# # # issue - the bed is rising on its own without any input from the control panel. # # # impact - patient safety hazard : the bed moving unexpectedly could cause the patient to fall or be injured. - disruption of patient care : the bed not functioning properly could delay or prevent patient care procedures. # # # possible causes - malfunctioning control panel : the control panel may be sending incorrect signals to the bed, causing it to rise on its own. - faulty wiring : there may be a problem with the wiring between the control panel and the bed, causing intermittent or incorrect signals to be sent. - mechanical issue : there may be a mechanical problem with the bed, such as a jammed motor or a faulty sensor, causing it to move unexpectedly. - software glitch : if the bed is controlled by a computer, there may be a software glitch that is causing the bed to move on its own. # # # recommendations - * * inspect the control panel : * * check the control panel for any visible damage or loose connections. - * * test the wiring : * * use a multimeter to check the continuity of the wiring between the control panel and the bed. - * * examine the mechanical components : * * check the bed for any jammed motors, loose screws, or other mechanical issues. - * * update the software : * * if the bed is controlled by a computer, check for any available software updates that may resolve the issue. - * * contact the manufacturer : * * if the issue cannot be resolved by the technician, contact the manufacturer for further assistance. @topic:1


# # # issue - wood bed rails are not locking # # # impact - patient safety hazard - inconvenience to staff # # # possible causes - broken or missing latch - misaligned bed rails - faulty wiring - software glitch # # # recommendations - inspect the latch for damage or breakage. replace if necessary. - ensure that the bed rails are properly aligned and secured. - check the wiring for any loose or damaged connections. - contact the manufacturer for software updates if necessary. - replace the equipment if all else fails. @topic:6


* * issue : * * values measured by the device are low. * * impact : * * compromised accuracy of vital signs monitoring, potentially leading to delayed or incorrect diagnosis and treatment. * * possible causes : * * * * * faulty sensor or transducer : * * the sensor responsible for measuring vital signs may be malfunctioning, leading to inaccurate readings. * * * calibration issues : * * the device may not be properly calibrated to provide accurate measurements. * * * electrical interference : * * electromagnetic interference from other devices in the vicinity can disrupt the device's functioning. * * * loose or damaged connections : * * loose or damaged wires or connectors can disrupt the transmission of signals from the sensor to the device. * * * user error : * * the device may not be used correctly, leading to incorrect readings. * * recommendations : * * * * * inspect the sensor : * * check the sensor for any physical damage or defects. * * * calibrate the device : * * follow the manufacturer's instructions to properly calibrate the device. * * * validate the device : * * conduct a validation test with known calibration standards to ensure accurate measurements. * * * check for electrical interference : * * identify and eliminate sources of electromagnetic interference in the vicinity of the device. * * * inspect connections : * * ensure all wires and connectors are securely connected and in good condition. * * * train users : * * provide adequate training to users on the proper operation of the device. @topic:20


# # # issue - difficulty adjusting the foot section of the ob / gyn bed # # # impact - delays or interruptions in patient care - discomfort or injury to patients # # # possible causes - worn or damaged foot section adjustment mechanism - misalignment of foot section components - obstruction or binding of foot section components # # # recommendations - inspect the foot section adjustment mechanism for signs of wear or damage. - ensure proper alignment of foot section components. - lubricate or adjust foot section components as necessary to reduce friction and binding. @topic:1


# # # issue : missing connector # # # impact : device malfunction # # # possible causes : - the connector may have become detached during transportation or handling. - the connector may have been damaged due to excessive force or improper use. - the connector may have corroded or rusted over time. # # # recommendations : - ensure that the device is handled and transported carefully to prevent connectors from becoming detached. - inspect connectors regularly for signs of damage or corrosion. - replace damaged or corroded connectors promptly. @topic:2


# # # issue - stand of the monitor is falling. # # # impact - monitor may not be positioned correctly, which can lead to patient discomfort or inaccurate readings. # # # possible causes - loose locking mechanism on the stand. - wear and tear on the stand components. - damage to the stand due to mishandling or excessive force. - improper assembly of the stand. # # # recommendations - tighten the locking mechanism on the stand. - inspect the stand components for any signs of wear or damage. - if any damage is found, replace the affected components. - ensure that the stand is assembled correctly according to the manufacturer's instructions. - handle the stand with care to avoid damage. @topic:12


# # # issue the bed is not adjustable. # # # impact the patient cannot be positioned properly for treatment or care. # # # possible causes - the electrical system is not functioning properly. - the mechanical system is not functioning properly. - the control panel is not functioning properly. - the battery is not functioning properly. # # # recommendations - check the electrical system for any loose connections or damaged wires. - check the mechanical system for any broken or jammed parts. - check the control panel for any broken buttons or switches. - check the battery for any leaks or damage. - if the problem cannot be identified or fixed, contact the manufacturer for assistance. @topic:1


# # # issue - monitor is not displaying on the lcd screen. # # # impact - the patient's vital signs cannot be monitored. # # # possible causes - hdmi splitter is not turned on. - hdmi cable is not properly connected. - monitor is not receiving power. - monitor is not set to the correct input. # # # recommendations - turn on the hdmi splitter. - check the hdmi cable connections. - ensure that the monitor is receiving power. - set the monitor to the correct input. @topic:2


# # # issue - monitor, physiologic, vital signs is not functioning due to a broken power cord and plug. # # # impact - inability to monitor vital signs, which can delay or compromise patient care. # # # possible causes - * * broken power cord : * * the power cord may have been damaged due to excessive bending, twisting, or pulling. - * * broken plug : * * the plug may have been damaged due to improper insertion or removal from the power outlet. # # # recommendations - * * replace the power cord : * * obtain a new power cord compatible with the monitor. replace the damaged power cord with the new one. - * * inspect the plug : * * examine the plug for any damage. if the plug is damaged, it may need to be replaced. - * * secure the cables : * * ensure that the power cord and cables are properly secured to prevent them from being accidentally disconnected or damaged. - * * avoid bending or twisting the cords : * * handle the power cord and cables with care to avoid kinking or damaging them. @topic:2


# # # issue - n / a # # # impact - medical cart cannot be used as intended. # # # possible causes - n / a # # # recommendations - perform regular maintenance checks on the medical cart to ensure that all components are secure and functioning properly. @topic:13


# # # issue - ekg interference # # # impact - inaccurate ekg readings - difficulty in diagnosing and monitoring patient's condition # # # possible causes - faulty or damaged reddot sensor - loose or disconnected connections between the sensor and the monitor - electrical interference from other nearby equipment - improper grounding of the patient or the monitor # # # recommendations - replace the reddot sensor - check and tighten all connections between the sensor and the monitor - identify and eliminate sources of electrical interference - ensure proper grounding of the patient and the monitor @topic:18


# # # issue - bed is too small and slippery, which poses a safety hazard to elderly patients who may fall off. - bed is too small for larger patients, making it difficult to examine them. # # # impact - patient safety is compromised. - patient satisfaction is decreased. # # # possible causes - incorrect bed size or model was ordered. - bed was not assembled or adjusted properly. - bed is old and worn out. # # # recommendations - order the correct bed size and model for the intended use. - assemble and adjust the bed according to the manufacturer's instructions. - inspect the bed regularly and replace any worn or damaged parts. - consider adding a non - slip surface to the bed to prevent patients from sliding off. - consider purchasing a larger bed for larger patients. @topic:9


# # # issue - loose connector # # # impact - device malfunction or failure # # # possible causes - wear and tear - improper connection - damage to the connector # # # recommendations - regular inspection of equipment connections - proper training on equipment usage - use of appropriate tools for connection and disconnection - keep the equipment in a proper environment @topic:12


# # # issue - battery failure in vital signs monitor # # # impact - device may not be able to provide accurate readings - device may not be able to operate - patient safety may be compromised # # # possible causes - battery is old and worn out - battery is not properly charged - battery is faulty # # # recommendations - replace the battery with a new one - ensure that the battery is properly charged - contact the manufacturer for further assistance @topic:11


# # # issue - touch screen is not working # # # impact - inability to use the monitor for patient monitoring # # # possible causes - damaged touch screen - software glitch - loose connection # # # recommendations - replace the touch screen - update the software - check the connections @topic:2


# # # issue - drawer does not open smoothly. # # # impact - difficulty accessing medical supplies. - potential delay in providing patient care. # # # possible causes - friction in the drawer track. - wear and tear of the drawer mechanism. - foreign object obstructing the drawer's movement. # # # recommendations - * * lubricate the drawer track : * * apply a small amount of lubricant, such as wd - 40, to the drawer track. - * * inspect the drawer mechanism : * * check for any loose or damaged components that may be hindering the drawer's movement. - * * remove any obstructions : * * remove any foreign objects or debris that may be blocking the drawer's path. - * * replace worn components : * * if the drawer mechanism is worn or damaged, consider replacing the affected components. - * * calibrate the drawer : * * ensure that the drawer is properly aligned and calibrated to prevent friction and smooth operation. @topic:21


# # # issue defective telemetry electrocardiography monitor # # # impact inability to monitor patient's electrocardiogram remotely # # # possible causes - malfunctioning telemetry transmitter or receiver - incorrect frequency setting - interference from other medical devices # # # recommendations - verify that the telemetry transmitter and receiver are functioning properly - confirm that the frequency setting is correct - identify and eliminate any sources of interference @topic:16


# # # issue - the monitor, central station from hewlett packard ( model m3150a ) is experiencing freezing issues. # # # impact - the freezing issue prevents the device from functioning properly, potentially impacting patient care and monitoring. # # # possible causes - * * software issues : * * outdated or corrupted software can cause the device to freeze. - * * hardware malfunction : * * faulty components, such as memory or storage devices, can also lead to freezing. - * * power supply problems : * * intermittent power supply or voltage fluctuations can cause the device to reset or freeze. - * * network connectivity issues : * * poor network connectivity can disrupt communication between the device and other components, causing freezing. - * * environmental factors : * * extreme temperatures, humidity, or dust can affect the device's performance and cause freezing. # # # recommendations - * * update software : * * ensure that the device is running the latest software version to address potential bugs or security issues. - * * inspect hardware : * * physically inspect the device for any loose connections, damaged components, or signs of overheating. - * * check power supply : * * verify that the device is receiving a stable power supply and that the voltage is within acceptable limits. - * * optimize network connectivity : * * ensure that the device has a strong and stable network connection. if necessary, troubleshoot any network issues or consider upgrading the network infrastructure. - * * control environmental factors : * * maintain the device in an environment with appropriate temperature and humidity levels. minimize exposure to dust or other contaminants. @topic:2


# # # issue - equipment : telemetry ekg module - description : the patient's ekg signal is not being transmitted properly. # # # impact - the patient's ekg signal is essential for monitoring their heart rate and rhythm. if the signal is not transmitted properly, the doctor may not be able to make an accurate diagnosis or provide appropriate treatment. # # # possible causes - the ekg cable may be loose or damaged. - the ekg module may be malfunctioning. - there may be interference from other electrical devices in the area. # # # recommendations - check the ekg cable for any loose connections or damage. - if the ekg cable is not the issue, replace the ekg module. - if the issue persists, move the patient to a different location away from any potential sources of interference. @topic:18


* * issue : * * test monitor is unable to read values from module bis. * * impact : * * the patient's condition cannot be accurately monitored, which could lead to serious health risks. * * possible causes : * * * defective test monitor * defective module bis * incompatible software or firmware between the test monitor and module bis * loose or damaged cables * incorrect settings on the test monitor or module bis * * recommendations : * * * check the cables connecting the test monitor and module bis to ensure they are securely connected and undamaged. * verify that the test monitor and module bis are using compatible software and firmware versions. * inspect the test monitor and module bis for any physical damage or defects. * reset both the test monitor and module bis to their default settings. * if the issue persists, contact the manufacturer of the test monitor or module bis for further assistance. @topic:22


manufacturer. these updates may address any software bugs or improve spo2 measurement accuracy. - * * contact technical support : * * if the issue persists after troubleshooting, contact the manufacturer's technical support team for further assistance. @topic:4


# # # issue : display screen failure # # # impact : unable to use the medical cart for patient care # # # possible causes - * * battery power issue : * * - the battery may have become completely discharged, preventing the device from powering on. - * * power supply failure : * * - the power supply to the device may have been interrupted due to a power outage or loose connection. - * * display module malfunction : * * - the display module responsible for displaying information on the screen may have malfunctioned or become disconnected. - * * software glitch : * * - a software glitch or bug may have caused the display screen to stop working. # # # recommendations - * * check battery : * * - ensure that the battery is fully charged or replaced with a new one. - * * inspect power supply : * * - check the power supply cable and connections for any damage or loose connections. - * * reset device : * * - try resetting the device by unplugging it from the power source and plugging it back in after a few minutes. - * * contact manufacturer : * * - if the issue persists after trying the above steps, contact the manufacturer for further technical support and repair options. @topic:13


# # # issue - the bed electric equipment, manufactured by huntleigh and model mb - acc73, is experiencing a failure due to a faulty control box. # # # impact - the equipment may not function properly, leading to potential disruptions in patient care and workflow. # # # possible causes - * * internal component failure : * * the control box may have experienced a failure due to worn - out or damaged components, such as capacitors, resistors, or transistors. - * * power surge : * * a sudden surge in voltage or power supply fluctuations can damage sensitive electronic components within the control box. - * * environmental factors : * * excessive heat, humidity, or vibration can also contribute to the failure of electronic components. # # # recommendations - * * replace the control box : * * the primary recommendation is to replace the faulty control box with a new one from the manufacturer. - * * test functionality : * * after replacing the control box, thoroughly test the equipment's functionality to ensure proper operation. - * * regular maintenance : * * implement regular maintenance schedules to inspect and service the equipment, including checking the control box for any potential issues. - * * staff training : * * provide training to staff on proper usage and handling of the equipment to minimize the risk of future failures due to misuse or mishandling. - * * monitor performance : * * regularly monitor the equipment's performance and keep a record of any issues or malfunctions. this information can help identify patterns and trends that may indicate underlying problems or the need for proactive maintenance. @topic:9


# # # issue - the medical cart is not connecting to wifi. # # # impact - the cart cannot be used to access patient records or other hospital systems. # # # possible causes - the cart's wifi adapter is not enabled. - the cart is not within range of a wifi network. - the wifi network is not functioning properly. - the cart's software is not up to date. # # # recommendations - check to make sure that the cart's wifi adapter is enabled. - move the cart to a location that is within range of a wifi network. - contact the hospital's it department to verify that the wifi network is functioning properly. - update the cart's software to the latest version. @topic:13


# # # issue the electric bed is not able to adjust. # # # impact the patient may not be able to get into or out of bed, or may be uncomfortable in the bed. # # # possible causes - the bed's motor may be malfunctioning. - the bed's power supply may be interrupted. - the bed's controls may be damaged. - the bed's frame may be damaged. # # # recommendations - check the bed's motor for any signs of damage or malfunction. - check the bed's power supply to make sure that it is working properly. - check the bed's controls for any signs of damage. - check the bed's frame for any signs of damage. - if any of the above components are damaged, they should be replaced. - if the problem persists, contact the manufacturer of the bed for further assistance. @topic:1


# # # issue - the basket holding the monitor fell off the stand. # # # impact - the monitor could have fallen and caused injury to a patient or staff member. - the monitor could have been damaged, resulting in loss of data or patient information. # # # possible causes - the locking screws on the basket were not tightened properly. - the basket was not properly attached to the stand. - the stand was not stable or secure. # # # recommendations - ensure that the locking screws on the basket are tightened properly. - verify that the basket is properly attached to the stand. - check that the stand is stable and secure. - regularly inspect the monitor and stand for any signs of damage or wear and tear. @topic:12


# # # issue - bed rail detachment # # # impact - patient safety hazard - potential for falls and injuries # # # possible causes - loose or damaged fasteners - improper installation - wear and tear - excessive force applied to the rail # # # recommendations - inspect the bed rail and fasteners regularly for any signs of damage or wear - ensure that the rail is installed correctly according to the manufacturer's instructions - do not apply excessive force to the rail - replace the rail if it becomes damaged or loose @topic:6


no. to re - enable touchscreen functionality change the touch enable selection to yes. printer test report to verify your printer configuration you may want to print a test report. to print a test report select main setup - > reports - > setup printers - > print test rep. your test report should look like this : | g philips general hospflal courier ( non - proportional ) abcd ^ ghijiamiopqbstuvwxyzabcde l £ghi3klinnopqrstuvwxyz0123456789 abcdefghijklmnopqrstuvwxyzabcdefghij klmnopqrstuvwxyz0123456789 abcdefghijklmnopqrstuvwxyzabcdefqhij klranopqrstuvwx'abcdefghijklmnopqrstuvwxyzabcdefghij'univers ( proportional ) abcdefghijklmnopqrstuvwxyzabcdefghijklmnopqrstuvwxyzoi 23456789 abcdefghijklmnopqrstuvwxyzabcdefghijklmnopqrstuvwxyz0123456785 abcdefghijklmnopqrstuvwxyzabcdefghijklmnopc arial ( proportional ) l l i, ii'l l l il.! inrx > pcjrsuivwxyz0123456709 abcdefghi jklmnopqrstuvwxyzabcdefghijklmnopqretuvwxyzoi 23456789 abcdefghi @topic:2


# # # issue - remote control line failure # # # impact - the bed cannot be adjusted via the remote control. # # # possible causes - damaged or broken remote control line. - poor connection between the remote control line and the bed. - electrical fault within the remote control or the bed. # # # recommendations - inspect the remote control line for any visible damage or breaks. - ensure that the remote control line is properly connected to both the remote control and the bed. - test the remote control and the bed to ensure proper functionality. - if the issue persists, contact the manufacturer for further troubleshooting and repair. @topic:7


ps power supply model : fw8031m / dt / 15. 82 disassembly and repair welch [UNK] connex® spot monitor note see the model number and label for further information. kit item : 7000 - ps, connex spot power supply. 1. holding the power supply unit snugly to the support bracket, attach the power supply to the bracket using the 2 phillips screws. thread the power cable from the power supply through one of the spaces next to the pole to reach the device, and then connect the power cable to the device. 2. 3. connect the power cord to the mains outlet. power up the device. 4. section c, remove the power supply from the mobile stand this section applies to csm devices with a 4400 - mbs or 7000 - ms3 mobile stand meeting these criteria : 4400 - mbs or 7000 - ms3 mobile stand manufactured before 02 / 2022 * or earlier. 7000 - ps power supply manufactured after 02 / 2022 * or later power supply model : fw8031m / dt / 15. service manual note see the model number and label for further information. power down the device. 1. disconnect the power cable from the device and disconnect the power cord from the wall. 2. with one hand holding the power supply, remove the 2 phillips screws that attach the power supply to the bracket. with the other hand, hold the adapter plate and the well nuts. 3. note set the screws, adapter plate, and well nuts aside for eassembly. remove the power supply unit. 4. recycle or dispose of the power supply unit in an environmentally safe manner consistent with local regulations. section c, attach the power supply to the stand this section applies to csm devices with a 4400 - mbs or 700 @topic:19


* * issue : * * bed is not lowering at the head. * * impact : * * patient discomfort, inconvenience, and potential risk of injury. * * possible causes : * * - * * electrical : * * faulty electrical connections, damaged wiring, or malfunctioning motor. - * * mechanical : * * obstructed gears or tracks, loose or worn components, broken springs or cables. - * * control system : * * malfunctioning control panel, faulty sensors, or software errors. - * * operator error : * * incorrect use or maintenance, exceeding weight capacity. - * * other : * * wear and tear, inadequate lubrication, environmental factors ( e. g., dust, humidity ). * * recommendations : * * * * electrical : * * - check electrical connections and wiring for damage or loose contacts. - inspect motor for any signs of burning or damage. - test electrical continuity and resistance to identify potential faults. * * mechanical : * * - inspect gears and tracks for obstruction or wear. - check components for looseness or damage. - replace any broken springs or cables. - lubricate moving parts as necessary. * * control system : * * - examine control panel buttons, switches, and sensors for damage or malfunction. - check software updates for any known issues. - reset or reconfigure the control system if necessary. * * operator error : * * - review proper operating procedures with staff. - ensure weight capacity is not exceeded. * * other : * * - clean and inspect the bed regularly for any signs of wear or damage. - lubricate moving parts as recommended by the manufacturer. - check for any environmental factors that may be contributing to the issue. @topic:1


# # # issue the medical bed cannot be raised at the head. # # # impact this can make it difficult for patients to get in and out of bed, and can also be a safety hazard. # # # possible causes - the remote control is not working properly. - the motor that raises the head of the bed is not working properly. - the gears that raise the head of the bed are not working properly. - the wiring that connects the motor to the remote control is damaged. # # # recommendations - check the remote control to make sure that it is working properly. - check the motor that raises the head of the bed to make sure that it is working properly. - check the gears that raise the head of the bed to make sure that they are working properly. - check the wiring that connects the motor to the remote control to make sure that it is not damaged. @topic:1


# # # issue - the cause of failure is : the module slot is loose. # # # impact - the module may not be properly seated in the slot, which could lead to a loss of data or power to the device. # # # possible causes - the module may not have been properly inserted into the slot. - the module slot may be damaged or misaligned. - the screws that secure the module in the slot may be loose. # # # recommendations - ensure that the module is properly inserted into the slot. - inspect the module slot for any damage or misalignment. - tighten the screws that secure the module in the slot. @topic:12


# # # issue - medical cart ( bdms 01 ) cannot connect to wi - fi, resulting in no internet connection. # # # impact - inability to access necessary medical information and software applications on the medical cart. # # # possible causes - wi - fi connection settings not properly configured - network issues or wi - fi signal interference - software or firmware issues with the medical cart # # # recommendations - verify wi - fi connection settings and adjust them if necessary. - troubleshoot network issues by checking for any interruptions or signal strength issues. - update software or firmware on the medical cart if available. - contact the vendor or manufacturer for further assistance if the issue persists. @topic:13


malaysia 50 k g cm iec maldives 60 ib in aami mali 50 k g cm iec malta 50 k g cm iec marshall islands 60 lb in aami martinique 60 k g cm iec mauritania 50 k g cm iec mauritius 60 ib in aami mayotte 60 ib in aami mexico 60 k g cm aami micronesia, federal states of 60 lb in aami moldova, republic of 60 lb in aami monaco 60 lb in aami mongolia 60 ib in aami montenegro 50 k g cm iec montserrat 50 k g cm aami morocco 50 k g cm iec mozambique 50 k g cm iec myanmar 60 lb in aami namibia 50 k g cm iec nauru 60 lb in aami nepal 60 ib in aami netherlands 50 k g cm iec netherlands antilles 50 k g cm aami new caledonia 60 ib in aami new zealand 50 k g cm aami nicaragua 60 k g in aami niger 50 k g cm iec nigeria 50 k g cm iec niue 60 ib in aami norfolk islands 60 ib in aami northern mariana islands 60 ib in aami norway 50 k g cm iec oman 50 k g cm aami pakistan 50 k g cm iec palau 60 lb in aami palestinian territory 50 k g cm aami panama 60 ib in aami papua new guinea 60 ib in aami paraguay 50 k g cm aami peru 60 k g cm aami 456 country - specific default settings 39 default settings appendix philippines 60 k g cm aami pitcairn 60 lb in aami poland 50 k g cm iec portugal 50 k g cm iec puerto rico 60 lb in aami qatar 50 k g cm aami reunion 60 ib in aami romania 50 k g cm iec russian federation 50 k g cm iec rwanda 50 k g cm iec saint helena 60 lb in aami saint kitts and nevis 60 k g cm aami saint lucia 50 k g cm @topic:19


# # # issue - broken brown ekg lead wire # # # impact - patient monitoring may be inaccurate or incomplete. - the monitor may not be able to display or record ekg data. - the monitor may generate false alarms. # # # possible causes - the lead wire may have been damaged during use or storage. - the lead wire may have been improperly connected to the patient or the monitor. - the lead wire may have been exposed to excessive heat or moisture. # # # recommendations - inspect the lead wire for damage. - replace the lead wire if it is damaged. - ensure that the lead wire is properly connected to the patient and the monitor. - store the lead wire in a cool, dry place. - train staff on the proper use and care of lead wires. @topic:18


# # # issue - cuff is damaged. # # # impact - patients may not receive accurate blood pressure readings. # # # possible causes - cuff may have been over - inflated or damaged due to improper use. - cuff may be old or worn out. # # # recommendations - replace the cuff with a new one. - inspect the cuff for any damage before each use. - train staff on proper use of the blood pressure monitor. @topic:3


# # # issue the test monitor is unable to read the values from the module bis. # # # impact the technician is unable to determine if the module bis is functioning properly. # # # possible causes - the test monitor is not properly calibrated. - the module bis is not properly connected to the test monitor. - the module bis is defective. # # # recommendations - calibrate the test monitor. - check the connections between the module bis and the test monitor. - replace the module bis if necessary. @topic:22


# # # issue - cuff bp is broken and cannot be repaired. # # # impact - patient's blood pressure cannot be accurately measured. # # # possible causes - cuff is old and worn out. - cuff was not properly inflated. - cuff was damaged during use. # # # recommendations - replace the cuff with a new one. - ensure that the cuff is properly inflated before use. - handle the cuff with care to avoid damage. @topic:3


a system test is required to ensure that the resulting equipment leakage current does not exceed the limits of en / iec 60601 - 1 - 1. for detailed information see the testing and maintenance chapter. compatible printers printers connected to the usb port of the monitor need to have native pcl5 capability or higher. the following printers have native pcl5 capability or higher and work when plugged into the usb port on intellivue patient monitors : • hp laserjet pro 400 m401 • hp laserjet pro 400 m451dn ( color ) • hp laserjet enterprise p3015dn • hp laserjet p2050 series models ( p2055d, p2055dn and p2055x ) if your selected printer has a comparable specification to the printers listed above, it is very likely that the printer will work with the intellivue patient monitor when connected to the usb port of the monitor. 262 7 installation instructions ] note • some printers ship with the " hp smart install " feature enabled. in these cases, the printer will not be identified by the monitor and the " hp smart install " feature needs to be disabled. for further details on how to disable the " hp smart install " feature, please refer to the documentation which ships with the printer. after disabling the " hp smart install " feature, the printer needs to be power cycled in order to make the printer work with the monitor. • in rare cases, a specific native pcl5 or higher printer may not be supported. this is typically caused by different printer firmware implementations of the printer manufacturers. we recommend testing the functionality of any printer before installation. • printing a report on both sides of a page is not supported. configuring the usb interface for use with a programmabie barcode scanner note a programmable barcode scanner must be pre - configured to @topic:28


# # # issue - the issue is that the electric bed is not functioning properly. # # # impact - the impact is that the bed cannot be used to care for patients. # # # possible causes - the possible causes of the failure are : - the bed is not plugged in. - the bed is not turned on. - the bed is damaged. - the bed is not calibrated. # # # recommendations - the recommendations for fixing the bed are : - check that the bed is plugged in. - check that the bed is turned on. - inspect the bed for damage. - calibrate the bed. @topic:9


# # # issue - the electrical bed is unable to adjust its height. # # # impact - the patient is unable to get in and out of the bed safely and comfortably. - the patient is unable to be positioned properly for medical procedures. - the medical staff is unable to work efficiently. # # # possible causes - * * electrical : * * - loose connections in the electrical wiring - faulty motor - faulty control panel - * * mechanical : * * - damaged gears or bearings - obstruction in the bed frame - * * user error : * * - incorrect operation of the bed controls # # # recommendations - * * electrical : * * - check for loose connections in the electrical wiring. - test the motor and control panel. - replace any faulty components. - * * mechanical : * * - inspect the gears and bearings for damage. - remove any obstructions from the bed frame. - * * user error : * * - train staff on the correct operation of the bed controls. - post instructions for patients on how to operate the bed. @topic:1


# # # issue - * * vital signs monitor not able to measure blood pressure ( bp ) * * # # # impact - * * delayed or inaccurate patient care * * - * * potential safety hazard if bp is not closely monitored * * # # # possible causes - * * loose bp cuff or tubing * * - * * defective bp sensor * * - * * malfunctioning monitor or software * * - * * operator error * * # # # recommendations - * * inspect the bp cuff and tubing for any loose connections or damage. * * - * * check the bp sensor for any visible damage or contamination. * * - * * ensure that the monitor and software are up to date and functioning properly. * * - * * review the operator's manual for proper use of the device. * * - * * consult with the manufacturer or a qualified biomedical engineer for further troubleshooting and repairs. * * @topic:15


performance test this test checks the performance of the ecg and respiration measurements. tools required : patient simulator. ecg performance 1. connect the patient simulator to the ecg / resp connector. 2. configure the patient simulator as follows : - ecg sinus rhythm. - hr = 100 bpm or 120 bpm ( depending on your patient simulator ). 3. check the displayed ecg wave and hr value against the simulator configuration. 4. the value should be 100bpm or 120 bpm + / - 2 bpm. respiration performance 1. change the patient simulator configuration to : - base impedance line 1500 ohm. - delta impedance 0. 5 ohm. - respiration rate 40 rpm or 45 rpm. 2. the value should be 40 rpm + / - 2 rpm or 45 rpm + / - 2 rpm. ecg performance test 100bpm + / - 2bpm or 120bpm + / - 2bpm respiration performance test 40 rpm + / - 2 rpm or 45 rpm + / - 2 rpm 77 ecg sync performance test this test checks the performance of ecg synchronization between the monitor and a defibrillator. it only needs to be performed when this feature is in use as a protocol at the customer site. tools required : - defibrillator with ecg sync and marker output. - patient simulator. 1. connect the patient simulator to the ecg connector and the defibrillator to the ecg sync output on the monitor. 2. set the patient simulator to the following configuration : - hr = 100 bpm or 120 bpm ( depending on your patient simulator ). - ecg sinus rhythm. 3. switch the defibrillator to simulation mode. 4. check that the marker pulse is displayed before the t - wave begins. expected test results ecg sync performance @topic:8


# # # issue - drawer is off track # # # impact - inability to access drawer contents, affecting patient care or workflow # # # possible causes - mechanical failure of drawer guides - damage to drawer or cabinet frame - improper installation or handling # # # recommendations - inspect drawer guides for damage or wear. tighten or replace as needed. - check drawer and cabinet frame for any signs of damage or misalignment. repair or replace damaged components. - ensure proper installation and handling of drawer, following manufacturer's instructions. - consider regular maintenance and inspections to prevent future drawer failures. @topic:21


# # # issue electric bed is not adjustable. # # # impact patients cannot be positioned or moved safely. # # # possible causes - motor malfunction - power supply issue - control panel failure - damaged cables or connections # # # recommendations - check the power supply and connections. - inspect the motor for signs of wear or damage. - test the control panel to ensure proper functionality. - replace any damaged cables or connections. - if the issue persists, escalate to a qualified biomedical engineer for further troubleshooting and repair. @topic:1


* * issue : * * unable to proceed with the repair due to missing component ( pressure sensor cuff ). * * impact : * * delay in repair of vital signs monitor, potentially affecting patient care. * * possible causes : * * - the pressure sensor cuff was not included in the initial inventory of spare parts for the repair. - the pressure sensor cuff was misplaced or lost during the repair process. * * recommendations : * * - check inventory for the missing pressure sensor cuff and replenish if necessary. - search thoroughly for the pressure sensor cuff within the repair area. - contact the manufacturer or supplier to inquire about the availability of the pressure sensor cuff. - if the pressure sensor cuff cannot be located or obtained promptly, consider postponing the repair until a replacement can be procured. @topic:5


# # # issue * * defect in detecting acceleration cord * * # # # impact * * inability to use the equipment for measuring neurological activity related to acceleration. * * # # # possible causes * * possible causes of the failure could be : * * * * * loose connection or damage to the acceleration sensor or its wiring. * * * * * malfunction of the electronic circuitry responsible for signal processing and detection. * * * * * software bug or glitch in the device's firmware. * * * * * manufacturing defect in the device. * * # # # recommendations * * to address the issue, the following recommendations can be explored : * * * * * inspect and test the acceleration sensor, its wiring, and the electronic circuitry. * * * * * update the device's firmware to the latest version, if available. * * * * * consult the manufacturer for further troubleshooting guidance or spare parts availability. * * * * * consider replacing the device if repairs are not feasible or cost - effective. * * @topic:2


# # # issue - loose nut on bed leg # # # impact - potential safety hazard for patients and staff - reduced functionality of bed - discomfort for patients # # # possible causes - improper installation - wear and tear - excessive force applied to bed - mechanical failure # # # recommendations - inspect all nuts on bed legs regularly and tighten as needed - use proper torque when tightening nuts - avoid applying excessive force to bed - replace worn or damaged nuts promptly @topic:6


# # # issue - cuff is tightening too much during blood pressure measurement. # # # impact - pain and discomfort for the patient. # # # possible causes - pressure settings on the monitor are set too high. # # # recommendations - adjust the pressure settings on the monitor to an appropriate level. @topic:3


# # # issue * monitor batteries need to be replaced. # # # impact * the monitor will not be able to operate without a power source. # # # possible causes * the batteries have reached the end of their lifespan. * the batteries are not properly seated. * there is a problem with the battery charger. * there is a problem with the electrical system of the monitor. # # # recommendations * replace the batteries with new ones of the same type and capacity. * ensure that the batteries are properly seated. * check the battery charger and electrical system of the monitor for any problems. @topic:11


a minimum distance between portable and mobile rf communications equipment and the monitor as recommended below, according to the maximum output power of the communications equipment. in the following table, p is the maximum output power rating of the transmitter in watts ( w ) according to the transmitter manufacturer and d is the recommended separation distance in meters ( m ). the values given in brackets are for respiration and bis. 80 mhz to 800 mhz 800 mhz to 2, 5 ghz √p equation for respiration anfor respiratiofor respiration and bated max. outpu. 1 ( 0. 4 ). 1 ( 0. 4 ). 2 ( 0. 7 ). 01. 7 ( 2. 2 ). 4 ( 1. 1 ). 4 ( 1. 1 ) 0. 1 w. 3 ( 7. 0 ). 3 ( 3. 5 ). 3 ( 3. 5 ) 1 w. 3 ( 22. 1 ). 8 ( 11. 1 ). 8 ( 11. 1 ) 10 w 2. 0 ( 35. 0 ) 2. 0 ( 35. 0 ) 3. 0 ( 70. 0 ) 00 w 498 40 specifications recommended separation distances for wlan operation to avoid interference in the presence of other rf sources, following separation distances are recommended ; a separation distance of 1 m to bluetooth ( class 1 ), dect and wlan devices ( if connected to a different wlan infrastructure ) operating in the same frequency band. a separation distance of 3 m to other radio transmitters with a pulsed mode of operation and a duty cycle less than 50 %, operating in the same frequency band. • a separation distance of 10 m to a consumer grade microwave oven and 30 m to a commercial en, if operating in the 2, 4 ghz band. • if other continuous wave transmitters operate in the same frequency band, a minimum signal - to - noise ratio ( snr @topic:10


# # # issue - spo2 sensor cable is broken. # # # impact - the patient's oxygen saturation levels cannot be monitored accurately. # # # possible causes - the cable may have been damaged during use. - the cable may have been kinked or pinched. - the cable may have been exposed to moisture or chemicals. # # # recommendations - replace the spo2 sensor cable. - inspect the cable for any damage and replace it if necessary. - protect the cable from moisture and chemicals. @topic:4


# # # issue - battery failure in philips intellivue mp40 ( m8003a ) monitors used for patient transport. # # # impact - potential disruption of patient monitoring during transport. - increased risk of adverse events due to inaccurate or interrupted patient data. - reduced patient safety and comfort. # # # possible causes - * * scheduled battery replacement : * * batteries have a limited lifespan and need to be replaced regularly as part of preventive maintenance. - * * battery degradation over time : * * batteries can degrade over time due to factors such as temperature, usage patterns, and charging cycles. - * * manufacturing defects : * * occasionally, batteries may have manufacturing defects that can lead to premature failure. - * * environmental factors : * * extreme temperatures or excessive humidity can contribute to battery failure. - * * improper handling or storage : * * damage to batteries can occur during handling or storage, affecting their performance. # # # recommendations - * * establish a regular battery replacement schedule : * * replace batteries according to manufacturer guidelines or based on usage data. - * * monitor battery performance : * * regularly check battery levels and performance indicators to identify any potential issues. - * * use high - quality batteries : * * invest in batteries from reputable manufacturers to minimize the risk of defects. - * * optimize battery usage : * * avoid excessive charging or discharging, and follow recommended charging practices. - * * handle and store batteries properly : * * protect batteries from physical damage and extreme environmental conditions. @topic:11


solves the problem perform the “ final actions ” on page 2 - 12, otherwise contact hill - rom technical support. 10. replace the knee motor ( refer to procedure 4. 7 on page 4 - 23 ), then go to “ final actions ” on page 2 - 12, otherwise contact hill - rom technical support. 11. using an ohmmeter, check the state of the foot section motor intermediate switch. • when the rod is extended, contacts 3 and 4 ( see figure 2 - 1 on page 2 - 22 ) of the connector ( socket # 2, green ) are short - circuited. • when the rod is retracted, the circuit is open. the foot section motor intermediate switch is working. yes no ■i — » replace the power supply ( a model beds ) ( refer to procedure 4. 3 ) or the motor control p. c. board ( b model and newer beds ) ( refer to procedure 4. 27 ). if this solves the problem perform the “ final actions ” on page 2 - 12, otherwise contact hill - rom technical support. 12. replace the foot motor ( refer to procedure 4. 8 on page 4 - 27 ). this solves the problem. yes no ■i — » contact hill - rom technical support. 13. go to “ final actions ” on page 2 - 12. page 2 - 22 careassist® and careassist® es bed service manual ( man330 rev 4 ) 2. 8 trendelenburg / reverse trendelenburg malfunction chapter 2 : troubleshooting procedures 2. 8 trendelenburg / reverse trendelenburg malfunction initial conditions : the hilow function works ( see “ function checks ” on page 2 - 5 ). 1. at least one of the other functions is working. yes no i - > go to rap 2. 1. 2. do the following to make sure the @topic:9


# # # issue - malfunctioning motor in the foot of the electric bed # # # impact - the bed is unable to move or adjust, which could impact patient care and comfort. # # # possible causes - wear and tear from frequent use - electrical issues - manufacturing defects # # # recommendations - replace the malfunctioning motor with a new one - inspect the bed's electrical system for any loose connections or повреждения - contact the manufacturer for further troubleshooting and support @topic:9


# # # issue : equipment not finding patient's name # # # impact : delays in patient care, increased workload for healthcare professionals, potential for errors # # # possible causes : - misconfigured software or settings - incorrect patient information entered - network connectivity issues - hardware malfunction # # # recommendations : - verify software and settings are correct - ensure patient information is entered accurately - check network connectivity and resolve any issues - if issue persists, troubleshoot hardware @topic:11


# # # issue - ekg not displaying on monitor # # # impact - inability to monitor patient's ekg, which is essential for diagnosis and treatment. # # # possible causes - faulty lead wires - loose connection between lead wires and monitor - problem with monitor's ekg input port - software issue on the monitor # # # recommendations - check the lead wires for any damage or loose connections. - ensure that the lead wires are securely connected to both the monitor and the patient. - inspect the monitor's ekg input port for any debris or damage. - restart the monitor to reset any software issues. - if the issue persists, contact the manufacturer for further troubleshooting assistance. @topic:18


# # # issue : trolley difficult to open ; malfunctioning wheels # # # impact : - impaired patient mobility and transportation - increased risk of injury to patients and staff - reduced efficiency in patient care # # # possible causes : * * trolley difficult to open : * * - obstruction or damage to the trolley's opening mechanism - foreign objects lodged in the track or hinges - misalignment or wear and tear of components * * malfunctioning wheels : * * - broken or damaged wheel mechanisms - missing or loose wheel bearings - uneven or excessive wear on wheels - obstructions or debris blocking wheel movement # # # recommendations : * * trolley difficult to open : * * - inspect the track and hinges for any obstructions or damage. - remove any foreign objects that may be hindering movement. - realign or replace any misaligned or worn components. * * malfunctioning wheels : * * - check the wheel mechanisms for any broken or damaged parts. - replace missing or loose bearings. - adjust or repair uneven or excessive wheel wear. - remove any obstructions or debris that may be hindering wheel movement. @topic:24


# # # issue - misalignment of the pin connector on the monitor and patient transport device, resulting in difficulty connecting the accessories. # # # impact - inability to effectively monitor and transport patients, leading to potential delays in patient care. # # # possible causes : - # # # # * * defective pin connector : * * the pin connector on either the monitor or patient transport device may be damaged or deformed, causing misalignment and preventing proper connection. - # # # # * * improper handling : * * rough handling or misuse of the equipment could have caused the pin connector to bend or become damaged. - # # # # * * incorrect pin placement : * * the pin connector may not have been properly aligned or inserted into the corresponding port, causing misalignment. # # # recommendations - # # # # * * inspect the pin connector : * * carefully examine the pin connector on both the monitor and patient transport device for any visible damage or deformation. - # # # # * * realign or repair the pin connector : * * if the pin connector is bent, attempt to carefully straighten it using a pair of pliers or tweezers. if the damage is severe, it may require replacement. - # # # # * * ensure proper pin placement : * * when connecting the monitor and patient transport device, make sure that the pin connector is aligned and inserted correctly into the port. @topic:2


# # # issue - bed alarm is not connected to the bed. # # # impact - patients may not be able to call for help if they need it. - staff may not be able to hear the bed alarm if it is not connected. # # # possible causes - the bed alarm cord may be damaged or disconnected. - the bed alarm may not be turned on. - the bed alarm may be malfunctioning. # # # recommendations - inspect the bed alarm cord for damage or disconnection. - turn on the bed alarm. - test the bed alarm to make sure it is working properly. - if the bed alarm is still not working, replace it. @topic:26


# # # issue - the tail of the bed cannot be put on. # # # impact - the bed cannot be used. # # # possible causes - the tail of the bed is not compatible with the bed frame. - the tail of the bed is damaged. - the bed frame is damaged. # # # recommendations - check to ensure that the tail of the bed is compatible with the bed frame. - inspect the tail of the bed for damage. - inspect the bed frame for damage. @topic:1


# # # issue * bed adjustment malfunction # # # impact * inability to adjust the bed's height and position, affecting patient comfort and mobility. # # # possible causes * * * faulty remote control : * * the replacement remote may be defective or incompatible with the bed. * * * electrical connection issues : * * loose wires or damaged connectors can interrupt the communication between the remote and the bed's control unit. * * * motor problems : * * the motor responsible for adjusting the bed's head may be faulty or damaged, preventing it from responding to input from the remote. * * * control unit malfunction : * * the electronic control unit that receives and processes input from the remote may be experiencing issues, affecting the bed's ability to adjust. # # # recommendations * * * verify remote functionality : * * test the replacement remote with another compatible bed or use a remote tester to confirm its proper operation. * * * inspect electrical connections : * * check for loose wires, damaged connectors, or corrosion, and tighten or replace any faulty connections. * * * troubleshoot motor : * * perform continuity tests on the motor and its wiring to identify any electrical faults. replace the motor if necessary. * * * examine control unit : * * conduct diagnostics on the control unit to identify any software or hardware issues. update firmware or replace the control unit as needed. @topic:7


# # issue - the provided information does not contain a specific cause of failure for the medical device. # # impact - the lack of a clear cause of failure makes it difficult to understand why the device is not functioning as intended. # # possible causes - the provided information is limited to a general statement that the device was tested and found to be operating normally. # # recommendations - further investigation is needed to determine the root cause of the failure. - the technician should perform additional tests or diagnostics to identify the specific component or issue that is causing the device to fail. - once the root cause is identified, the technician can take appropriate action to fix the device and restore it to normal operation. @topic:12


# # # issue the electric bed manufactured by hill - rom ( model 900b ) is currently experiencing a fault where the head of the bed is unable to be adjusted. # # # impact - the inability to adjust the head of the bed could result in patient discomfort and make it difficult for healthcare professionals to provide adequate care. # # # possible causes - electrical issue : faulty wiring or connections could be preventing the bed's motor from receiving power or responding to commands. - mechanical issue : wear and tear or misalignment of mechanical components, such as gears or pulleys, could be hindering the bed's movement. - control system issue : a problem with the bed's control panel or software could be causing it to malfunction. - user error : if the bed has not been operated correctly or has been subjected to excessive force, it could have caused damage to its components. # # # recommendations to resolve this issue and prevent future occurrences, the following steps are recommended : - * * inspect electrical connections : * * check all electrical connections, including plugs, wires, and terminals, for any loose, damaged, or corroded components. - * * examine mechanical components : * * inspect the bed's gears, pulleys, and other mechanical parts for any signs of wear, misalignment, or damage. ensure that all components are properly lubricated. - * * test control system : * * using the bed's manual or diagnostic tools, test the control panel and software to identify any errors or malfunctions. - * * review user instructions : * * ensure that the bed is being operated according to the manufacturer's instructions and that it is not being subjected to excessive force or misuse. - * * contact manufacturer or service technician : * * if the issue cannot be resolved through the above steps, contact the manufacturer or a qualified @topic:1


##116c recorder module these parts are exclusively for m1116c part number 12 nc description 453564452831 453564452831 m _ rec recorder module text 453564452841 453564452841 m _ rec recorder module symbol support parts 453564456821 453564456821 m _ rec mechasy roller 240 6 parts ) part number 12 nc description 453564456791 453564456791 m _ rec plast snap lock kit ( 2ea ) 865115 intellibridge ec10 module part number 12nc description 453564116661 453564116661 ib - ecio module rs232 / lan 453564116931 453564116931 ib - ecio frt housg assy module rs232 / lan m1031 - 42201 453564116891 ib - eclo bezel rj45 865383 nmt module part number 12nc description 453564279141 453564279141 m nmt module nmt, eng 453564279161 453564279161 m _ nmt module nmt, inti 453564257851 453564257851 m _ nmt plast bezel 453564279181 453564279181 m _ nmt frt housg assy nmt, eng 453564279191 453564279191 m _ nmt frt housg assy nmt, inti 989803174581 989803174581 cbl nmt patient cable * 453564464081 453564464081 m @topic:19


# # # issue - med admin system cannot connect # # # impact - delay in medication administration - potential patient safety risks # # # possible causes - network connectivity issues - software glitches - hardware malfunction - configuration errors # # # recommendations - * * restart the device * *. this can resolve temporary software glitches or network connectivity issues. - * * check network connections * *. ensure that the device is connected to a stable network. - * * verify software updates * *. check if there are any pending software updates that may resolve the issue. - * * inspect hardware * *. check if there is any visible damage or loose connections that may be causing the malfunction. - * * reconfigure the device * *. reset the device to its factory settings and reconfigure it according to the manufacturer's instructions. - * * contact the manufacturer * *. if the issue persists, contact the manufacturer for technical assistance. @topic:16


# # # issue - bed does not adjust # # # impact - patient discomfort - delays in treatment - increased risk of injury to patient or staff # # # possible causes - motor malfunction - control panel failure - electrical issue - mechanical obstruction # # # recommendations - check the motor for any signs of damage or wear. - inspect the control panel for any loose connections or damaged components. - check the electrical connections for any loose or damaged wires. - inspect the bed frame for any obstructions that may be preventing the bed from adjusting. - if the issue cannot be resolved by the technician, contact the manufacturer for further assistance. @topic:1


# # # issue - battery failure of welch allyn connex spot monitor # # # impact - inability to monitor vital signs of patients # # # possible causes - battery degradation due to extended usage # # # recommendations - replace the battery with a new one @topic:11


# # # issue - motor failure in the huntleigh contour 390 electric bed # # # impact - the bed is not able to move or adjust, which can impact patient care and comfort. # # # possible causes - the motor may have become misaligned or damaged due to use or wear and tear. - the motor may have been overloaded, causing it to fail. - there may be a problem with the electrical connection to the motor. # # # recommendations - replace the motor with a new one. - check the alignment of the motor and ensure that it is properly secured. - check the electrical connections to the motor and ensure that they are secure. - regularly inspect and maintain the bed to prevent future failures. @topic:9


# # # issue - cuff failure in philips suresigns vs4 vital signs monitor # # # impact - inaccurate or unreliable blood pressure measurements - potential patient safety concerns # # # possible causes - cuff deterioration due to age, wear, or misuse - cuff puncture or tear - cuff bladder leakage - loose or disconnected cuff connections # # # recommendations - replace the cuff with a new one - inspect the cuff regularly for signs of damage or wear - handle the cuff carefully to avoid punctures or tears - ensure that the cuff connections are secure - calibrate the monitor after replacing the cuff - document the repair and testing results for future reference @topic:15


- 64134 451261000241 m3012a front bezel ( pressure, temp, press / temp ),, dutch not shown m3012 - 64135 451261000251 m3012a front bezel ( pressure, temp, press / temp ), spanish not shown 198 measurement server extension parts ( m3012a, m3014a, m3015a and m3016a ) 6 parts part number 12nc part no. description item m3012 - 64137 451261000261 m3012a front bezel ( pressure, temp, press / temp ), norwegian not shown m3012 - 64138 451261000271 m3012a front bezel ( pressure, temp, press / temp ), swedish not shown m3012 - 64139 451261000281 m3012a front bezel ( pressure, temp, press / temp ), finnish not shown m3012 - 64144 451261000291 m3012a front bezel ( pressure, temp, press / temp ), portuguese not shown m3012 - 64145 451261000301 m3012a front bezel ( pressure, temp, press / temp ), greek not shown m3012 - 64147 451261000311 m3012a front bezel ( pressure, temp, press / temp ), russian not shown m3012 - 64149 451261000321 m3012a front bezel ( pressure, temp, press / temp ), czech not shown m3012 - 64150 451261000331 m3012a front bezel ( pressure, temp, press / temp ), polish not @topic:19


# # # issue : bed not adjusting # # # impact : patient discomfort and potential safety risks # # # possible causes : - * * remote control malfunction : * * the remote control may be faulty or damaged, preventing it from transmitting signals to the bed. - * * damaged actuator : * * the actuators responsible for adjusting the bed may be malfunctioning or physically damaged, preventing them from responding to commands. - * * electrical issues : * * loose or disconnected electrical connections, power outages, or faulty wiring can disrupt the bed's operation. - * * software problems : * * a software glitch or corruption can interfere with the bed's functionality, causing it to become unresponsive. - * * mechanical issues : * * mechanical components such as gears, bearings, or hydraulic cylinders may have worn out or become damaged, hindering the bed's ability to adjust. # # # recommendations : - * * troubleshoot remote control : * * check the remote control for any visible damage or loose buttons. replace the batteries if necessary. - * * inspect actuators : * * examine the actuators for any signs of wear or physical damage. test their functionality using the remote control or manual controls. - * * check electrical connections : * * inspect all electrical connections for looseness, damage, or corrosion. ensure that the bed is properly plugged in and receiving power. - * * contact manufacturer : * * if the issue persists despite troubleshooting, contact the manufacturer of the bed for technical support or warranty coverage. - * * perform regular maintenance : * * conduct regular preventative maintenance on the bed to identify and resolve potential problems before they escalate into failures. this includes checking electrical connections, cleaning components, and lubricating moving parts. @topic:1


# # # issue battery not charging # # # impact patient monitoring may be compromised if the battery is not able to provide backup power during a power outage. # # # possible causes - battery has reached the end of its service life - battery is not properly seated in the device - battery charger is not functioning properly # # # recommendations - replace the battery with a new one - ensure that the battery is properly seated in the device - test the battery charger to ensure that it is functioning properly @topic:11


# # # issue - loose button on the physiologic monitor # # # impact - ekg measurements cannot be obtained # # # possible causes - loose nut holding the button in place # # # recommendations - tighten the nut holding the button in place @topic:16


" page 77 ). • risk preventiogeneral recommendations in general : [UNK] check that nothing ( e. g. objects, accessories or power cable ) or any persons ( e. g. children, limbs ) will interfere with the movement of the mobile parts of the bed before actuating them. an intermittent beep sounds when one of the bed's movements is hindered. always check ( e. g. to and fro movements ) that the various locking mechanisms are in good working order ( e. g. siderails, extensions, grip handles, brakes ). sufficiently qualified nursing staff determine the usage condition suitable for the various functions and the degree of supervision to ensure that the patient uses the bed safely. [UNK] when the patient is left unattended : apply the brakes to prevent any risks of falling, especially if the patient leans on the bed when getting in or out, • leave the sleep surface in the lowest position to avoid serious consequences in the event of falling, use the siderails to secure the patient and reduce the risk of falling accidentally,. • lock any function that, if misused, could worsen existing injuries or pathologies, or even result in bodily injury, • never leave the bed in the trendelenburg position. a never modify the bed without hill - rom's prior written consent. alterations could result in injury to the patient or damage to the bed. only use manufacturer's parts and accessories. never place objects or equipment on the chassis or use it to support a person. do not use of the bed with loads in excess of the safe working load. in the case of patients suffering from particular behavioral difficulties ( e. g., agitation, mental confusion, loss of sense of direction, obsessive behavior, old patients, weakness, etc. ), properly trained medical staff should ascertain how the siderails should be used ( irres @topic:9


service tool ( wast ) can create the'waclientcert. pim'file using only a'ca cert'( server verification ), or a'ca cert'and'client cert'( for client authentication ). create a folder on the pc or thumb drive and place the ca cert, and if needed, the client cert in the same folder. 1. connect a device to the pc running the wast and allow the device to come up to a usable state. 2. click select to choose the desired device from the wast'device list '. 3. se welch allyn service tool [UNK] file administration help welcome, administhursday 2021 / 07 / 08 11 : 41 log off evice ork liwelch allyn connex device to view device details, click on the desired row, and then click select. serial number ip address devices asset tag location onnex spot monitor 100001774714 100011654819 hca. 0. 0welch allyn connex device how online and offline devices [UNK] select click the configure tab and then click change... in the current settings pane. 4. 230 connex direct welch [UNK] connex® spot monitor i welch allyn service tool [UNK] file administration help welcome, administrator log off ednesday 2021 / 07 / 21 welch allyn connex device x device list work list radio ip address. 0. 0. 0 ip address device name serial number ref welch allyn connex device 00011654819 0. 0. 0. 0 view log ethernet mac address 00 : 1a : fa : 2a : 2d : 24 asset tag vsm 6000 series ocation id refresh erify and calibrate device information reports pgradconfigure ocation id : current settings asset tag @topic:10


# # # issue - the scanner of the medical cart is not working. # # # impact - the medical cart cannot be used to transport patients or equipment. # # # possible causes - the scanner may be defective. - the scanner may not be properly connected to the medical cart. - the scanner may not be properly configured. - the scanner may be experiencing a software issue. # # # recommendations - inspect the scanner for any damage. - ensure that the scanner is properly connected to the medical cart. - verify that the scanner is properly configured. - update the scanner software to the latest version. - if the scanner is still not working, contact the manufacturer for assistance. @topic:23


# # # issue - broken sensor wire # # # impact - inability to accurately measure patient's vital signs # # # possible causes - * * defective sensor : * * the sensor itself may be faulty and causing the wire to break. - * * physical damage : * * the wire may have been damaged due to improper handling or external forces. - * * improper maintenance : * * lack of regular maintenance and calibration can lead to wear and tear of the wire. # # # recommendations - * * replace sensor : * * the defective sensor should be replaced with a new one. - * * inspect wiring : * * check the entire length of the wire for any visible damage or loose connections. - * * proper handling and maintenance : * * ensure the equipment is handled and maintained according to manufacturer's instructions to prevent future damage. @topic:22


# # # issue - the electric bed is vibrating when being raised or lowered. # # # impact - the vibration makes the bed unsafe and uncomfortable for patients to use. # # # possible causes - the hi - low column is worn and needs to be replaced. # # # recommendations - replace the hi - low column. @topic:1


# # # issue monitor not displaying accurate data from brown lead # # # impact inaccurate patient monitoring data due to disconnected lead. # # # possible causes - broken brown lead wire # # # recommendations * * immediate actions : * * - change with new lead - check the lead for any cuts or damage. - if the lead is damaged, replace it with a new one. * * preventive actions : * * - inspect the leads regularly for any damage. - handle the leads with care to avoid damage. - store the leads in a safe place when not in use. @topic:18


# # # issue - side rail does not lock # # # impact - patient safety risk - patient discomfort - increased workload for nursing staff # # # possible causes - damaged side rail mechanism - loose or missing screws - worn or broken locking mechanism - misalignment of side rail with bed frame # # # recommendations - inspect side rail mechanism for damage or wear - tighten loose screws and replace missing screws - replace worn or broken locking mechanism - ensure side rail is properly aligned with bed frame @topic:6


the check boxes reflect the current treatment status. ending the sepsis management bundle the sepsis management bundle normally ends 24 hours after the sepsis resuscitation bundle started, as documented by the bundle timer on the protocolwatch screen. you are asked to confirm with the yes key that routine monitoring should now resume. by selecting no you can extend the sepsis management bundle phase by an additional 6 hours. after completion of the sepsis management bundle a sepsis standby phase begins which allows time for further patient stabilization and recovery. during this phase severe sepsis screening is suspended. the standby phase is by default 7 days but the length can be adjusted in configuration mode. 318 ssc sepsis protocol 25 protocolwatch viewing and printing the protocol log the protocol log stores all user interactions, alarms and phase transitions while the ssc sepsis protocol is running. to view the log select main setup - > protocolwatch - > protocol log you can choose between three different views of the protocol log : • summary view - a short form which does not include when the windows were triggered • full view - a complete log of all user interactions, alarms and phase transitions • phase view - you can view log entries for one phase by selecting that phase on the orientation line when the log consists of several pages, you can move through the pages with the previous page and next page keys. to start a protocol log printout, select print. the log is cleared whenever a patient is discharged. always discharge patients it is essential that patients are discharged when monitoring ends. as a reminder, the monitor can be configured to display a pop - up window when the monitor has been switched off, in standby mode, or not monitoring basic vitals for a certain time. the window asks whether a new patient is being monitored and offers a pop - up key to discharge the previous patient if this is the case. the window must be configured in configuration @topic:8


that equipment. relevant standards for some non - medical electrical equipment may have limits for equipment leakage currents higher than required by the standard lec / en 60601 - 1 - 1 or lec / en 60601 - 1 + a1 ed. 3 clause 16. these higher limits are acceptable only outside the patient environment. it is essential to reduce equipment leakage currents to values specified in lec / en 60601 - 1 when non - medical electrical equipment is to be used within the patient environment. 52 3 testing and maintenance ] system example this illustration shows a system where both the medical electrical equipment and the non - medical electrical equipment are situated at the patient ’ s bedside. non - medical devices medical devices personal computer distance to patient must be > = 1. 5m key : power cables : - isolation transformer data cables : ; intelllvue patient monitoi warning • do not use additional ac mains extension cords or multiple portable socket - outlets. if a multiple portable socket - outlet is used, the resulting system must be compliant with lec / en 60601 - 1 - 1 or lec / en 60601 - 1 + a1 ed. 3 clause 16. do not place multiple socket - outlets on the floor. do not exceed the maximum permitted load for multiple socket - outlets used with the system. do not plug additional multiple socket outlets or extension cords into multiple socket outlets or extension cords used within the medical electrical system. • do not connect any devices that are not supported as part of a system. • do not use a device in the patient vicinity if it does not comply with lec / en 60601 - 1 or lec 60601 - 1 edition 3 clause 16. the whole installation, including devices outside of the patient vicinity, must comply with lec / en 60601 - 1 - 1 or lec / en 60601 - 1 @topic:10


# # # issue - spo2 probe failure # # # impact - inability to accurately monitor oxygen saturation levels - potential delays in patient care - increased risk of patient complications # # # possible causes - * * device malfunction : * * the spo2 probe or monitor may be defective. - * * incorrect probe placement : * * the probe may not be securely attached to the patient's finger, or it may be placed on the wrong finger. - * * patient movement : * * excessive patient movement can interfere with the probe's ability to obtain an accurate reading. - * * environmental factors : * * bright ambient light or electromagnetic interference can affect the probe's signal. # # # recommendations - * * check the device : * * ensure that the spo2 probe and monitor are functioning properly by testing them on a known good patient. - * * reposition the probe : * * securely attach the probe to the patient's finger and ensure that it is placed on the correct finger. - * * minimize patient movement : * * instruct the patient to remain still during the spo2 measurement. - * * eliminate environmental interference : * * move the patient to a location with minimal ambient light or electromagnetic interference. - * * contact the manufacturer : * * if the problem persists, contact the manufacturer for further troubleshooting and repair options. @topic:4


* * issue : * * equipment readings are inaccurate compared to calibration device. * * impact : * * inaccurate readings can lead to misdiagnosis or mistreatment of patients. * * possible causes : * * * * * defective sensor : * * the sensor responsible for measuring the physiological parameter may be faulty or have become damaged. * * * electronic component failure : * * electronic components within the monitor, such as capacitors or resistors, may have failed, leading to incorrect signal processing. * * * software glitch : * * software errors or bugs can disrupt the monitor's ability to interpret and display accurate readings. * * * environmental conditions : * * extreme temperature, humidity, or electromagnetic interference can affect the monitor's performance. * * * improper calibration : * * the monitor may not have been calibrated properly or the calibration parameters were not followed correctly. * * recommendations : * * * * * inspect the sensor : * * check the sensor for any physical damage, such as loose connections or leaks, and ensure it is properly connected to the monitor. * * * test electronic components : * * use a multimeter or other diagnostic tools to test the electronic components for continuity, voltage, and resistance. * * * update software : * * check for software updates from the manufacturer that may address known issues. * * * adjust environmental conditions : * * ensure the monitor is operating in a temperature - and humidity - controlled environment free from electromagnetic interference. * * * recalibrate the monitor : * * perform a comprehensive calibration using an appropriate calibration device and follow the manufacturer's calibration procedure precisely. * * * refer to the manufacturer : * * if the issue persists after troubleshooting, consult the manufacturer for technical support or replacement parts. @topic:20


# # # issue - inability to adjust the bed rails # # # impact - patient safety compromised due to potential falls or entrapment - reduced functionality of the bed # # # possible causes - defective or missing wheels on the bed rails - misalignment or damage to the bed rail adjustment mechanism - electrical or mechanical failure # # # recommendations - inspect the bed rails and replace any missing or damaged wheels. - verify the alignment and functionality of the adjustment mechanism. - check electrical connections and ensure that the motor is operating correctly. - consider a full system diagnostic test to identify any underlying issues. @topic:9


# # # issue the feet of the electric bed cannot be adjusted up or down using the remote control. # # # impact the patient may not be able to get into or out of bed comfortably, which could lead to falls or other injuries. # # # possible causes - the remote control is not working properly. - the electrical cord is damaged. - the motor that adjusts the feet is not working properly. - the mechanical linkage between the motor and the feet is damaged. # # # recommendations - check the remote control batteries and replace them if necessary. - inspect the electrical cord for damage and replace it if necessary. - contact a qualified biomedical engineer to troubleshoot and repair the motor or mechanical linkage. @topic:1


# # # issue > spo2 sensor failure on a philips intellivue mp20 ( m8001a ) physiological monitor. # # # impact > inability to accurately measure and monitor patient's oxygen saturation levels ( spo2 ), which is crucial for proper patient care. # # # possible causes * * * defective spo2 sensor : * * the sensor itself may be damaged or malfunctioning due to manufacturing defects, physical damage, or prolonged use. * * * loose or disconnected sensor cable : * * the cable connecting the sensor to the monitor may have become loose or disconnected, resulting in intermittent or lost signal. * * * interference : * * external factors such as ambient light, nail polish, or patient movement can interfere with the sensor's ability to accurately measure oxygen saturation levels. * * * software malfunction : * * a software issue or glitch within the monitor may prevent proper communication with the spo2 sensor or interfere with its functionality. # # # recommendations * * * inspect and replace the spo2 sensor : * * verify if the sensor is properly attached and undamaged. if necessary, replace the sensor with a new one. * * * check cable connections : * * ensure that the sensor cable is securely connected to both the sensor and the monitor. if loose or damaged, replace the cable. * * * rule out interference : * * ensure that there are no sources of interference in the patient's environment that could affect the sensor's readings. * * * restart the monitor : * * power cycling the monitor may resolve software glitches or temporary communication issues. * * * contact technical support : * * if the issue persists, contact the manufacturer's technical support for further troubleshooting and repair. @topic:4


# # # issue - the electric bed is not functioning properly. # # # impact - the patient cannot be adequately cared for. - the staff cannot move the patient safely or comfortably. # # # possible causes - the bed is not plugged in. - the power cord is damaged. - the control panel is not functioning properly. - the bed motor is malfunctioning. - the bed frame is damaged. # # # recommendations - check that the bed is plugged in and that the power cord is not damaged. - if the control panel is not functioning properly, try resetting it. - if the bed motor is malfunctioning, contact a biomedical engineer. - if the bed frame is damaged, do not use the bed and contact a biomedical engineer. @topic:9


# # # issue - bed cannot be adjusted # # # impact - patient discomfort - difficulty in performing medical procedures - increased workload for healthcare staff # # # possible causes - * * electrical fault : * * loose connections, damaged wiring, or faulty electrical components can prevent the bed from receiving power or responding to control inputs. - * * mechanical failure : * * worn or broken gears, motors, or actuators can impede the bed's movement. - * * software glitch : * * a software error or malfunction could disrupt the bed's control system, causing it to become unresponsive. - * * lack of maintenance : * * regular lubrication and inspection of the bed's components can help prevent failures. - * * patient overload : * * exceeding the weight capacity of the bed can strain its components and lead to malfunction. # # # recommendations - * * electrical troubleshooting : * * inspect for loose connections and damaged wiring. test electrical components with a multimeter to identify and replace faulty parts. - * * mechanical inspection : * * check for worn gears, broken motors, or damaged actuators. lubricate moving parts and replace any defective components. - * * software update : * * contact the manufacturer to obtain and install any available software updates that may resolve software glitches. - * * preventive maintenance : * * establish a regular schedule for inspecting and servicing the bed to prevent future failures. - * * weight management : * * ensure that the patient's weight does not exceed the bed's weight capacity to avoid overloading. @topic:1


# # # issue - battery is not charging # # # impact - device cannot be used without battery power - monitoring of vital signs may be interrupted # # # possible causes - faulty adapter - loose connection between adapter and device - damaged battery # # # recommendations 1. replace the adapter with a new one 2. check the connection between the adapter and device to ensure it is secure. 3. if the battery is damaged, replace it with a new one. @topic:11


* * possible cause of failure : * * * * * power supply issues : * * the device may not be receiving sufficient power, leading to unexpected shutdowns. * * * malfunctioning mainboard : * * the mainboard is responsible for controlling various functions of the device, including power management. a faulty mainboard can cause erratic operation and shutdowns. * * * faulty sensors or connections : * * sensors and connections ensure proper data collection and communication within the device. loose or damaged connections can lead to intermittent power shutdowns. * * * battery issues : * * if the device relies on a battery as a backup power source, a failing battery can cause unexpected shutdowns. * * * environmental factors : * * extreme temperatures, humidity, or vibrations can affect the device's performance and contribute to unexpected shutdowns. @topic:11


# # # issue - the bed cannot be adjusted due to a stuck lock button on the remote control. # # # impact - patients may be unable to get into or out of bed, or may be uncomfortable due to the bed being in an awkward position. # # # possible causes - the remote control may have been dropped or damaged. - the lock button may be jammed due to debris or dirt. - the remote control may be malfunctioning. # # # recommendations - inspect the remote control for damage. - clean the remote control with a damp cloth. - try resetting the remote control by pressing and holding the power button for 10 seconds. - if the remote control is still not working, replace it. @topic:7


laws. philips medizin systeme boblingen gmbh hewlett - packard str. 2 71034 boblingen, germany the information contained in this document is subject to change without notice. philips makes no warranty of any kind with regard to this material, including, but not limited to, the implied warranties or merchantability and fitness for a particular purpose. philips shall not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance, or use of this material. passwords in order to access different modes within the monitor a password may be required. the passwords are listed below. monitoring mode : no password required configuration mode : 71034 demo mode : 14432 service mode : 1343 consult the configuration guide before making any changes to the monitor configuration. warnings and cautions in this guide : • a warning alerts you to a potential serious outcome, adverse event or safety hazard. failure to observe a warning may result in death or serious injury to the user or patient. • a caution alerts you where special care is necessary for the safe and effective use of the product. failure to observe a caution may result in minor or moderate personal injury or damage to the product or other property, and possibly in a remote risk of more serious injury. 14 and cautions 1 introduction note when an intellivue mp40 / mp50, software revision b. o with battery option installed is used together with an intellivue infromation center d. 01 or e. o and the monitor issues battery - related inops, these inops are displayed as “ unknown ” on the intellivue information center. upgrade the information center text catalog to e. 01 if using an mp40 / mp50 monitor with the information center. 15 1 introduction warnings and cautions 16 2 ■ ) theory of operation monitor @topic:10


# # # issue - battery failure in philips suresigns vs4 monitor. # # # impact - the monitor may not be able to provide accurate vital signs readings. - the monitor may not be able to be used in emergency situations. - the monitor may need to be replaced, which can be costly. # # # possible causes - the battery is old and has reached the end of its lifespan. - the battery has been damaged due to physical or electrical abuse. - the battery has been exposed to extreme temperatures or humidity. # # # recommendations - replace the battery with a new one. - inspect the battery for any signs of damage. - store the battery in a cool, dry place. - follow the manufacturer's instructions for battery care and maintenance. @topic:11


* * issue : * * - monitor, patient transport brand philips model intellivue mp40 ( m8003a ) failed. * * impact : * * - the device is not functional and cannot be used to monitor or transport patients. * * possible causes : * * - the battery is depleted and needs to be replaced. * * recommendations : * * - replace the battery with a new one. - test the device to ensure it is functioning properly. @topic:11


# # # issue - the electric bed model 900b by hill - rom cannot be adjusted up or down. # # # impact - the patient may not be able to get in or out of bed. - the patient may not be able to be positioned comfortably for medical procedures. # # # possible causes - the remote control may not be working properly. - the bed may not be plugged in properly. - the bed may be in the wrong mode. - the bed may be damaged. # # # recommendations - check the remote control batteries. - make sure the bed is plugged in properly. - check the bed mode. - if the bed is still not working, contact a qualified biomedical engineer. @topic:1


# # # issue - cuff of the bedside physiologic monitor is damaged. # # # impact - the monitor cannot accurately measure patient's vital signs, such as blood pressure, heart rate, and oxygen saturation. # # # possible causes - * * excessive use : * * the cuff may have been used too frequently or for too long, causing it to stretch and lose its elasticity. - * * improper handling : * * the cuff may have been handled roughly, causing it to puncture or tear. - * * environmental factors : * * the cuff may have been exposed to extreme temperatures or humidity, causing it to deteriorate. # # # recommendations - * * replace the cuff : * * the damaged cuff should be replaced with a new one. - * * regular maintenance : * * the monitor and cuff should be inspected regularly for signs of damage or wear and tear. - * * proper handling : * * the cuff should be handled carefully and stored in a protected environment. @topic:5


network cables must be unshielded. connectors the actual placement of boards and configuration of connections for your monitor depends on how your hardware has been configured. see the symbols table on page 403 to see which symbols are used to mark the connections. 405 38 installation and specifications installation safety information warning connect only medical devices to the ecg output connector socket. mp20 / mp30 mp20 / mp30 left side of monitor 1 mms and one extension 2 ecg analog output / marker input connector mp20 / mp30 rear of monitor 406 installation safety information 38 installation and specifications 1 measurement link connector 2 ac power inlet 3 remote alarm connector 4 analog video out connector 5 wired network connector 6 wireless network connector 7 battery compartment 8 equipotential grounding 9 protective earth 10 space for optional interface boards, e. g. serial / mib ( rs232 ) connectors, or optional parallel printer connection 11 side cover with antenna for intellivue instrument telemetry wireless network 407 38 installation and specifications installation safety information mp40 / mp50 mp40 / mp50 rear of monitor 1 space for optional interface boards, e. g. serial / mib ( rs232 ) connectors, or optional parallel printer connection 2 equipotential grounding 3 protective earth 4 ac power inlet 5 wireless network connector for use with wireless lan or instrument telemetry adapter 6 remote alarm connector 7 analog video out connector 8 wired network connector 9 battery compartment 10 measurement link connector 11 intellivue instrument telemetry adapter 12 monitor connector ( to 5 above ) 13 12v dc power inlet ( to 5 above ) 14 antenna connector 408 installation safety information 38 installation and specifications mp40 / mp50 left side of monitor 3 2 mp60 / mp70 right side of monitor ( mp60 / 70 ) 1 2 3 4 5 6 7 8 9 10 11 1 mms and one extension 2 plug @topic:19


# # # issue - bed is not adjustable # # # impact - patient discomfort, reduced mobility # # # possible causes - failure of electrical components in the control box - malfunctioning of the motor or gears - damage to the wiring or cables # # # recommendations - replace or repair the faulty electrical components in the control box - inspect and repair or replace the motor or gears - check and secure the wiring and cables - perform regular maintenance and inspections to prevent future failures @topic:1


# # # issue - the electric bed is unable to adjust its head down. # # # impact - patients may be unable to get into or out of bed comfortably. - patients may be at risk of injury if they attempt to adjust the bed themselves. - the bed may be unusable, which could lead to delays in patient care. # # # possible causes - faulty remote control - loose wires or connections - malfunctioning motor - software issue # # # recommendations - reset the remote control. - check the wires and connections for any loose or damaged connections. - contact the manufacturer for assistance if the problem persists. @topic:1


# # # issue - cuff failure # # # impact - inability to measure patient's vital signs accurately # # # possible causes - cuff damage due to improper use or handling - cuff size mismatch - cuff tubing leaks # # # recommendations - ensure proper cuff usage and handling techniques - select the appropriate cuff size for the patient - inspect cuff tubing regularly for leaks - replace damaged cuffs promptly @topic:5


# # # issue - drawer cannot be closed. # # # impact - the medical cart cannot be used properly. # # # possible causes - the solenoid is not working properly. - the drawer tracks are damaged. - the drawer latch is not engaging properly. # # # recommendations - replace the solenoid. - inspect the drawer tracks for damage and replace if necessary. - inspect the drawer latch for damage and replace if necessary. @topic:21


# # # issue - battery deterioration # # # impact - the device may not function properly or may not be able to be used at all. # # # possible causes - age of the battery - use of the battery in extreme temperatures - overcharging or undercharging of the battery # # # recommendations - replace the battery with a new one. - charge the battery regularly and according to the manufacturer's instructions. - avoid using the battery in extreme temperatures. @topic:11


# # # issue - monitor not displaying graphs # # # impact - inability to monitor patient's vital signs # # # possible causes - faulty mms module - loose connection between mms module and monitor - software glitch # # # recommendations - replace mms module - inspect and tighten all connections - update software if necessary @topic:22


##vo 8000 golvo 8008 2000015 olvo 8008 lowb2000019 when the viking xl is used with a bed equipped with 125 mm diameter casters *, when lowering the bed to the low position, make sure that the elevation arms do not hit the chassis of the patient hoist. recommended part numbers for overbed tables overbed table ta270 a519 overbed table overbed table ta529 endboards lockable * headboard lockable * footboard the afssaps headboard and footboard meet the demands stipulated by the afssaps in the " decision dated 26 / 04 / 2010 " ( article 2 ) for use with children measuring less than 146cm in height. headboard the headboard is fitted with fins that must point towards the sleep surface. if the headboard is installed in the bed frame the wrong way round, the risk of entrapment increases. standard headboard * ef 16947ef 17677atn2169an2292a if the headboard is removed from the bed frame, the risk of patient entrapment or falling increases. similarly, the use of the accessories installed at the head of the bed ( e. g., iv poles, helpers, etc. ) can incur risks for the patient. the headboard can be removed for easier access to the patient's head. endboard fastening system * bed frame extension * 5 1 & do not sit or climb on the extension. the extension can be pulled out by 20 cm in intermediate steps of 4 cm. cushion for extensions is available as accessory. all stop adlocated at the head of the bed, the extractible wall stop rotects the bed against impacts with walls or headwalls. a store the wall stop before moving. take out the wall stop store the wall stop mobilizing the patient electrical functions the @topic:9


# # # issue - siderail adjustment failure # # # impact - patient safety and comfort - reduced functionality of the bed # # # possible causes - cable damage or obstruction - board malfunction - software glitch - user error # # # recommendations - inspect and replace damaged cables - test board functionality and repair or replace if necessary - update software if available - provide training to users on proper siderail adjustment techniques @topic:6


# # # issue - broken solenoid # # # impact - inability to access drawers # # # possible causes - wear and tear - poor maintenance - manufacturing defect # # # recommendations - replace solenoid - lubricate solenoid regularly - conduct preventive maintenance @topic:21


# # # issue - cuff of philips suresigns vs4 vital signs monitor is deteriorated. # # # impact - inaccurate blood pressure readings. - potential for patient harm. # # # possible causes - normal wear and tear. - improper use or storage of the cuff. - exposure to extreme temperatures or humidity. # # # recommendations - replace the cuff with a new one. - ensure proper use and storage of the cuff. - inspect the cuff regularly for signs of wear and tear. - calibrate the monitor according to the manufacturer's instructions. @topic:15


# # # issue battery not charging, reporting - 1 error code on medical cart model bdms 01. # # # impact the medical cart cannot be used, potentially impacting patient care. # # # possible causes * battery is faulty * charging port is damaged * charging cable is faulty * power source is not providing enough power # # # recommendations * replace the battery * inspect the charging port for damage and clean if necessary * try using a different charging cable * check the power source and ensure it is providing the correct voltage and amperage * contact the manufacturer for further troubleshooting and support @topic:13


wiring schematic plug 8 - pin, plug banana male conductor coloring 1 « € ga blue 2 < ey 3 ae a 4 [ tk jreen e — 7 5 < / ellow 7 dk — are 8 « i me e > ne 41 spo2 ( m1020a ) : wiring schematic plug 12 - pin plug banana male conductor coloring blue — l ink green fellow brown t pi r = 100 ohm t oo ) o cm a p ou mm — shield n. c. es n. c. 42 invasive pressure : wiring schematic plug 12 - pin plug banana male conductor coloring a blue b grey c dink d green 1 ellow 5 brown 8 red | shield ne. > m. c. 43 m1006b # c01 : wiring schematic phone jack 1 / 8 " plug banana stereo wiring schematic plug # - contact plug banana male conductor coloring 1 | | a ee 44 co2 ( mp5, m3014a ) : wiring schematic plug 8 - pin, plug banana male conductor coloring i ae blue 2 « ix — grey 3 ake pink i je — green 4 5 < ru 6 e aoe 7 ak red 8 « | nc « e — — — — — — — — esheld nc 45 co2 ( m1016a, m3016a ) : wiring schematic plug 12 - pin plug banana male conductor coloring blue gre dink green „ 4 ellow j | brown red ( 4 ) co ue rm oi mm uu t ou — shield nc n. c. 4 = all resistors 120 kohm 46 cardiac output : wiring schematic plug 12 - pin plug banana male conductor coloring blue wey mj pink green ellow brown req do es oo lw pb ww n. c — — — — — — — @topic:10