using a soft brush or compressed air - check the motherboard and processor for any signs of damage or overheating - if necessary, replace the faulty keypad or motherboard 2. * * software troubleshooting : * * - update the firmware or drivers related to the numeric keypad - close any unnecessary software or applications that may be interfering with the keypad functionality - run a system diagnostic test to identify any potential software issues 3. * * user education : * * - ensure that users are properly trained on the operation of the medical cart, including the correct settings for the numeric keypad - provide clear instructions on how to enable and disable the numeric keypad functionality @topic:27


if you silence a yellow arrhythmia alarm and the alarm condition still exists, the visual indicators continue until the condition stops. you will get an alarm reminder every time the configured timeout period has expired. if you silence a yellow arrhythmia alarm and the alarm condition has stopped, the visual indicators are immediately cleared. silencing an alarm does not reset its time out period, so you will not get a realarm for the same condition or lower on the chain until the timeout expires. arrhythmia alarm chaining when arrhythmia analysis is switched on, multiple alarm conditions may be present. announcing all of the detected alarm conditions would be confusing, and might hide a more serious condition. for this reason, arrhythmia alarms are prioritized in three alarm “ chains ” : pvc alarms ; beat detection alarms, and rate alarms. only the highest priority alarm condition in each chain is announced. lower priority alarms in the same chain will not be announced while an alarm is active or during the configured timeout period. if alarm conditions of equal severity from different chains are detected, the alarm condition that occurred most recently is announced. the exception is irregular hr, which only occurs if no other alarms are occurring. see “ ecg and arrhythmia alarm overview ” on page 130 for information on which alarms are included in the different arrhythmia options. see “ arrhythmia alarm timeout periods ” on page 141 for an explanation of how alarm timeouts work. 142 lower priority higher priority arrhythmia alarms 6 ecg, arrhythmia, st and qt monitoring red arrhythmia alarms pvc alarms chain non - sustain vt / vent rhythm lil run pvcs pair f ^ / cs r - on - ^ pvcs - 9 vent bigeminy vent tri @topic:8


# # # issue - the bed's head cannot be adjusted down due to a dislodged backrest. # # # impact - impairs the ability to position the patient in a comfortable or appropriate position for medical care. # # # possible causes - the backrest may have been dislodged due to : - improper handling or maintenance - wear and tear over time - mechanical failure # # # recommendations - inspect the backrest and ensure it is securely fastened. - check for any loose connections, damaged components, or other mechanical issues. - replace or repair any faulty parts as necessary. - implement regular maintenance and inspection schedules to prevent such issues in the future. @topic:1


* * issue : * * inability to touch or interact with the patient monitor's screen * * impact : * * * inability to access patient data, monitor vital signs, or adjust settings * potential delays in patient care and decision - making * * possible causes : * * * * * touchscreen malfunction : * * the touchscreen itself may be damaged or defective, preventing proper registration of touch inputs. * * * control board failure : * * the control board responsible for processing touch inputs and managing screen functionality may have malfunctioned or sustained damage. * * recommendations : * * * * * replace touchscreen : * * if the touchscreen is deemed defective, replacing it with a new one can restore touch functionality. * * * repair or replace control board : * * if the control board is faulty, it should be repaired or replaced to ensure proper signal processing and screen control. * * * update software : * * occasionally, outdated or corrupt software can cause touchscreen issues. updating the device's software to the latest version may resolve the problem. * * * calibrate touchscreen : * * if the touchscreen is misaligned or not responding accurately, calibrating it can reset the touch sensitivity and improve its performance. * * * inspect for physical damage : * * check the monitor for any physical damage that may have compromised the touchscreen or control board, such as cracks or spills. @topic:2


# # # issue - equipment : monitors, physiologic, respiration, respiratory gas, inspired oxygen, bedside / intraoperative - band : teledyne - model : ax300 - cause of failure : unable to measure o2 due to missing sensor at nicu # # # impact - inability to accurately monitor inspired oxygen levels, which is crucial for patient safety, especially in critical care settings like nicu. # # # possible causes - absence of the sensor itself - disconnection of the sensor - faulty sensor # # # recommendations - ensure the sensor is present - check for proper connection of the sensor - if the sensor is present and properly connected, consider replacing the sensor with a new one. @topic:4


range input, no voltage selection required power consumption : ~ 48w cabiing options and conduit size requirements the following table describes the cabling options for the m8031a / b 15 ”, the m8033a / b / c 17 ” and the 865299 19 " tft medical grade touch displays. table 11 digital video cables part number 12nc description m8071 - 61001 453563484551 iv cbl 1. 5m cable kit m8071 - 61002 453563484561 iv cbl 3m cable kit m8071 - 61003 453563484571 iv cbl 10m cable kit. touch cabie part number 12nc description conduit sizes bend radius connector size ( l x w ) m8081 - 61010 451261006551 iv cbl touch, 1. 5m 30mm ^ 25 mm 35 x 16 mm m8081 - 61011 451261006561 iv cbl touch, 3m 30mm ^ 25 mm 35 x 16 mm m8081 - 61012 451261006571 iv cbl touch, 10m 30mm ^ 25 mm 35 x 16 mm m8081 - 61013 451261006581 iv cbl touch, 15m 30mm ^ 25 mm 35 x 16 mm 298 8 site preparation ^ ) part number 12nc description conduit sizes bend radius connector size ( l x w ) m8081 - 61014 451261006591 iv cbl touch, 25m 30mm ^ 25 mm 35 x 16 mm input devices part number 12nc description 453564115901 453564115901 kbd keyboard french trackbau usb 453564115911 453564115911 kbd keyboard italian trackbau usb @topic:19


# # # issue * * * bed cannot be adjusted * * # # # impact * * * patient discomfort * * * the patient may be unable to get into a comfortable position for sleep or medical procedures. * * * staff inconvenience : * * * the staff may have difficulty moving the patient or providing care. * * * potential safety hazard * * * the patient may fall out of bed if it cannot be adjusted to a safe position. # # # possible causes * * * power issue * * * the bed is not plugged in or the power cord is damaged. * * * remote control issue * * * the remote control is not working properly. * * * mechanical issue * * * the bed motor or gears are not working properly. * * * electrical issue : * * * there is a short circuit or other electrical problem in the bed. # # # recommendations * * * check the power cord and outlet * * * make sure the bed is plugged in and the power cord is not damaged. * * * check the remote control. * * * replace the batteries in the remote control. * * * reset the bed. * * * unplug the bed and then plug it back in. * * * inspect the bed for mechanical damage. * * * look for any loose or damaged parts. * * * contact a qualified biomedical engineer. * * * if you are unable to resolve the issue on your own, contact a qualified biomedical engineer to diagnose and repair the bed. @topic:1


* * issue : * * * medical cart is not functioning properly. * * impact : * * * inability to transport medical supplies and equipment efficiently. * delay in patient care. * increased risk of medical errors. * * possible causes : * * * * * battery failure * * : the medical cart may not be charging properly or the battery may have reached the end of its lifespan. * * * electrical malfunction * * : there may be a short circuit or other electrical issue preventing the cart from operating. * * * mechanical failure * * : the cart's wheels, brakes, or other mechanical components may be malfunctioning. * * * software glitch * * : the cart's software may have encountered an error or bug. * * * improper use * * : the cart may not have been used according to the manufacturer's instructions, leading to damage. * * recommendations : * * * * * inspect the battery * * : check the battery level and charge status. if the battery is low or not charging, replace it. * * * troubleshoot electrical connections * * : ensure that all electrical cables and connections are secure and functioning properly. * * * examine mechanical components * * : check for any loose or damaged wheels, brakes, or other mechanical parts. * * * update software * * : check for any available software updates and install them. * * * provide user training * * : educate staff on proper use and maintenance of the medical cart to prevent future issues. * * * contact manufacturer * * : if the issue cannot be resolved by troubleshooting, consult the manufacturer for assistance or technical support. @topic:13


# # # issue biomedical engineer ( bme ) found that the blood pressure ( bp ) cuff of the bedside physiologic monitor is not measuring accurately. # # # impact the inaccurate bp readings could lead to incorrect patient assessment and treatment decisions. # # # possible causes - faulty bp cuff - loose or disconnected connections - problem with the monitor's bp measurement system # # # recommendations - replace the bp cuff with a new one. - check the connections between the bp cuff and the monitor. - if the issue persists, contact the monitor's manufacturer for further troubleshooting and repair. @topic:14


# # # issue - monitor is unable to measure o2 saturation. # # # impact - patient monitoring may be compromised. # # # possible causes - broken spo2 cable. # # # recommendations - replace the spo2 cable. @topic:4


# # # issue battery replacement is required for the philips intellivue mp40 ( m8003a ) patient transport monitors. # # # impact the monitors will not be able to function properly without a functioning battery. this could lead to patient safety concerns if the monitors are not able to provide accurate and reliable vital sign readings. # # # possible causes the batteries in the monitors have reached the end of their lifespan and need to be replaced. # # # recommendations replace the batteries in the monitors with new batteries that meet the specified specifications ( 10. 8v, 6ah, x2 ). @topic:11


# # # issue according to the input, the nibp monitor ( model 740 from casmed ) is not functioning due to a faulty spo2 probe. # # # impact the faulty probe prevents the nibp monitor from accurately measuring spo2 ( peripheral capillary oxygen saturation ). this could delay or impede diagnosis and treatment, potentially impacting patient outcomes. # # # possible causes the cause of the probe failure is not explicitly stated in the input. however, * * some possible causes might include * * : - electrical failure in the probe itself - defective wiring or connection issues between the probe and the monitor - damage to the probe during handling or storage # # # recommendations as an interim solution, the input suggests using a backup spo2 probe from mtd ( lot 6272121 ds - 100a ) with the nibp monitor. however, the following recommendations are suggested for a comprehensive solution : - * * inspect the faulty probe : * * visually check for any physical damage or loose connections. use a multimeter to test electrical continuity. - * * inspect the monitor connection port : * * ensure a secure connection between the probe and the monitor. check for any bent or damaged pins. - * * check the monitor's software : * * update the monitor's software if necessary. reset the monitor to factory settings if the problem persists. - * * contact the manufacturer or service provider : * * if the cause of failure cannot be identified or resolved in - house, reach out to the manufacturer or an authorized service provider. - * * document the repair : * * maintain a record of the issue, troubleshooting steps, and any repairs or replacements made. @topic:4


# # # issue * keypad on the right side of the patient is not working, causing the bed to be unable to be adjusted up and down. # # # impact * the patient cannot be properly positioned in the bed, which can lead to discomfort, pain, and pressure sores. * the bed cannot be used for its intended purpose, which can disrupt patient care and workflow. # # # possible causes * keypad malfunction * loose or damaged wiring * power supply issue * software glitch # # # recommendations * replace the faulty keypad. * inspect and test the wiring and power supply. * update the software if necessary. * provide training to staff on how to properly use and maintain the bed. @topic:25


##hg ( 0. 1 kpa ) resolution wave : 0. 1 mmhg ( 0. 01 kpa ) tability : short term drift erm drift 0. 8 mmhg over foaccuracy specification will be maintained over a 120 hour period to 150 rprange 1 rpaccuracy 2 minutes with co2 transducer attached for full accuracy specification less than 60 ms ( with adult or infant reusable or disposable adapter ) 486 40 specifications m3014a sidestream co2 performance specifications 0 to 150 mmhg ( 0 to 20. 0 kpa ) co2 range accuracy fter 2 minutes warmupfor values between 0 and 40 mmhg : ±2. 0 mmhg ( ±0. 29 kpa ) for values from 41 to 70 mmhg : ±5 % of reading for values from 71 to 100 mmhg : + 8 % of reading for values from 101 to 150 mmhg : ±10 % of reading at respiration rates above 80 rpm, all ranges are ±12 % of actual. the specifications are valid for gas mixtures of co2, balance n2, dry gas at 760 mmhg within specified operating temperature range. numeric : 1. 0 mmhg ( 0. 1 kpa ) resolution wave : 0. 1 mmhg ( 0. 01 kpa ) stability : short term drift accuracy specification will be maintained over a 120 hour period to 150 rprange 1 rpaccuracy 2 minutes with co2 sensor attached for full accuracy specification 50 ±10 ml / minute 3 seconds total system response time 0 to 40°c ( 32 to 100°f ) operating temperature m3014a mainstream and sidestream co2 humidity correction factor either btps or atpd can be selected as the humidity correction factor for the co2 readings. the formula for the correction calculation is : where : p abs atpd @topic:17


unlikely event of a defect on arrival, please keep the packing materials until you have completed the initial inspection. 246 7 installation instructions ] initial inspection mechanical inspection open the shipping container ( s ) and examine each part of the instrument for visible damage, such as broken connectors or controls, or scratches on the equipment surfaces. if the shipping carton / container is undamaged, check the cushioning material and note any signs of severe stress as an indication of rough handling in transit. this may be necessary to support claims for hidden damage that may only become apparent during subsequent testing. electrical inspection the instrument has undergone extensive testing prior to shipment. safety testing at installation is not required ( except in situations where devices are interconnected forming a system, see connecting non - medical devices in the site preparation chapter of this manual ). an extensive self check may be performed. this recommendation does not supersede local requirements. all tests are described in the testing and maintenance section of this manual. claims for damage and repackaging claims for damage when the equipment is received, if physical damage is evident or if the device does not meet the specified operational requirements of the patient safety checks or the extended self check, notify the carrier and the nearest philips sales / support office at once. philips will arrange for immediate repair or replacement of the instrument without waiting for the claim settlement by the carrier. repackaging for shipment or storage if the instrument is to be shipped to a philips sales / support office, securely attach a label showing the name and address of the owner, the instrument model and serial numbers, and the repair required ( or symptoms of the fault ). if available and reusable, the original philips packaging should be used to provide adequate protection during transit. if the original philips packaging is not available or reusable please contact the philips sales / support office who will provide information about adequate packaging materials and methods. installing the mx400 @topic:10


# # # issue - broken remote # # # impact - inability to adjust the bed # # # possible causes - physical damage to the remote - electrical malfunction - software glitch # # # recommendations - replace the remote - inspect the wiring and connections - update the software @topic:7


# # # issue - the monitor is not compatible with the new software program. # # # impact - the monitor cannot be used with the new software program which may affect the patient care. # # # possible causes - the monitor is outdated and does not meet the system requirements of the new software program. - the monitor is not configured properly for use with the new software program. # # # recommendations - check the monitor's system requirements to ensure that it is compatible with the new software program. - if the monitor is compatible, ensure that it is configured properly. - if the monitor is not compatible, contact the vendor to see if there is a firmware update or other solution that will make it compatible. @topic:16


# # # issue - inability to adjust the bed - remote control malfunction # # # impact - patient discomfort and inability to adjust the bed for optimal positioning - delays in patient care due to the need for manual adjustments # # # possible causes - loose electrical connections in the remote control - damage to the remote control's circuitry - malfunctioning remote control receiver in the bed - power supply issues # # # recommendations - * * confirm proper electrical connections : * * ensure that the remote control cable is securely plugged into the bedside unit and that there are no loose connections. - * * inspect the remote control : * * check for any visible damage, such as cracked buttons or a damaged casing. if damage is found, the remote control may need to be replaced. - * * test the remote control receiver : * * use a multimeter to test the voltage output of the receiver when the remote control buttons are pressed. if there is no voltage output, the receiver may be faulty. - * * check the power supply : * * ensure that the bed is receiving power from the mains supply and that the circuit breaker or fuse is not tripped. if the power supply is not consistent, the bed may not function properly. * * additional considerations : * * - * * historical maintenance records : * * review previous maintenance logs for the bed and remote control to identify any recurring issues or patterns. - * * user training : * * provide training to the staff on the proper use and maintenance of the bed and remote control to prevent future malfunctions. - * * regular maintenance : * * conduct regular preventive maintenance on the bed and remote control to identify and address any potential issues before they become major failures. @topic:7


# # # issue - the battery in the physiologic monitor ( bedside ) was replaced. # # # impact - the device was not operational due to a lack of power. # # # possible causes - the battery was not properly installed. - the battery was defective. - the charging system was not functioning properly. # # # recommendations - verify that the battery is properly installed. - replace the battery with a known - good battery. - check the charging system for any damage or loose connections. @topic:11


# # # issue the device, a teledyne ax300 respiratory gas monitor, emitted an alarm. # # # impact the alarm indicated a potential issue with the device's functionality, which could affect patient monitoring and care. # # # possible causes - dirty or depleted battery terminals - battery failure # # # recommendations - clean the battery terminals. - replace the batteries. @topic:11


turn on the pressure meter and zero if necessary. 11. check the accuracy at 0 mmhg. 12. record the reading on the pressure meter and the service tool and compare the results. 13. using the hand bulb, pressurize the nibp system to 50 mmhg ± 5 mmhg and allow 10 seconds for the pressure to stabilize. 14. record the reading on the pressure meter and the service tool and compare the results. 15. using the hand bulb, pressurize the nibp system to 150 mmhg ± 5 mmhg and allow 10 seconds for the pressure to stabilize. 16. record the reading on the pressure meter and the service tool and compare the results. 17. using the hand bulb, pressurize the nibp system to 250 mmhg ± 5 mmhg and allow 10 seconds for the pressure to stabilize. 18. compare the reading on the pressure meter to the service tool reading and record the results. 19. click open valve to open the nibp valves. verify that the pressure on the meter and service tool change to zero. 20. click done to complete the check. 21. to record the results of your test, go to " service record. " perform an nibp accuracy test without the welch allyn service tool the following tools are required to perform this test : csm with swf ver. 1. 33. 00 - x0006 or greater • blood pressure bulb • pressure meter dual lumen pressure tubing. pressure test box service manual functional verification and calibration overview 115 1. in the advanced settings code box, enter your advanced settings password. 2. 3. the nibp pressure verification feature is located in the bottom right of the screen. connect the dual lumen tubing to the device. 4. 5. connect one end of the tubing to the pressure meter and the other end to the @topic:17


# # # issue - nbp unplug # # # impact - the physiologic monitor is unable to measure blood pressure. # # # possible causes - the nbp cable is not properly connected to the monitor. - the nbp cuff is not properly inflated. - the nbp sensor is faulty. # # # recommendations - check that the nbp cable is properly connected to the monitor. - check that the nbp cuff is properly inflated. - replace the nbp sensor if it is faulty. @topic:15


# # # issue - oxygen saturation sensor connected to the forehead is faulty # # # impact - inability to accurately measure oxygen saturation levels # # # possible causes - loose or damaged connection between the sensor and the patient - defective sensor # # # recommendations - inspect the connection between the sensor and the patient to ensure it is secure. - replace the sensor if it is suspected to be defective. @topic:4


# # # issue - signal loss during cautery use # # # impact - inability to monitor patient's vital signs during critical procedures # # # possible causes - faulty grounding system # # # recommendations - ensure a proper grounding system is in place by inspecting the grounding wire and connecting points. - replacing the grounding wire if necessary. @topic:12


# # # issue - cuff unable to be used. # # # impact - unable to obtain vital signs data from patient. # # # possible causes - the cuff may be damaged or defective. - cuff may not be attached properly to the monitor. - cuff may be too small or too large for the patient. # # # recommendations - replace the cuff with a new one. - ensure that the cuff is properly attached to the monitor. - check the size of the cuff and ensure that it is appropriate for the patient. - if the problem persists, contact the manufacturer for assistance. @topic:5


. wait until the conn. status field in the service window shows authenticatd ( forva. c. o monitors ) or connected ( for rel do or higher ). take the monitor approximately 5 m away from the access point. there should be no walls or other obstacles between the monitor and the access point. the following should apply : — observe the rssi ( received signal strength indicator ) value for at least 5 - 10 seconds. the rssi value wiu fluctuate but should stay above 30 in a 5 m distance from the access point used. the wireless link should be active, i. e. the conn. status field should authenticatd ( for rel. c. o monitors ) or connected ( for rel do or higher ), and the other fields should contain values. if the rssi value is significantly lower, check the distance to the access point and the antenna orientation at the monitor. the antenna orientation should be vertical, but the physical placement of the monitor or other equipment within its vicinity as well as walls or other obstacles may influence the antenna orientation required to receive the best rssi value. 5 if this test fails, retry in a different physical area with a different access point and / or check the credential settings in the monitor. test expected test results intellivue 802. 11 bedside adapter performance test rssi value above 30 iit communication test 1 make sure the lan cable is disconnected from the rear of the monitor, then switch on the monitor. 2 go into service mode and, select main setup - > network - > setup iit. in the setup iit menu, set the rf access code in each profile to match your installation. 3 go into service mode. select main setup - > network - > iit diagnostic to access the instrument telemetry diagnostic window. 4 proper installation of the iit module is assured by @topic:10


# # # issue - vital signs monitor intermittently fails to measure blood pressure ( bp ) # # # impact - inaccurate bp readings can lead to delayed or incorrect diagnosis and treatment # # # possible causes - connector leak between the monitor and the bp cuff # # # recommendations - replace the leaking connector - ensure proper connection between the monitor, bp cuff, and patient - test the device to ensure it is functioning properly @topic:14


# # # issue - display does not show on the connected monitor # # # impact - affects the monitoring and display of patient data # # # possible causes - loose or damaged video cable connection - incompatible video output settings on the device - malfunctioning display driver on the connected monitor - internal hardware failure within the device # # # recommendations - ensure that the video cable is securely plugged into both the medical device and the monitor. inspect the cable for any damage or wear. - verify the video output settings on the medical device and adjust them to match the monitor's capabilities. - test the monitor with a different device to isolate the issue. - if the issue persists, contact the manufacturer of the medical device for further troubleshooting and repair. @topic:2


the issue persists after troubleshooting the above steps, contact the manufacturer's technical support team for further assistance. - * * regular maintenance : * * implement a regular maintenance schedule to ensure that the system is synchronized and updated properly. @topic:12


# # # issue - monitor wheels are stuck and not moving smoothly. # # # impact - inability to properly maneuver the monitor within the hospital environment. - potential for accidents or injuries if the monitor is not stable. # # # possible causes - * * lubrication issues : * * lack of lubrication or use of incorrect lubricants can cause the wheels to stick. - * * accumulation of debris : * * dirt, dust, or other particles can accumulate on the wheels and impede their movement. - * * manufacturing defects : * * faulty or poorly designed wheels can lead to friction and sticking. - * * improper handling : * * rough handling or dropping the monitor can damage the wheels or their bearings. # # # recommendations - * * lubricate the wheels : * * apply a suitable lubricant ( e. g., wd - 40 ) to the wheels and bearing surfaces to reduce friction. - * * clean the wheels : * * remove any accumulated debris by wiping with a damp cloth or using a compressed air duster. - * * inspect the wheels : * * examine the wheels for any damage, such as cracks or broken bearings. replace any damaged wheels. - * * handle the monitor with care : * * avoid dropping or mishandling the monitor to prevent damage to the wheels. - * * consider maintenance schedule : * * implement a regular maintenance schedule to check and lubricate the wheels to prevent future issues. @topic:24


# # # issue - equipment not measuring co2 # # # impact - inability to accurately monitor patient's respiratory status # # # possible causes - dirty or malfunctioning co2 sensor - faulty electrical connection - software or firmware issue # # # recommendations - clean or replace the co2 sensor - inspect and repair any loose or damaged electrical connections - check for software or firmware updates and install as needed @topic:4


# # # issue - cuff deterioration # # # impact - inaccurate vital sign readings - patient discomfort - equipment downtime # # # possible causes - prolonged use - physical damage - improper storage - defective materials # # # recommendations - regularly inspect and maintain cuffs according to manufacturer's guidelines - replace damaged or deteriorated cuffs promptly - store cuffs in a clean, dry environment - use high - quality cuffs from reputable manufacturers - train staff on proper cuff handling and storage techniques - consider using disposable cuffs to minimize the risk of contamination and ensure patient safety @topic:5


ten most recently - used screens including up to three modified screens can be accessed. after a patient discharge, the monitor ’ s default screen is shown. modified screens are still available in the change screen menu. if the monitor is switched off and then on again, modified screens are erased from the monitor ’ s memory and cannot be recalled. if a modified screen was the last active screen when the monitor was switched off, it is retained ( unless automat. default is configured to yes ). connecting additional displays to the monitor a second display, showing the same screen as the main display, can be connected to any of the monitors, for viewing only. the mp90 can have two main displays and additionally, as a third main display, the d80 intelligent display connected. all main displays can be configured and operated individually using standard input devices. for mp90 monitors with multiple displays and multiple input devices, the usage and behavior can be configured according to specific requirements at installation ( for example, use for two independent operators or tracking of mouse input across two displays ). for details refer to the service guide. when two operators are using two displays, the scope of an action depends on the type of operation : • patient monitoring operations such as silence or pause alarms take effect for the monitor as a whole, the results will be seen on both displays. • display operations such as the main screen key and back hardkey will take effect only on the display being operated. if you are operating two displays with one remote speedpoint, to navigate from one display to another : 1 move the highlight to the main screen key and then turn one click further. the highlighting moves to a special “ jump ” field at the edge of the screen 2 press the knob on the speedpoint to confirm ; the highlighting will automatically move to the other display. 29 1 basic operation using the xds remote display the content of each screen can be changed individually as @topic:10


# # # issue * * * adjustable head failure * * # # # impact * inability to adjust the patient's head position during examinations or procedures * patient discomfort or pain # # # possible causes * * * motor head failure : * * the motor that controls the head adjustments may have malfunctioned or failed. * * * mechanical failure : * * the mechanical components ( e. g., gears, pulleys, cables ) that enable head movement may have broken or become damaged. * * * electrical failure : * * the electrical system that powers the motor and other components may have short - circuited or burned out. * * * software glitch : * * the software that controls the head adjustments may have encountered a bug or error. # # # recommendations * * * replace the motor head : * * if the motor has failed, it should be replaced with a new or refurbished unit. * * * inspect and repair mechanical components : * * the gears, pulleys, cables, and other mechanical components should be inspected for damage or wear and tear. any damaged parts should be replaced or repaired. * * * check electrical system : * * the electrical wiring, circuit boards, and other electrical components should be inspected for signs of burning, short - circuiting, or damage. any faulty components should be replaced. * * * update or patch software : * * if the failure is suspected to be software - related, the software should be updated or patched to the latest version. * * * perform regular maintenance and calibration : * * to prevent future failures, regular maintenance and calibration of all components, including the motor, mechanical system, and electrical system, is essential. @topic:1