10000 ft ) non - operating - 500 m to 12000 m ( 40000 ft ) docking station m8045a item condition range temperature range operating oto 40°c ( 32 to 100°f ) non - operating - 20 to 60°c ( - 4 to 140°f ) humidity range operating 20 % to 85 % relative humidity ( rh ) ( non condensing ) non - operating 5 % to 85 % relative humidity ( rh ) altitude range operating - 500 m to 3000 m ( 10000 ft ) non - operating - 500 m to 12000 m ( 40000 ft ) thermal array recorder module m1116b item condition range temperature range operating + 5 to 45°c ( 41 to 113°f ) non - operating - 10 to 70°c ( 14 to 158°f ) 418 monitor performance specifications 38 installation and specifications thermal array recorder module m1116b item condition range humidity range operating 95 % relative humidity ( rh ) max @ 40°c ( 104°f ) ( non condensing ) non - operating 95 % relative humidity ( rh ) max @ 65°c ( 150°f ) ( non condensing ) altitude range operating up to 3048 m ( 10000 ft ) non - operating up to 3048 m ( 10000 ft ) remote speedpoint m8026a item condition range temperature range operating 0 to 55°c ( 32... 130°f ) storage - 20 to 60°c ( - 4... 140°f ) humidity range operating 95 % rh max. at 40°c ( 100°f ) storage 85 % rh max. at 50°c ( 120°f ) altitude range operating - 500 to 4. 600 m ( - 1600... 15000 ft ) storage - 500 to 13. 100 m ( - 1600... 43000 ft ) remote alarm @topic:19


# # # issue - failure to connect to the saturation pulse oximetry ( spo2 ) signal. # # # impact - inability to monitor patient's oxygen saturation levels, which is critical for patient care. # # # possible causes - * * faulty spo2 cable : * * the spo2 cable may be damaged or defective, preventing proper signal transmission. - * * improper cable connection : * * the spo2 cable may not be securely connected to the monitor or the patient's sensor. - * * sensor malfunction : * * the spo2 sensor may be faulty or not properly placed on the patient's finger or toe. - * * software or hardware issue : * * there may be a problem with the monitor's software or hardware that is preventing it from detecting the spo2 signal. # # # recommendations - * * replace the spo2 cable : * * verify that the new cable is compatible with the monitor and the patient's sensor. - * * inspect cable connections : * * ensure that the cable is securely and properly connected to both the monitor and the patient's sensor. - * * check sensor placement : * * make sure that the sensor is correctly placed on the patient's finger or toe and that there is no interference from nail polish or other obstructions. - * * restart the monitor : * * power down and restart the monitor to refresh its software and hardware. - * * contact technical support : * * if the issue persists, contact the manufacturer's technical support team for further assistance. @topic:4


# # # issue : * missing connector # # # impact : * the equipment cannot be used for patient transport. # # # possible causes : * the connector may have been accidentally removed during maintenance or use. * the connector may have become damaged or worn out over time. # # # recommendations : * check the connector to ensure that it is securely attached. * if the connector is damaged, replace it with a new one. * inspect the equipment regularly to identify any potential issues that could lead to the connector becoming loose or damaged. @topic:2


# # # issue - spo2 cable is faulty # # # impact - the patient's vital signs may not be accurately monitored. # # # possible causes - the cable may be physically damaged. - the cable may be old and worn out. - the cable may be incompatible with the monitor. # # # recommendations - replace the faulty cable with a new one. - ensure that the new cable is compatible with the monitor. - regularly inspect the cables for damage and replace them as needed. - train staff on the proper use and care of the equipment. @topic:4


panel ” on page 5 - 11. 2 pull the antenna cable upward to disconnect it from the wireless board. 3 ‘ a a a v. j ( antenna waster jf l d yq 4 remove the torx t10 screw from the wireless board and remove the board from the front panel. repairing the monitor suresigns vs2 * and vsi service guide removing the antenna removing the antenna to remove the antenna : step 1 separate the front panel from the rear case assembly. see “ removing the front panel ” on page 5 - 11. disconnect the antenna cable from the wireless board and remove the washer. antenna cable unscrew the antenna and remove it from the front panel. repairing the monitor suresigns vs2 * and vsi service guide 5 - 33 removing the antenna repairing the monitor 5 - 34 suresigns vs2 * and vsi service guide spare parts 6 replacement parts and assembly drawings the following table lists the suresigns vs2 * and vsi spare parts, their philips part numbers, and in which of the following figures they appear. see “ assembly drawings ” on page 6 - 3 for the referenced figures. note — for clarity, some screws and connector lines do not appear in the drawings. description part number figure item antenna 4535 642 70421 6 - 1 5 battery ( lithium ion ) 989 803 174881 6 - 2 19 battery compartment cover 4535 642 00031 6 - 2 18 communications ( i / o ) board 4535 641 70121 | 6 - 2 15 cover ( faceplate, without recorder ) 4535 642 70081 6 - 2 10 front panel without lcd, vsi 4535 642 70161 6 - 1 6 front panel without lcd, vs2 * 4535 642 70171 6 - 1 6 lcd assembly 4535 64 @topic:19


# # # issue - battery depleted - broken storage basket - impeded wheels # # # impact - impeded patient monitoring - reduced storage capacity - impaired mobility # # # possible causes * * battery depleted * * - extended use without recharging - faulty charging mechanism * * broken storage basket * * - rough handling - loose screws or fixtures * * impeded wheels * * - accumulation of dirt and debris - worn - out bearings - uneven flooring # # # recommendations * * battery depleted * * - implement regular charging schedules - inspect charging mechanism for any defects * * broken storage basket * * - tighten loose screws or replace fixtures - handle equipment with care * * impeded wheels * * - clean wheels and bearings regularly - replace worn - out components - ensure smooth flooring surface @topic:30


# # # issue - battery indicator displays red color despite charging # # # impact - inability to monitor patient's physiological parameters accurately and continuously # # # possible causes - loose battery connection at the back of the device # # # recommendations - ensure that the battery is securely fitted into the designated compartment at the back of the device. - check the battery contacts for any damage or corrosion. clean the contacts if necessary. - if the issue persists, replace the battery with a new one. @topic:11


# # # issue - the server responsible for receiving patient data is experiencing issues. # # # impact - the inability to access patient data, potentially leading to delays in diagnosis and treatment. # # # possible causes - bug in the server software - network connectivity issues - hardware failure # # # recommendations - contact the vendor ( md ) and it team to resolve the server issue. - once the server is resolved, verify the functionality of the patient data scanning system. - implement regular monitoring and maintenance procedures to prevent similar issues in the future. @topic:16


# # # issue black screen on the monitor ( medical cart, bdms, bdms 01 ) # # # impact inability to use medical cart for patient care # # # possible causes - main computer overheating - faulty monitor - loose or damaged cables - power supply issues # # # recommendations - ensure that the main computer is adequately cooled and not overheating - check the monitor for any physical damage or loose connections - inspect all cables for damage or loose connections - verify that the power supply is providing the correct voltage and amperage - consult with the manufacturer for further troubleshooting and repair instructions @topic:13


# # # issue - remote control not responding # # # impact - inability to adjust bed settings, such as height, angle, and position. # # # possible causes - malfunctioning or damaged remote control - loose or disconnected wires or connections - power supply issues - software or firmware errors - mechanical failure in the bed's control system # # # recommendations - replace or repair the remote control. - check and tighten all wires and connections. - ensure the power supply is stable and working properly. - update the firmware or software if available. - contact the manufacturer for technical support or troubleshooting guidance. @topic:7


# # # issue - cuff bp is faulty. # # # impact - the patient's blood pressure cannot be accurately measured. # # # possible causes - the cuff may be damaged. - the cuff may be the wrong size for the patient. - the cuff may not be properly inflated. # # # recommendations - replace the cuff. - use the correct size cuff for the patient. - inflate the cuff properly. @topic:3


fill printer paper tray check cabling replace i / o board status message “ print device remote 1 ( remote 2, remote 3 ) unavailable ” is issued. printer job is stalled print error on philips information center network connection to philips information center not functioning print a test report on the philips information center. if this fails, refer to philips information center documentation check that the network connection between the monitor and the philips information center is working 97 4 troubleshooting troubleshooting guide symptoms cause of failure failure isolation and remedy status message “ printing on device remote 1... ( remote 2, remote 3 ) ” is issued but no report is printed print queue on philips information center is full. reasons for this may be : - printer is not switched on - printer paper tray is empty switch on printer power fill printer paper tray printouts are not as expected printer paper size is not correctly configured printer resolution is not correctly configured printer color support is configured to “ on ” although the printer does not support color printer not compatible configure the paper size according to the inserted print media configure the printer resolution according to the printer capabilities configure the printer color support to “ off ” check specifications recorder symptom possible cause corrective action system thinks that door is open when it is not. defective door switch. replace door switch. exchange module. system thinks that the recorder is out of paper when it is not. paper - out sensor dirty. clean paper - out sensor. recorder not communicating with system. poor connection to the front - end 4 - slot rack unplug the module. plug it back in and try it again in a few seconds. ( watch for the led to flash. ) only one recorder module may be used with each monitor. remove one of the recorder modules. system not configured properly. check the configuration of the connected monitor. too many modules connected. check and remove the extra modules. @topic:28


# # # issue - spo2 sensor failure # # # impact - inability to accurately monitor patient's oxygen saturation levels # # # possible causes - * * defective sensor * * : the spo2 sensor itself may be faulty or damaged, leading to incorrect readings or complete failure. - * * loose connections * * : loose connectors between the sensor and the monitor can cause intermittent or complete failure of the spo2 measurement. - * * software issues * * : software glitches or bugs within the monitor can interfere with the proper functioning of the spo2 sensor. - * * environmental factors * * : extreme temperatures or excessive moisture can adversely affect the performance of the spo2 sensor. # # # recommendations - * * replace the sensor * * : if the sensor is defective, it should be replaced with a new one. ensure proper connection and fit to prevent loose connections. - * * inspect connections * * : check all connections between the sensor and the monitor for any loose or damaged wires. tighten or replace connectors as necessary. - * * update software * * : if software issues are suspected, check for available software updates from the manufacturer and install them. - * * control environmental factors * * : ensure that the monitor and sensor are operated within the specified temperature and humidity ranges. @topic:4


* * issue : * * battery failure in philips intellivue mp40 ( m8003a ) patient transport monitor. * * impact : * * * disruption of patient monitoring during transport. * potential safety hazards due to unreliable vital sign measurements. * increased risk of patient discomfort or harm. * * possible causes : * * * battery degradation due to normal wear and tear. * improper charging or use of the batteries. * battery manufacturing defects. * * recommendations : * * * * * regular battery maintenance and replacement : * * follow manufacturer's guidelines for battery replacement based on usage and age. * * * proper charging practices : * * use the designated charger and follow charging instructions to prevent overcharging or undercharging. * * * careful handling : * * avoid dropping or exposing batteries to extreme temperatures or moisture. * * * battery storage : * * store spare batteries in a cool, dry place to extend their lifespan. * * * inspection and testing : * * regularly inspect batteries for any signs of damage or malfunction. * * * consider battery alternatives : * * explore alternative battery technologies or power sources that may provide longer life or better reliability. * * * manufacturer support : * * contact philips technical support if the issue persists or if further assistance is required. @topic:11


the mains power supply for the bed must comply with relevant standards : nf c 15 - 100 and nf c 15 - 211 ( france ),. international electrotechnical commission ( iec ) 364 for other locations. • check that the bed's power requirements on the identification label ( see " electrical characteristics " page 13 ) correspond to the power supply voltage of the hospital. the power supply should be equipped with a maximum 30 ma earth an = 30ma leakage circuit breaker, in compliance with iec 364 - 5 - 53. [UNK] the parts of the bed that are within the patient's reach, even if they are under the frame, are applied parts. if the integrity of the protective conductor is in doubt, the beds fitted with batteries must be used in battery mode. in compliance with standards relating to electromagnetic interference for medical equipment, this product does not interfere with other medical devices or is not susceptible to interference when combined with other medical devices that also omply with the electromagnetic standards in force. some devices, particularly older ones that do not comply with the electromagnetic compatibility standards, may however undergo interference or may themselves interfere with the working of this product. the users of such devices are responsible for ensuring that any malfunctions will not endanger the patient or any other person. ensure that the power cord is unplugged and hooked to the bed before moving the bed ( see " securing the power cable " page 76 ). only duly qualified and authorized staff should carry out electrical maintenance. never clean or service the bed without unplugging it from the mains power supply and disconnecting the battery. the battery backup must never be left in direct contact with fire, placed in liquid, or discarded in a refuse bin. in the event of the battery being damaged, see " de - commissioning " page 82. age 8 this label indicates that the bed must never be used with @topic:9


# # # issue leaking cuff in the bedside physiologic monitor # # # impact - inaccurate blood pressure measurements - false alarms - patient discomfort # # # possible causes - cuff material degradation - loose connections - damaged tubing - software glitch # # # recommendations - replace the cuff with a new one of the same size and type. - inspect the tubing and connections for damage and replace as necessary. - update the software if a newer version is available. - calibrate the monitor according to the manufacturer's instructions. - train staff on proper use and maintenance of the monitor. @topic:3


# # # issue - the cuff of the physiologic monitor ( bedside ) is leaking air, causing inaccurate blood pressure ( bp ) measurements. # # # impact - inaccurate bp readings can lead to incorrect diagnoses and treatment decisions. - delays in patient care due to the need to repeat bp measurements and troubleshoot the device. - loss of confidence in the device and its readings. # # # possible causes - cuff was not properly attached or inflated. - cuff bladder is damaged or punctured. - air leak in the tubing or connectors. - malfunctioning pressure sensor. - software or hardware issue with the monitor. # # # recommendations - inspect the cuff, tubing, and connectors for any damage or leaks. - replace the cuff if necessary. - calibrate the monitor according to the manufacturer's instructions. - contact the manufacturer for technical support if the issue persists. - consider using an alternative bp measurement device until the issue is resolved. @topic:3


# # # issue - alarm systolic is set to 120 mmhg # # # impact - the alarm may not sound when the patient's systolic blood pressure drops below 120 mmhg, which could lead to a delay in treatment. # # # possible causes - the alarm was accidentally set to 120 mmhg. - the alarm settings were not properly checked after the device was installed. - the alarm system is malfunctioning. # # # recommendations - check the alarm settings on the device and make sure that they are set to the appropriate values. - test the alarm system to make sure that it is functioning properly. - document the alarm settings and the results of the test in the device's maintenance log. @topic:2


# # # issue - remote control malfunction # # # impact - inability to operate the electric bed effectively. # # # possible causes - * * defective remote control : * * the remote control may have internal malfunctions or damage, leading to failure. - * * battery depletion : * * low or exhausted batteries can prevent the remote control from functioning properly. - * * interference or signal disruption : * * external factors like electromagnetic interference or obstacles between the remote and the bed receiver can disrupt communication. - * * loose connections or wiring issues : * * faulty connections or frayed wires within the remote or the bed can cause intermittent or complete loss of functionality. - * * software or firmware glitches : * * in advanced remote controls, software errors or firmware updates can introduce unexpected issues. # # # recommendations - * * inspect remote control : * * examine the remote for any visible damage or loose components. check connections and ensure the battery is charged. - * * test with different batteries : * * replace the old batteries with new ones to eliminate any issues related to battery power. - * * rule out interference : * * move the bed away from potential sources of electromagnetic interference and ensure there are no obstructions blocking the signal path. - * * check wiring and connections : * * inspect the remote control and bed for loose wires or damaged connectors. tighten or replace as necessary. - * * reset or update firmware : * * if the remote has advanced features, consider resetting it or updating the firmware. consult the manufacturer's guidelines for specific instructions. - * * consult technical support : * * if the issue persists after troubleshooting, seek assistance from the manufacturer's technical support team or a qualified biomedical engineer. @topic:7


# # # issue - cuff leak # # # impact - inaccurate blood pressure measurements - misdiagnosis or delayed treatment # # # possible causes - worn or damaged cuff - loose or broken connections - malfunctioning cuff pump # # # recommendations - replace the cuff with a new one - inspect the connections and ensure they are secure - check the cuff pump for proper operation @topic:3


# # # issue - equipment : bed, electric - brand : huntleigh - model : contoura 390 - symptom : bed not adjusting # # # impact - patient discomfort - delays in treatment - increased risk of injury to patients and staff # # # possible causes - broken remote control cord - faulty motor - electrical issues - software errors # # # recommendations - inspect remote control cord for damage and replace if necessary - test motor and electrical connections - update software if available - contact manufacturer for further troubleshooting assistance @topic:7


# # # issue - toe adjustment not working # # # impact - patient discomfort - prolonged immobilization - delays in patient care # # # possible causes - electrical fault in the toe adjustment motor - mechanical failure in the toe adjustment mechanism - software glitch in the bed control system # # # recommendations - inspect the toe adjustment motor and wiring for any damage or loose connections. - check the toe adjustment mechanism for any broken or worn components. - reset the bed control system and update the software. - if the issue persists, contact the manufacturer for further troubleshooting and repairs. @topic:1


# # # issue - the bed, electric, huntleigh, enterprise 8000 is not adjusting. # # # impact - patients may not be able to be positioned correctly for treatment. - staff may have to manually adjust the bed, which could be difficult and time - consuming. # # # possible causes - error in the control box - electrical problem - mechanical problem # # # recommendations - refresh the control box. - check the electrical connections. - inspect the mechanical components for damage. @topic:1


# # # issue : - 木 [UNK] かロックされない 。 # # # impact : - [UNK] [UNK] かヘットから [UNK] [UNK] する [UNK] [UNK] [UNK] 。 # # # possible causes : - 木 [UNK] のラッチか [UNK] [UNK] している 。 - 木 [UNK] の [UNK] り [UNK] けか [UNK] んている 。 - [UNK] [UNK] モータか [UNK] [UNK] している 。 # # # recommendations : - 木 [UNK] のラッチを [UNK] [UNK] し 、 [UNK] [UNK] に [UNK] して [UNK] [UNK] する 。 - 木 [UNK] の [UNK] り [UNK] けを [UNK] め [UNK] ける 。 - [UNK] [UNK] モータを [UNK] [UNK] し 、 [UNK] [UNK] に [UNK] して [UNK] [UNK] する 。 @topic:12


software issue is suspected, contact the manufacturer to obtain the latest software updates and follow the installation instructions carefully. - * * calibrate display : * * after any repairs or replacements, recalibrate the display to ensure accurate readings and optimal performance. @topic:2


# # # issue monitor not linking # # # impact - inability to monitor patient's vital signs - delay in diagnosis and treatment # # # possible causes - loose or damaged cables - software glitch - hardware failure # # # recommendations - inspect and tighten cables - restart the monitor - contact the manufacturer for further troubleshooting and possible repair @topic:16


# # # issue - broken lead wire ( yellow ) on a bedside physiologic monitor # # # impact - inability to monitor patient's electrocardiogram ( ecg ) # # # possible causes - physical damage to the lead wire, such as a cut or tear - loose connection between the lead wire and the monitor or patient - electrical fault within the lead wire # # # recommendations - replace the broken lead wire with a new one - ensure that all connections are secure - inspect the lead wire for any other damage or defects - test the monitor and lead wire to verify that they are functioning properly @topic:18


# # # issue - bed not adjustable # # # impact - patient discomfort - difficulty in providing medical care # # # possible causes - loose ac connection # # # recommendations - verify ac connection is secure - test bed operation to ensure proper functionality @topic:1


# # # issue - the electric bed's headrest is not adjusting up or down. # # # impact - the patient may be uncomfortable if they cannot adjust the headrest to a comfortable position. - the patient may be unable to see the television or read if the headrest is not raised high enough. - the patient may be unable to eat if the headrest is not lowered low enough. # # # possible causes - the motor that adjusts the headrest may be faulty. - the wiring that connects the motor to the bed may be damaged. - the control panel that adjusts the headrest may be faulty. - the headrest may be obstructed by something, preventing it from moving. # # # recommendations - inspect the motor, wiring, and control panel for any damage. - if any damage is found, replace the damaged component. - if no damage is found, reset the bed by unplugging it from the power outlet and then plugging it back in. - if the bed still does not adjust, call a qualified technician to service it. @topic:1


* * issue : * * - cuff detaching due to adhesive failure * * impact : * * - inaccurate blood pressure readings - patient discomfort - potential for harm if blood pressure readings are not accurate * * possible causes : * * - cuff not properly attached - cuff adhesive worn or degraded - skin preparation not done correctly - patient sweating or movement - other factors that affect adhesive performance, such as humidity, temperature, or skin oils * * recommendations : * * - ensure cuff is properly attached to patient's arm. - inspect cuff regularly and replace if adhesive is worn or degraded. - prepare patient's skin properly before attaching cuff. - consider using a different type of cuff if patient's skin is particularly sensitive or prone to movement. - use a cuff cover or other device to protect the cuff from moisture and other environmental factors that could affect adhesive performance. @topic:3


* * issue : * * cuff failure in vital signs monitor * * impact : * * * inaccurate vital signs readings * delays in patient monitoring * increased patient discomfort * * possible causes : * * * * * cuff degradation : * * cuffs deteriorate over time due to repeated use, exposure to moisture, and cleaning chemicals. this can weaken the material and lead to leaks or tears. * * * improper inflation : * * over - or under - inflation can damage the cuff, especially if it occurs repeatedly. * * * external damage : * * the cuff can be accidentally punctured or damaged by sharp objects, such as needles. * * * manufacturing defects : * * rarely, cuffs may have manufacturing flaws that compromise their integrity. * * recommendations : * * * * * regular maintenance and inspection : * * schedule routine inspections and maintenance to identify and replace worn cuffs promptly. * * * proper use and inflation : * * train staff on the correct inflation technique and ensure that cuffs are not over - or under - inflated. * * * avoid damage : * * store cuffs in a safe and dry location, away from sharp objects or hazards. * * * use high - quality cuffs : * * invest in durable cuffs from reputable manufacturers to minimize the risk of premature failure. * * * consider alternatives : * * if repeated cuff failures occur, consider using non - cuff - based methods for monitoring vital signs, such as fingerpulse oximetry or telemetry. @topic:5


# # # issue - wifi not accessible on medical cart # # # impact - medical cart cannot be used wirelessly # # # possible causes - wifi network not configured - incorrect network password - wifi signal strength too weak - wifi router not working properly - firewall blocking connection - network adapter not enabled - [ 6bih - n2 ] wifi not encrypted # # # recommendations - check if wifi network is configured and configured correctly - check if network password is correct - check if wifi signal strength is strong enough - check if wifi router is working properly - check if firewall is blocking connection - check if network adapter is enabled - [ 6bih - n2 ] encrypt wifi network @topic:13


common mode rejection ratio diagnostic mode : > 86 db ( with a 51 kq2 / 47 nf imbalance ). filter mode : > 106 db ( with a 51 k2 / 47 nf imbalance ). electrode offset potential tolerance + 500 mv 476 40 specifications ecg / arrhythmia / st performance specifications auxiliary curr ( leads off detection ) active electrode : < 100 na reference electrode : < 900 na input signal range + 5 mv cg / arrhythmia / st / qt adjustment range alarm specifications 15 to 300 bpm adult : 1 bpm steps ( 15 to 40 bpm ) hr maximum delay : 10 seconds according to aami ec 13 - 1992 pedi / neo : 1 bpm steps ( 15 to 50 bpm ) eps ( 50 to 300 bpm ) difference to high limit 0 t50 bpm extreme tachy bpm stepclamping at 150 to 300 bpdifference to low limit 0 to 50 bpm bpm stepxtreme bradbpm stepbpm stepclamping at 15 to 100 bpm one, fixed setting 2 pvcs not adjustable by user run pvcs pvcs rate 1 to 99 pvcs / minute 1 pvc 0 to 300 bpbpvent tach hr 3 to 99 pvcs / minute vent tach run vent rhythm run 1 pvc 3 to 99 pvcs / minute 1 pvc 120 to 300 bpm bpsvt hr svt run 3 to 99 sv beats 1 sv beat t hig - 19. 8 to + 20 mm. 2 m - 20 to + 19. 8 mm. 2 mst low - 20 to + 20 mm ste limit 0. 2 mm qtc high 0 ms step200 ms to 800 ms 0 ms stepaqtc high 30 ms to 200 ms ecg / arrhythmia / st supplemental information as required by @topic:8


# # # issue medical cart model bdms 01 is unable to connect to the bhq _ corp wifi network. # # # impact the medical cart cannot be used to access the hospital's electronic health records system. # # # possible causes - the medical cart's wifi adapter is not working properly. - the medical cart is not within range of the bhq _ corp wifi network. - the bhq _ corp wifi network is down. - the medical cart's wifi settings are not configured correctly. # # # recommendations - check the medical cart's wifi adapter to make sure it is working properly. - move the medical cart closer to the bhq _ corp wifi network. - check the status of the bhq _ corp wifi network. - check the medical cart's wifi settings and make sure they are configured correctly. @topic:13


# # # issue - oxygen saturation ( spo2 ) function is not working properly and cannot measure spo2. # # # impact - inability to accurately monitor patient's oxygen saturation levels, which is crucial for assessing respiratory function and providing appropriate care. # # # possible causes - * * faulty spo2 sensor : * * the sensor that detects the amount of oxygen in the blood may be malfunctioning or damaged. - * * loose or damaged spo2 cable : * * the cable that connects the sensor to the monitor may be loose or damaged, causing intermittent or unreliable readings. - * * probe placement error : * * the spo2 sensor may not be placed properly on the patient's finger or earlobe, resulting in inaccurate measurements. - * * equipment malfunction : * * there may be an issue with the internal components of the monitor that is affecting the spo2 function. # # # recommendations - * * inspect spo2 sensor : * * check the sensor for any visible damage or wear. clean the sensor tip gently with an alcohol swab. - * * check spo2 cable : * * ensure that the cable is securely connected to both the sensor and the monitor. inspect the cable for any cuts or breaks. - * * reposition probe : * * remove the sensor from the patient and reapply it, making sure it is properly positioned on the patient's finger or earlobe. - * * restart equipment : * * turn off the monitor and restart it to reset any software glitches that may be causing the issue. - * * contact manufacturer : * * if the above steps do not resolve the problem, contact the manufacturer of the equipment for further troubleshooting and possible repair or replacement. @topic:4


# # # issue remote control is not functioning properly, affecting the adjustment of the head and leg positions of the bed. # # # impact the patient may experience discomfort or difficulty in finding a comfortable position, which can hinder their recovery or treatment. # # # possible causes 1. * * electrical or mechanical failure in the remote control : * * the remote control may have a faulty battery, loose wiring, or a malfunctioning internal component. 2. * * signal interference : * * the remote control may not be able to send or receive signals due to interference from other electronic devices, such as cell phones or wi - fi routers. 3. * * misconfiguration or user error : * * the remote control may have been accidentally reprogrammed or misused, leading to its failure to function as expected. # # # recommendations 1. * * check the battery and replace it if necessary : * * a weak or dead battery can prevent the remote control from operating. 2. * * inspect the wires and connections : * * ensure that all wires and connections are properly secured and free of damage. 3. * * re - pair or replace the remote control : * * if the issue persists after checking the battery and wires, the remote control may need to be repaired or replaced. 4. * * investigate possible signal interference : * * identify any potential sources of interference and relocate the bed or remote control to minimize its impact. 5. * * review the user manual and ensure proper configuration : * * follow the manufacturer's instructions to ensure that the remote control is correctly configured for the specific bed model. @topic:7


# # # issue the nibp monitor, dinamap procare 100, manufactured by ge healthcare, is not charging properly. # # # impact the device is unable to function properly without a charged battery. this could lead to delays in patient care or inaccurate readings. # # # possible causes * battery is faulty * charging port is damaged * charger is not working properly * software issue # # # recommendations * replace the battery. * inspect the charging port for damage. * try a different charger. * contact the manufacturer for technical support. @topic:11


# # # issue - inability to adjust the bed using the remote control # # # impact - difficulty for patients to adjust their bed position, leading to discomfort and potential complications. # # # possible causes - faulty remote control - circuit issues causing interference with remote signals - malfunction of the bed's electrical system # # # recommendations - * * verify the remote control : * * ensure the remote control is properly connected, has fresh batteries, and is functioning correctly. test on another bed if possible. - * * inspect the wiring : * * check if there are any loose connections or damaged wires between the remote control and the bed. tighten connections and replace faulty wires as needed. - * * test the bed's electrical system : * * use a multimeter to measure voltage and continuity throughout the electrical system. look for any abnormal readings or open circuits that may interfere with the remote's operation. - * * replace components : * * if the issue persists, consider replacing the remote control or the bed's electrical components, such as the control unit or actuators. @topic:7


# # # issue - rust on the electric bed. # # # impact - compromised bed safety and stability. - potential electrical hazards. - patient discomfort and dissatisfaction. # # # possible causes - * * corrosion : * * exposure to moisture, bodily fluids, or cleaning agents can cause metal components to rust. - * * poor maintenance : * * lack of regular cleaning and inspection can lead to rust buildup. - * * defective coating : * * faulty or damaged paint or protective coatings can expose metal surfaces to corrosive environments. # # # recommendations - * * regular cleaning and inspection : * * establish a maintenance schedule for routine cleaning and inspection of equipment to identify and address potential rust issues. - * * proper cleaning agents : * * use non - corrosive cleaning solutions and avoid exposing the bed to harsh chemicals. - * * protective coatings : * * ensure proper paint or protective coatings are applied and maintained to shield metal surfaces from moisture and corrosion. - * * regular inspection for rust : * * inspect equipment regularly for signs of rust and take prompt action to remove it. - * * immediate repair : * * if rust is detected, schedule repairs immediately to prevent further damage and ensure patient safety. @topic:9


# # # issue - failed spo2 cable # # # impact - inability to monitor patient's oxygen saturation levels # # # possible causes - physical damage to the cable - electrical malfunction in the cable - compatibility issues between the cable and the monitor # # # recommendations - replace the spo2 cable with a new one - inspect the cable for any visible damage - ensure that the cable is compatible with the monitor - test the new cable to ensure that it is functioning properly @topic:4


elch allynii hillrom tmn connex® spot monitor h hillrom. clinician name : location 03 : 00 00993369000 barker - scotch, david, a d & b. ulse rate nibp 93 a 129 / 67 s002 h 4 start sys / dia mmhg ( maps ) 20 hillrom. bpm spo2 source : sp spo2 temperature 97. 9 a 94 % a °f ( 36. 6°c ( pi 56 ) vrbl cvdr resp bhvr eye gcs clear save home patient alarms review settings welch allyn connex spot monitor ao when. com service manual software version 1. 5x © 2023 welch allyn. all rights are reserved. the purchaser is permitted to copy this publication, for internal distribution only, from the media provided by welch allyn. no other use, reproduction, or distribution of this publication, or any part of it, is permitted without written permission from welch allyn. legal statement. welch allyn, inc. ( " welch allyn " ) assumes no responsibility for any injury to anyone that may result from ( i ) failure to properly use the product in accordance with the instructions, cautions, and warnings published in this manual or the statement of intended use published in the instructions for use, or ( ii ) any illegal or improper use of the product. welch allyn, connex, suretemp, flexiport, and surebp are registered trademarks of welch allyn. eco cuff is a trademark of welch allyn. lncs, resposable, and rainbow are trademarks of, and set, lnop, and masimo are registered trademarks of masimo corporation. possession or purchase of a masimo spo2 - equipped device does not convey any express or implied license to use the @topic:8


# # # issue - faulty ekg cable ( red ) # # # impact - inability to accurately monitor patient's electrocardiogram ( ecg ) # # # possible causes - wear and tear of the cable over time - improper handling or storage - manufacturing defect # # # recommendations - replace the faulty ekg cable with a new one - ensure proper handling and storage of cables - perform regular inspections and maintenance of medical equipment - contact the manufacturer if the issue persists or for guidance on troubleshooting @topic:18


# # # issue the electric bed manufactured by hill - rom ( model : basic care, p1440a ) is not adjustable. # # # impact the inoperability of the electric bed hinders its intended purpose of providing support and mobility for patients, affecting their comfort and overall care. # # # possible causes * * electrical malfunction : * * - faulty wiring or loose connections within the bed's electrical system - damaged or malfunctioning electrical components ( e. g., motor, controller ) * * mechanical issues : * * - broken or worn - out gears or bearings in the adjustment mechanism - obstructed or misaligned parts preventing proper movement * * software error : * * - corrupted or outdated software controlling the bed's functions - communication issues between the control panel and the bed's components # # # recommendations * * troubleshooting and inspection : * * - verify power supply and ensure that the bed is properly plugged in - inspect the electrical wiring and connections for any visible damage or loose contacts - check the motor and other mechanical components for signs of wear, damage, or obstruction * * electrical repairs : * * - replace or repair damaged electrical components as necessary - ensure proper connections and grounding of the electrical system * * mechanical adjustments : * * - lubricate and adjust gears or bearings to ensure smooth movement - realign or replace misaligned parts to restore proper operation * * software updates : * * - check for and install any available software updates for the bed's control system - reset the system or reprogram the software if necessary @topic:9


# # # issue - frequent rebooting of a medical cart # # # impact - interruption of patient care - delays in treatment - loss of data - safety hazards # # # possible causes - software bugs - hardware malfunctions - power supply issues - environmental factors ( e. g., extreme temperatures, humidity ) - user error # # # recommendations - * * software : * * - update the operating system and applications to the latest versions. - run virus and malware scans. - check for software conflicts. - * * hardware : * * - inspect the cart for loose connections or damaged components. - clean the cart and its components regularly. - check the power supply and ensure it is providing a stable voltage. - * * environmental : * * - ensure the cart is operated in a clean and temperature - controlled environment. - avoid exposing the cart to extreme temperatures or humidity. - * * user : * * - train users on proper operation of the cart. - encourage users to report any issues promptly. - * * other : * * - monitor the cart's performance regularly and keep a maintenance log. - consider implementing a preventive maintenance program. @topic:13


troubleshooting is required, reach out to the medical device manufacturer or it support for assistance. @topic:12


connector with the attached and bent msl flex. be careful not to damage the msl flex when pushing the frame connector downwards. 5 position the connector correctly and push it into place. refitting the dc / dc board note this step only has to be done on hw a / b position the dc / dc board and press it down gently. make sure it is connected properly to both connectors indicated in the picture. 200 5 repair and disassembly ) refitting the top cover note to change the top cover of a hw rev c mms ( s / n prefix de610xxxxx ) you have to attach the two cushions which are part of the top cover assembly. these two cushions secure the connection of the msl flex and the nbp flex. note perform the following two steps only on an mms hw rev c 2 position the top cover, then press it back into place until you hear a click or there is no longer a gap between the two covers. 3 the cover has a rubber seal, press the cover firmly together. 201 [ 5 repair and disassembly ] 4 holding the bottom cover firmly in place, slide the two long mounting pins completely back into the mms. make sure there is no gap between the top and bottom cover. refitting the front cover to refit the front cover, press it back into place over the measurement connector hardware until you hear a click. finai inspection perform a final inspection to ensure that : • the msl connector is positioned correctly • there are no gaps between the msl connector and the cover • there is no gap between the top and bottom cover testing to ensure that the mms is functioning correctly, you must perform safety tests and a performance check on it. please refer to the " testing and maintenance " chapter of this service guide. warning when you disassemble / assemble an mms @topic:19


# # # issue - bed brake failure # # # impact - patient safety hazard - injury to patients or staff - equipment damage - downtime and inconvenience # # # possible causes - worn or damaged brake pads - loose or missing brake caliper bolts - brake caliper piston seizure - hydraulic fluid leak - electrical fault in brake control system # # # recommendations - inspect and replace brake pads as needed. - tighten or replace brake caliper bolts. - inspect and lubricate brake caliper piston. - check for and repair any hydraulic fluid leaks. - troubleshoot and repair any electrical faults in the brake control system. - follow manufacturer's guidelines for maintenance and repair procedures. - regularly test the bed brakes to ensure proper functionality. @topic:9


# # # issue - ekg lead wire is not reading values. # # # impact - patient monitoring is compromised. - inability to accurately assess patient's cardiac status. - delays in diagnosis and treatment. # # # possible causes - faulty ekg lead wire. - loose connection between the lead wire and the monitor. - damaged or loose electrodes. - interference from other electrical devices. - improper settings on the monitor. # # # recommendations - replace the ekg lead wire with a known - good one. - ensure that the lead wire is securely connected to the monitor and the patient. - check the electrodes for damage or loose connections. - move the monitor away from any other electrical devices that may be causing interference. - verify that the monitor settings are correct for the patient's condition. - if the problem persists, contact the manufacturer for technical support. @topic:18


# # # issue - equipment : electrical bed - manufacturer : hill - rom - model : care assist es220 - description : bed weight scale function failure during testing # # # impact - inability to accurately measure patient weight - potential impact on patient care and treatment # # # possible causes - faulty weight sensor - electrical component failure - software malfunction # # # recommendations - inspect and replace weight sensor if faulty - check electrical connections and components for damage or loose connections - update software to the latest version - recalibrate the weight scale - if the issue persists, consult the manufacturer for further assistance @topic:25


# # # issue cuff is leaking # # # impact the cuff is not able to properly inflate and measure blood pressure, which can lead to inaccurate readings. # # # possible causes - the cuff is damaged or has a hole. - the cuff is not the correct size for the patient. - the cuff is not properly inflated. # # # recommendations - inspect the cuff for any damage or holes. - make sure the cuff is the correct size for the patient. - properly inflate the cuff. @topic:3


# # # issue : central monitor is not connected to all the rooms. # # # impact : - inability to monitor patient data in all rooms. - delayed or inaccurate patient care. # # # possible causes : - * * network connection issue : * * the central monitor may not be properly connected to the network, preventing it from communicating with the bedside monitors in other rooms. - * * software glitch : * * a software error or bug may be preventing the central monitor from establishing connections with the bedside monitors. - * * hardware malfunction : * * a hardware component in the central monitor or bedside monitors may be malfunctioning, causing the connection issue. - * * incorrect configuration : * * the central monitor or bedside monitors may not be configured properly, preventing them from connecting to each other. # # # recommendations : - * * check network connection : * * verify the network connectivity between the central monitor and the bedside monitors. - * * restart devices : * * restart both the central monitor and the bedside monitors to clear any software glitches. - * * inspect hardware : * * check the cables and connections between the devices to ensure they are secure and undamaged. replace any faulty cables or connectors. - * * review configuration settings : * * confirm that the central monitor and bedside monitors are configured with the correct ip addresses, subnet masks, and gateway information. - * * contact manufacturer support : * * if the issue persists, contact the manufacturer of the central monitor or bedside monitors for technical assistance. @topic:16


##ghijklmnopqrstuvwxyzabcdefghijklmnopqrstuvwxyz0123456789 abcdefghijklmnopqrstunaa / xyzabcdefghijklmnopq 71 3 testing and maintenance printer test report 72 t roubleshooting introduction this section explains how to troubleshoot the monitor if problems arise. links to tables that list possible monitor difficulties are supplied, along with probable causes, and recommended actions to correct the difficulty. how to use this section use this section in conjunction with the sections testing and maintenance and parts. to remove and replace a part you suspect is defective, follow the instructions in the section repair and disassembly. the theory of operation section offers information on how the monitor functions. who should perform repairs only qualified service personnel should open the monitor housing, remove and replace components, or make adjustments. if your medical facility does not have qualified service personnel, contact philips ’ response center or your local philips representative. warning high voltage - voltages dangerous to life are present in the instrument when it is connected to the mains power supply or a battery. do not perform any disassembly procedures ( other than server, extension or module removal ) with power applied to or batteries inserted into the instrument. failure to adhere to this warning could cause serious injury or death. replacement level supported the replacement level supported for this product is to the printed circuit board ( pcb ) and major subassembly level. once you isolate a suspected pcb, follow the procedures in the repair and disassembly section, to replace the pcb with a known good pcb. check to see if the symptom disappears and that the monitor passes all performance tests. if the symptom persists, swap back the replacement pcb with the suspected mal @topic:16


# # # issue - defective cuff # # # impact - inaccurate blood pressure measurements # # # possible causes - excessive wear and tear - improper maintenance - damage to the cuff during use or storage # # # recommendations - replace the cuff with a new one that is compatible with the monitor - ensure that the cuff is properly fitted to the patient's arm - clean and inspect the cuff regularly for any signs of damage or wear - store the cuff in a protected environment when not in use @topic:3


# # # issue - faulty ekg lead cable # # # impact - interfered display - inaccurate readings # # # possible causes - damaged ekg lead cable # # # recommendations - replaced the ekg lead cable @topic:18


error occurred during the last zero calibration. check the airway adapter and clean, if necessary. perform another zero calibration. if the inop persists, contact your service personnel. numeric is replaced by -? inop tone co2change scale the co2 wave is clipped. select a more appropriate wave scale to display the whole wave. 93 4 patient alarms and inops what to do nop message, indicatioco₂notransducer there is no co2 transducer connected. if you replace the transducer, the new transducer must be calibrated. if you silence this inop the co2 measurement will be switched off. from mainstream co2 ( excepm3014a ) numeric is replaced by -? - nop tonco₂zero running wait until zero calibration is finished. co₂zerorequired umeric is replaced by -? nop tone perform zero calibration for the co2 sensor. if the inop persists, contact your service personnel. ipi check pat. age for adult patients : a date of birth has been entered that results in a calculated patient age of less than 12 years. enter correct date of birth or, if patient is pediatric, correct patient category. for pediatric patients : either no date of birth has been entered ( it is required for calculation of ipi for pediatric patients ) or the calculated age is below 1 year or above 12 years. check that correct date of birth is entered. ipi check sources the ipi numeric cannot be provided as one or more of the measurement sources required for ipi is not available. check availability of etco2, awrr, spo2, and pulse rate. ipi neo patient? check patient category. the ipi numeric is not available for neonatal patients. so2 inops in @topic:17


on page 38 ). performance tests see performance assurance tests ( on page 76 ). reporting of results evaluation of results check and prepare for normal use see reporting of test results ( on page 95 ) see evaluation of test results ( on page 97 ) if any single test fails, testing must be discontinued immediately and the device under test must be repaired or labeled as defective. visual inspection before each use check all exterior housings for cracks and damage. check the condition of all external cables, especially for splits or cracks and signs of twisting. if serious damage is evident, the cable should be replaced immediately. check that all mountings are correctly installed and secure. refer to the instructions that accompany the relevant mounting solution. after each service, maintenance or repair event ensure all fuses accessible from the outside comply with the manufacturer ’ s specification. check : - the integrity of mechanical parts, internally and externally. - any damage or contamination, internally and externally - that no loose parts or foreign bodies remain in the device after servicing or repair. - the integrity of all relevant accessories. power on test 1. connect the monitoring system to mains and switch it on. this includes connected displays and mms extensions. 2. make sure that all steps listed in the table initial instrument boot phase in the troubleshooting section are completed successfully and that an ecg wave appears on the screen. the expected test result is pass : the monitor boots up and displays an ecg wave. the wave might be a flat line if no simulator is attached. 35 safety tests 36 the following safety test needs to be performed on the monitoring system : - applied part leakage current - system test ( 1f required ) safety test requirements are set according to international standards, their national deviations and specific local requirements. the safety tests detailed in this service guide are derived from international standards but may not be sufficient to meet local requirements. we recommend that you file the results @topic:10


# # # issue - alarm mms unplugged # # # impact - monitor is unable to work properly and display parameter values - alarm mms unplugged is triggered # # # possible causes - mms ( multi - measurement server ) module may be unplugged or malfunctioning # # # recommendations - check if mms module is properly connected to the device - check if the mms module is functioning properly - if the mms module is malfunctioning, it should be replaced @topic:16


# # # issue - cuff deterioration # # # impact - inability to accurately measure patient vital signs # # # possible causes - normal wear and tear - excessive use - improper storage - exposure to extreme temperatures # # # recommendations - regular maintenance and inspection of the cuff - proper storage and handling of the cuff - replacement of the cuff as needed @topic:5


* * issue : * * - equipment does not turn on. * * impact : * * - patient monitoring and transport is affected. * * possible causes : * * - main board failure or power supply issue. * * recommendations : * * - contact the manufacturer or authorized service provider for repair. - ensure proper power supply and electrical connections. - regular maintenance and inspections to prevent such failures in the future. @topic:12


# # # issue - bed is not adjusting up or down # # # impact - patient discomfort - impaired patient care # # # possible causes - power issue : - loose plug - faulty power cord - power outage - mechanical issue : - broken motor - damaged gear assembly - obstruction in the bed frame # # # recommendations - check power supply : - ensure the power plug is securely connected to the outlet. - inspect the power cord for damage and replace if necessary. - inspect for mechanical issues : - listen for any unusual noises from the motor or gear assembly. - check for any obstructions or damage to the bed frame. - contact manufacturer for further assistance if necessary : - provide the equipment details ( band, model ) to the manufacturer for troubleshooting and repair. @topic:1


6 design capacity, 2 - 3 diagnosing problems, 4 - 3 diagnostics audio testing, 4 - 31 component usage, tracking, 4 - 35 display testing, 4 - 33 keys test, 4 - 32 recorder testing, 4 - 34 self test, 4 - 30 serial numbers, resetting, 5 - 25 system, running, 4 - 29 display, testing, 4 - 33 disposal, battery, 5 - 3 documentation, monitor, 1 - 2 e error codes description, 4 - 12 service, 4 - 12 error log clearing, 4 - 37 exporting, 4 - 36 printing, 4 - 36 viewing, 4 - 36 errors, tracking, 4 - 35 exploded views, replacement parts, 6 - 3 exporting, error log, 4 - 36 f faceplate, recorder, removing, 5 - 7 flash drive, recommended, 3 - 7 frequency maintenance, 2 - 1 performance tests, 3 - 2 front panel assembly, theory of operation, a - 5 front panel, removing, 5 - 11 full capacity, 2 - 3 fuse, replacing, 5 - 18 index - 1 suresigns vs2 * and vsi service guide g guidelines, testing and inspection, 3 - 1 h hard shutdown, performing, 5 - 2 hardware version, viewing, 4 - 2 i information, system, viewing, 4 - 2 intended audience, 1 - 1 k keys test, running, 4 - 32 l lan ip address, viewing, 4 - 2 lan mac address, viewing, 4 - 2 language in use, viewing, 4 - 2 lcd removing, 5 - 28 usage, tracking, 4 - 35 leds testing, 4 - 34 troubleshooting, 4 - 6 m main board removing, 5 - 23 theory of operation, a - 3 maintaining the battery, 2 - 2 maintenance menu, accessing, 4 - 29 maintenance, routine checks, 2 - 1 max error, 2 - 3 @topic:19


# # # issue : bed not locking in the correct position # # # impact : patient safety hazard # # # possible causes : - loose or missing screws - worn or damaged locking mechanism - electrical fault # # # recommendations : - check and tighten all screws - inspect the locking mechanism for damage or wear - check the electrical connections for continuity and proper function @topic:6


# # # issue : patient transport monitor leaking cuff # # # impact : - incorrect blood pressure measurements - patient discomfort - interruption of patient care # # # possible causes : - torn or damaged cuff bladder - loose connections between the cuff and the monitor - malfunctioning pressure transducer # # # recommendations : - inspect the cuff for any visible damage or tears. - check the connections between the cuff and the monitor to ensure they are secure. - calibrate the monitor to ensure the pressure transducer is functioning correctly. - replace the cuff if it is damaged or leaking. @topic:3


* * issue : * * knee motor adjustment failure * * impact : * * the bed cannot be adjusted to the desired height or position, which can be inconvenient or uncomfortable for patients. * * possible causes : * * * * * defective motor : * * the knee motor may be faulty, preventing it from adjusting properly. * * * electrical fault : * * there may be an electrical problem in the motor or its wiring, causing it to malfunction. * * * mechanical obstruction : * * something may be blocking the motor or its gears, preventing it from moving freely. * * * software error : * * the software controlling the motor may be malfunctioning, causing it to receive incorrect instructions. * * * incorrect installation : * * the motor may not have been installed correctly, resulting in misalignment or interference with other components. * * recommendations : * * * * * inspect the motor : * * check the motor for visible signs of damage or defects. * * * check the electrical connections : * * ensure that the motor is connected properly and that there are no loose wires or connectors. * * * inspect for obstructions : * * remove any objects or debris that may be blocking the motor or its gears. * * * update or reset the software : * * if necessary, update the software or reset it to factory settings. * * * reinstall the motor : * * if all else fails, the motor may need to be reinstalled or replaced by a qualified technician. @topic:9


# # # issue - bp calibrate fails # # # impact - unable to accurately measure blood pressure # # # possible causes - faulty bp cuff - faulty bp cuff connector - air leak in bp cuff or connector # # # recommendations - replace bp cuff and connector - inspect bp cuff and connector for leaks - confirm proper calibration of bp monitor @topic:14


# # # issue the telemetry monitor is not working properly in department 6h. # # # impact the telemetry monitor is used to monitor patients'vital signs, including heart rate, blood pressure, and oxygen levels. if the telemetry monitor is not working properly, it could lead to delays in diagnosis and treatment, and could even put patients'lives at risk. # # # possible causes there are several possible causes for the telemetry monitor not working properly. these include : - the telemetry monitor is not properly connected to the patient. - the telemetry monitor is not properly configured. - the telemetry monitor is malfunctioning. # # # recommendations to troubleshoot the telemetry monitor, the following steps should be taken : 1. check the connections between the telemetry monitor and the patient. make sure that the cables are securely connected and that there are no loose wires. 2. check the configuration of the telemetry monitor. make sure that the monitor is set to the correct settings for the patient's age, weight, and height. 3. if the telemetry monitor is still not working properly, it may be necessary to replace the monitor. @topic:16


# # # issue - open not start # # # impact - patients cannot be monitored for ecg and vital signs. # # # possible causes - the battery terminals are damaged. - the battery is not properly connected. - there is a loose connection in the power supply. - the power switch is not turned on. # # # recommendations - check the battery terminals and make sure they are not damaged. - make sure the battery is properly connected. - check the power supply for any loose connections. - turn on the power switch. @topic:11


# # # issue - keyboard of the medical cart is not responsive ( g ) # # # impact - the medical cart cannot be used to enter patient information or access medical records. # # # possible causes - the batteries of the keyboard are dead. - the keyboard is not properly connected to the medical cart. - the keyboard is damaged. # # # recommendations - replace the batteries of the keyboard. - check the connection between the keyboard and the medical cart. - if the keyboard is damaged, it will need to be repaired or replaced. @topic:27


# # # issue - medical cart keeps rebooting intermittently. # # # impact - the medical cart is not available for use, which can delay patient care. # # # possible causes - software issue - hardware issue # # # recommendations - * * software issue : * * - update the software to the latest version. - check for and install any available software patches. - reset the software to its default settings. - * * hardware issue : * * - inspect the medical cart for any loose connections or damaged components. - replace any faulty components. - contact the manufacturer for further troubleshooting and repair. @topic:13


# # # issue cuff pressure monitoring system is not functioning because the cuff does not have a proper connection joint. # # # impact infant patients cannot have their blood pressure accurately measured using the device's bp cuff. # # # possible causes the cuff that is being used on the patient is not the correct size. the cuff is not properly attached to the device. the cuff is damaged. the cuff is not compatible with the device. # # # recommendations verify that the cuff that is being used is the correct size for the patient. check that the cuff is properly attached to the device. inspect the cuff for any damage. ensure that the cuff is compatible with the device. @topic:5


casters, raise the brake and steer bar to the " steer " position. • fter movinconnect the bed and accessory power cords, • connect the nurse call system communications cable. securing the power cable always correctly store the power cable. failure to follow this recommendation may result in damage to the cable by crushing and create the risk of electric shock. the power cable must be hooked in place before moving the bed. attachment with cable tie ad292a * * h'hillrom. removable frame ( ad270b ) * * the detachable tube helps to guide the bed when transferring. decontamination, maintenance decontaminating the bed ensure that the bed cannot move. • lock out all electrical functions. disconnect the bed and stow the power cable ( see " securing the power cable ". check that all plugs are well connected ( control unit, electric motors on the power • never clean the bed by pouring water on it, nor with high - pressure hoses nor in tunnel washes. never use water at a temperature of more than 60°c. avoid excess water on the connectors. • refer to the recommendations of the cleaning product manufacturer. • thoroughly dry before reusing. • in order to guarantee the performances of the bed, all components that are removed for cleaning purposes must be returned to exactly the same place. • failure to implement one or more of these recommendations may lead to damage or deterioration, preventing use of the bed and rendering the warranty void. recommendations personnel must be trained to perform appropriate cleaning and disinfection. the instructor must carefully read the instructions and follow them while the trainee is attending the course. the trainee must : take all the time needed to read the instructions and ask questions. • clean and disinfect the product under the instructor's supervision. • during and / or after this process, the instructor must @topic:9


battery and the ac / dc power module can power the system through several dc / dc converters that provide various voltage outputs required by various chips. isolation transformers provide the power to the front ends floating area. speaker the speaker is driven by an amplifier and audio codec that interfaces to the microprocessor. the speaker provides the monitor ’ s audio during alarm conditions. the microprocessor outputs different audio tones specified by different alarm priorities and conditions. theory of operation a - 4 suresigns vs2 * and vsi service guide block diagram components nurse call contacts the nurse call alarm contacts are available through a 3. 5 mm, miniature insulated phone jack on the back of the monitor. the alarm contacts close ( open ) for any alarm condition detected by the monitor. the alarm contacts are the three switching signals associated with a single pole, double throw relay as follows : normally open contact ( no ) common contact normally closed contact ( nc ) contact rating of : 1a @ 25 vac, 60 vdc isolation of 1. 5 kv front panel assembly the front panel assembly contains these elements : lcd — 43 " wvga display — 480 pixels x 272 lines monitor keypad navigation wheel ( vs2 * only ) navigation wheel ( vs2 * only ) the navigation wheel is a rotating, push - switch wheel. the associated wheel circuitry generates a pulse when pressed and generates a digitally encoded pair of quadrature signals whose relative magnitudes and polarities represent the angular position of the wheel. these outputs are connected to the microprocessor where they are interpreted as required for the functions involved. successive angular positions determine the direction of wheel rotation. in addition to the functions performed in conjunction with the keypad, the wheel operates in conjunction with the display to select menus and lists of parameter variables. recorder ( vs2 * only ) the recorder contains the following features : digital thermal head array assembly printing @topic:10


# # # issue the reusable blood pressure cuff on the philips suresigns vs2 physiological vital signs monitor has deteriorated. # # # impact the monitor's blood pressure readings may be inaccurate or unreliable due to the deteriorated cuff. # # # possible causes 1. * * age / wear : * * reusable cuffs have a limited lifespan and can deteriorate over time due to repeated use and cleaning. 2. * * improper storage : * * cuffs should be stored in a cool, dry place to prevent damage from extreme temperatures or moisture. 3. * * overinflation : * * overinflating the cuff can put excessive strain on the material and lead to premature deterioration. 4. * * exposure to chemicals : * * cuffs can be damaged by exposure to harsh chemicals, such as cleaning agents or disinfectants. # # # recommendations 1. * * replace the cuff : * * the deteriorated cuff should be replaced with a new one to ensure accurate blood pressure readings. 2. * * inspect cuffs regularly : * * biomedical engineers should inspect reusable cuffs regularly for signs of wear or damage. 3. * * follow proper storage guidelines : * * cuffs should be stored in a clean, dry environment according to the manufacturer's instructions. 4. * * avoid overinflation : * * ensure that the cuff is inflated to the correct pressure range specified by the manufacturer. 5. * * protect from chemicals : * * cuffs should be handled with care and protected from exposure to harsh chemicals. @topic:5


# # # issue - motor failure # # # impact - inability to adjust bed height # # # possible causes - electrical fault in motor - mechanical failure in motor - damage to motor housing # # # recommendations - replace motor - inspect electrical connections - inspect motor housing for damage - test bed functionality after repairs @topic:9


# # # issue - battery depletion # # # impact - inability to use the nibp monitor due to lack of power # # # possible causes - battery degradation over time - incorrect battery installation - faulty battery charger # # # recommendations - replace the battery with a new one - ensure proper battery installation according to the manufacturer's instructions - test the battery charger to rule out any potential issues @topic:11


# # # issue - cuff bp has deteriorated due to age and use, causing the velcro to become ineffective. # # # impact - the device is unable to accurately measure blood pressure. # # # possible causes - prolonged use and wear - and - tear - exposure to extreme temperatures or humidity - improper storage or handling # # # recommendations - replace the cuff bp with a new one. - regularly inspect and maintain the device to prevent similar issues. - ensure proper storage and handling of the device and its accessories. @topic:3


# # # issue the issue is that the bed is not adjusting. # # # impact the patient may not be able to be positioned correctly for treatment. # # # possible causes * the remote is not working. * the bed is not connected to power. * the bed motor is not working. * the bed frame is damaged. # # # recommendations * check the remote to make sure it is working and that the batteries are not dead. * make sure the bed is plugged into a power outlet and that the power cord is not damaged. * check the bed motor to make sure it is working. * inspect the bed frame for any damage. @topic:1


# # # issue - spo2 sensor cable malfunction # # # impact - inability to accurately monitor patient's oxygen saturation levels # # # possible causes - damaged or loose spo2 sensor cable - faulty spo2 sensor - loose connection between the spo2 sensor and the monitor # # # recommendations - * * check the spo2 sensor cable for damage or loose connections. * * if the cable is damaged, it should be replaced. - * * inspect the spo2 sensor for any damage. * * if the sensor is damaged, it should be replaced. - * * ensure that the spo2 sensor is properly connected to the monitor. * * the connection should be secure and free of any debris. - * * if the problem persists, contact the manufacturer for further assistance. * * @topic:4


# # # issue the philips efficia cm10 monitor failed to turn on after being examined by the biomedical engineer ( bme ). # # # impact the failure of the monitor could have a significant impact on patient care, as it is used to monitor vital signs such as heart rate, respiration, and blood pressure. without a functioning monitor, healthcare professionals may not be able to accurately assess a patient's condition and provide appropriate treatment. # # # possible causes - * * power supply issues : * * the monitor may not be receiving power from the outlet or the power cord may be damaged. - * * battery issues : * * the battery may be dead or not making proper contact with the monitor. - * * hardware malfunction : * * there may be a problem with the monitor's internal circuitry or components, such as the motherboard or power supply. - * * software issues : * * the monitor's software may be corrupted or malfunctioning, preventing it from booting up properly. # # # recommendations the bme should perform the following steps to troubleshoot and fix the issue : 1. * * check the power supply : * * ensure that the monitor is plugged into a working outlet and that the power cord is securely connected. 2. * * replace the battery : * * if the monitor is using a battery, replace it with a new one to ensure that it has sufficient power. 3. * * inspect the hardware : * * visually inspect the monitor for any signs of damage, such as loose connections or burnt components. if any damage is found, the monitor may need to be repaired or replaced. 4. * * reinstall the software : * * if the monitor's software is malfunctioning, it may need to be reinstalled. the bme should contact the manufacturer for instructions on how to do this. @topic:2


# # # issue : - equipment malfunction due to deteriorated lead and spo2 cables. # # # impact : - interruption in patient monitoring, leading to potential missed critical events. - reduced accuracy and reliability of physiological data. # # # possible causes : - * * normal wear and tear : * * prolonged use and aging of cables can cause wear on insulation, connectors, and other components. - * * environmental factors : * * exposure to moisture, extreme temperatures, or harsh cleaning agents can accelerate cable deterioration. - * * mishandling or improper storage : * * dropping, bending excessively, or storing cables without proper protection can cause damage. - * * defective cables : * * manufacturing defects or material imperfections can lead to premature cable failure. # # # recommendations : - * * regular inspection and maintenance : * * implement a schedule for regular inspection of cables to identify signs of deterioration and perform preventive maintenance. - * * proper handling and storage : * * train staff on proper handling and storage techniques to minimize cable damage. - * * use high - quality cables : * * invest in cables that meet industry standards for durability and resistance to environmental factors. - * * monitor usage and record history : * * track cable usage and keep records of any incidents or repairs to identify patterns and improve maintenance practices. - * * train staff on signs of cable deterioration : * * educate staff on the signs of cable deterioration and how to report them promptly. @topic:20


818 ( 134 ) nbp equip malfunc ( 134 ) crc mismatch in error log data. shut down ( see “ shutting down the monitor ” on page 5 - 2 ) and restart the monitor. if the problem persists, replace the main board. see “ removing the main board ” on page 5 - 31. 818 ( 135 ) nbp equip malfunc ( 135 ) crc mismatch in inflation cycle count and runtime data. shut down ( see “ shutting down the monitor ” on page 5 - 2 ) and restart the monitor. if the problem persists, replace the main board. see “ removing the main board ” on page 5 - 31. 818 ( 136 ) nbp equip malfunc ( 136 ) the pneumatic safety switch malfunctioned. shut down ( see “ shutting down the monitor ” on page 5 - 2 ) and restart the monitor. if the problem persists, replace the main board. see “ removing the main board ” on page 5 - 31. if the problem persists, replace the nbp module. see “ removing the nbp module ” on page 5 - 36. 818 ( 137 ) nbp equip malfunc ( 137 ) valve 2 is defective or blocked. shut down ( see “ shutting down the monitor ” on page 5 - 2 ) and restart the monitor. if the problem persists, replace the nbp module. see “ removing the nbp module ” on page 5 - 36. if the problem persists, replace the main board. see “ removing the main board ” on page 5 - 31. 818 ( 141 ) nbp equip malfunc ( 141 ) inflation time for selected patient type exceeded. verify that the appropriate cuff size is being used for the selected patient @topic:20


# # # issue - bed lock malfunction # # # impact - patient safety risk due to bed instability - difficulty in positioning patient for procedures or comfort - inconvenience for healthcare staff # # # possible causes - broken lock component - malfunctioning button - damage to electrical wiring # # # recommendations - * * broken lock component : * * - replace the broken lock with a new one. - ensure the new lock is compatible with the bed model. - * * malfunctioning button : * * - inspect the button for physical damage or loose connections. - replace the button if necessary. - * * damage to electrical wiring : * * - check for any visible damage to the electrical wiring connecting the button to the lock. - repair or replace damaged wiring. - * * additional recommendations : * * - perform regular maintenance checks on the bed to identify potential issues early. - train healthcare staff on proper use and handling of the bed. - document the repair process and any observations made during troubleshooting. @topic:6


* * issue : * * ups failure * * impact : * * intermittent power supply to medical device, potentially leading to data loss or device malfunction. * * possible causes : * * * * * battery failure : * * batteries may have reached the end of their lifespan or may have been improperly maintained. * * * power supply failure : * * the ups may have experienced a power outage or surge, causing it to malfunction. * * * component failure : * * internal components within the ups, such as capacitors or transistors, may have failed. * * * software glitch : * * a software issue may have caused the ups to malfunction. * * recommendations : * * * * * inspect and replace batteries : * * regularly inspect batteries for signs of damage or end of life, and replace them promptly as needed. * * * test ups regularly : * * conduct regular power failure simulations to ensure the ups is functioning properly. * * * protect against power outages : * * provide backup power sources, such as generators or battery backups, to minimize the impact of power outages. * * * update software : * * ensure the ups software is up - to - date to address potential glitches. * * * troubleshoot : * * if the ups is malfunctioning, consult the manufacturer's instructions or contact a qualified technician to diagnose and repair the issue. @topic:11


# # # issue - remote malfunction # # # impact - inability to adjust bed settings # # # possible causes - faulty remote control unit - damaged wiring or connections - software malfunction # # # recommendations - replace the remote control unit - inspect and repair any damaged wiring or connections - update the software, if necessary @topic:7


# # # issue * * cuff leak * * # # # impact bp measurement is inaccurate or impossible # # # possible causes - cuff damage - loose or неправильно подключенныи манжет - defective valve # # # recommendations - * * inspect the cuff for damage. * * if there are any holes, tears, or other damage, the cuff will need to be replaced. - * * check the cuff connection. * * the cuff should be securely connected to the monitor and to the patient's arm. if the connection is loose, the cuff may not be able to inflate properly and may leak air. - * * test the valve. * * the valve should be able to hold air when the cuff is inflated. if the valve is defective, the cuff may leak air. # # # additional notes - if the cuff is leaking air, it will not be able to inflate properly. this will result in an inaccurate bp measurement. - if the cuff is not properly connected, it may not be able to inflate at all. - if the valve is defective, the cuff will not be able to hold air and will deflate quickly. @topic:3


# # # issue - the bed's side rails cannot be raised. # # # impact - the patient may fall out of bed. # # # possible causes - the hydraulic pump is not working properly. - the hydraulic lines are clogged. - the side rails are not properly aligned. - the side rails are not properly lubricated. # # # recommendations - check the hydraulic pump and replace it if necessary. - flush the hydraulic lines and replace the hydraulic fluid. - align the side rails and tighten the bolts. - lubricate the side rails. @topic:6


* * issue : * * medical device generated a continuous alarm line. * * possible causes : * * * * * faulty sensor : * * the sensor responsible for monitoring exhaled carbon dioxide may be malfunctioning, leading to erroneous readings. * * * loose connections : * * the cables or connections between the sensor and the device may be loose or damaged, causing intermittent data transmission and triggering alarms. * * * systemic error : * * a software or hardware glitch within the device itself could lead to continuous alarm generation, even in the absence of an actual issue. * * * wrong settings : * * incorrect configuration of the alarm parameters or thresholds could cause the device to generate false alarms. * * * environmental factors : * * extreme temperature or humidity levels, power fluctuations, or electromagnetic interference can affect the device's performance and trigger alarms. * * recommendations : * * * * * check the sensor : * * inspect the sensor for any visible damage or contamination. clean or replace the sensor as necessary. * * * inspect the connections : * * ensure all cables and connections between the sensor and the device are secure and intact. * * * restart the device : * * power cycle the device to potentially resolve any software or hardware issues. * * * review the alarm settings : * * verify that the alarm parameters and thresholds are set appropriately. adjust as needed to reduce false alarms. * * * check environmental conditions : * * ensure the device is operating within the specified temperature, humidity, and electromagnetic compatibility specifications. * * * consult with the manufacturer : * * contact the manufacturer for technical support or replacement parts if the issue persists after troubleshooting. @topic:20


# # # issue - secondary lead not displaying v lead waveform. # # # impact - inability to monitor the patient's cardiac activity accurately. # # # possible causes - faulty secondary lead. - loose connection between the secondary lead and the monitor. - incorrect settings on the monitor. - damage to the monitor's input circuitry. # # # recommendations - inspect the secondary lead for any visible damage. - ensure that the secondary lead is securely connected to the monitor. - verify the monitor's settings and adjust as necessary. - contact the manufacturer for further troubleshooting and repair if necessary. @topic:18


# # # issue bp cup failure # # # impact - inaccurate blood pressure readings - delays in treatment - patient discomfort # # # possible causes - physical damage to the cup - electrical malfunction - software error # # # recommendations - replace the bp cup with a new one - check the electrical connections - update the software if necessary @topic:14


# # # issue - low blood pressure ( bp ) reading # # # impact - inaccurate monitoring of vital signs - potential for misdiagnosis and inappropriate treatment # # # possible causes - faulty bp sensor - loose or damaged cable connections - software issues - user error ( e. g., incorrect cuff size or placement ) # # # recommendations - check and replace the bp sensor if necessary - inspect and tighten all cable connections - update software if required - train users on proper bp measurement techniques - ensure the cuff is the correct size for the patient and is positioned correctly on the arm @topic:14


# # # issue - battery failed to stimulate. # # # impact - patient monitoring may be interrupted or compromised. - increased risk of patient harm. - wasted time and resources. # # # possible causes - defective battery. - battery not compatible with the device. - battery not properly installed. - battery depleted. - battery charger malfunction. - device malfunction. - software issue. # # # recommendations - * * verify that the device is compatible with the battery. * * - * * check that the battery is properly installed. * * - * * replace the battery with a new one. * * - * * check the battery charger for any damage or malfunction. * * - * * test the device with a known - good battery. * * - * * contact the device manufacturer or a biomedical engineer for further troubleshooting. * * - * * update the device software to the latest version. * * @topic:11


# # # issue : sphygmomanometer cuff failure # # # impact : - inaccurate blood pressure readings, potentially leading to improper patient care. # # # possible causes : - * * wear and tear * * : the cuff is a movable component that is subjected to repeated inflation and deflation, which can weaken the material over time. - * * exposure to heat or moisture * * : extremes in temperature and moisture can damage the cuff material, making it less flexible and likely to fail. - * * improper handling * * : rough handling or incorrect storage can also contribute to cuff damage. # # # recommendations : - * * regular inspection * * : periodically inspect the cuff for signs of wear, such as cracks, tears, or fraying. - * * careful storage * * : store the cuff in a cool, dry place when not in use. - * * proper handling * * : handle the cuff gently, avoiding overinflation or rough handling. - * * regular replacement * * : replace the cuff according to the manufacturer's recommendations or when signs of deterioration are observed. @topic:3


# # # issue - the electric ob / gyn bed is not fully functional after being moved from 3divf to 5divf. # # # impact - the bed is essential for patient care and its malfunction can disrupt patient flow and safety. # # # possible causes - loose connections or damaged components - software issues or incorrect configuration - electrical power issues - mishandling or improper storage during the move # # # recommendations - inspect the bed thoroughly for loose connections and damaged components. - check the software settings and ensure proper configuration. - verify the electrical power supply and ensure it meets the bed's requirements. - review the handling and storage procedures to prevent future damage during transportation. - consult the manufacturer's guidelines for maintenance and troubleshooting. - test the bed thoroughly before returning it to patient use to ensure it is fully functional and safe. @topic:9


# # # issue the ekg cable is faulty. # # # impact the patient's ekg cannot be accurately monitored. # # # possible causes * the ekg cable is damaged. * the ekg cable is not properly connected to the monitor. * the ekg cable is not compatible with the monitor. # # # recommendations * replace the ekg cable. * ensure that the ekg cable is properly connected to the monitor. * verify that the ekg cable is compatible with the monitor. @topic:18


# # # issue battery failure in a philips intellivue mp40 ( m8003a ) monitor, patient transport device. # # # impact the device may not be able to function properly due to insufficient power supply. this could potentially lead to inaccurate readings or a loss of monitoring capabilities. # # # possible causes - battery has reached the end of its life cycle and needs to be replaced. - battery was not properly charged or maintained. - battery was damaged or defective. # # # recommendations - replace the battery with a new one. - ensure that the battery is charged and maintained according to the manufacturer's instructions. - inspect the battery for any signs of damage or defects. - schedule regular maintenance checks to ensure that the device is functioning properly and that the battery is in good condition. @topic:11


# # # issue : - cuff bp is defective. # # # impact : - the patient's blood pressure cannot be accurately measured. - the device may provide inaccurate readings. - the patient may not receive the appropriate medical care. # # # possible causes : - the cuff is old and has deteriorated. - the cuff has been damaged. - the cuff is not the correct size for the patient. - the cuff is not properly inflated. # # # recommendations : - replace the cuff with a new one. - ensure that the cuff is the correct size for the patient. - properly inflate the cuff. - regularly inspect the cuff for damage. @topic:3


# # # issue - the cause of failure is not clear. # # # impact - the technician may not be able to fix the device. - the patient may not be able to use the device. # # # possible causes - the technician may have misinterpreted the cause of failure. - the device may have multiple causes of failure. - the device may have a latent defect. # # # recommendations - the technician should re - evaluate the device to determine the cause of failure. - the technician should consult with a more experienced technician or engineer for assistance. - the device should be replaced if it cannot be repaired. @topic:12


to users on the proper use of the ear temperature probe to minimize operator error. @topic:20


# # # issue the cause of failure is that the spo2 value is not increasing. # # # impact this issue can impact the patient's care by providing inaccurate or incomplete information about the patient's vital signs. # # # possible causes the possible causes of this issue could include : - the spo2 sensor is not properly connected. - the spo2 sensor is defective. - the patient is not wearing the spo2 sensor correctly. - the patient has a low spo2 level. # # # recommendations the following recommendations can be made to address this issue : - check the connection between the spo2 sensor and the monitor. - replace the spo2 sensor. - ensure that the patient is wearing the spo2 sensor correctly. - monitor the patient's spo2 level closely. @topic:4


# # # issue : - battery failure # # # impact : - the physiologic monitor may not be able to function properly or may shut down if the battery is not replaced. # # # possible causes : - battery is old and needs to be replaced. - the battery is not making good contact with the monitor. - there is a problem with the monitor's charging system. # # # recommendations : - replace the battery with a new one. - clean the battery contacts on the monitor and the battery. - check the monitor's charging system for any problems. - if the problem persists, contact the manufacturer for further troubleshooting. @topic:11


* * issue : * * broken electric bed * * impact : * * patients cannot use the bed for treatment or recovery * * possible causes : * * - normal wear and tear - misuse by patients or staff - improper maintenance - manufacturer defects - power surges or outages * * recommendations : * * - regularly inspect and maintain the bed according to the manufacturer's instructions - train staff on proper use and maintenance of the bed - replace any worn or damaged parts promptly - install surge protectors to prevent damage from power surges - work with the manufacturer to identify and resolve any design or manufacturing issues @topic:9


# # # issue - remote control for the electric hospital bed is broken. # # # impact - patients may not be able to adjust their bed position, which can lead to discomfort or pain. - nurses and other healthcare providers may have to manually adjust the bed, which can be time - consuming and inefficient. # # # possible causes - the remote control may have been dropped or damaged accidentally. - the remote control may have been exposed to water or other liquids. - the batteries in the remote control may be dead. - the remote control may have a loose connection or a faulty circuit board. # # # recommendations - replace the broken remote control with a new one. - check the batteries in the remote control and replace them if necessary. - inspect the remote control for any loose connections or damage to the circuit board. - if the remote control cannot be repaired, contact the manufacturer for assistance. @topic:7


# # # issue - motor making excessive noise # # # impact - unable to adjust bed height # # # possible causes - motor seizure # # # recommendations - replace motor - test functionality @topic:9


# # # issue - equipment : telemetric electrocardiography monitor - brand : philips - model : intellivue mx40 - description : the wifi receiver on the device was not functioning. # # # impact - the device could not communicate with the central monitor. # # # possible causes - temporary power outage due to electrical cabinet maintenance. # # # recommendations - ensure that the electrical cabinet supplying power to the device is functioning properly. @topic:16


for use for further details. units and care groups can be configured at the information center. the available sizes for units and care groups are defined by the information center. monitors can be added to or removed from an existing care group at the monitor by assigning a new caregiver. 126 5 managing patients and equipment care groups in piic there are two main types of care groups : standard care group ( bed - based care group ) - up to 12 patients monitored by up to • 4 information centers unit unit group ( unit - based care group ) - for a complete unit with up to 64 patients monitored by care groups are configured at the information center. monitors can be added to or removed from an existing care group at the monitor. there is a third care group which can be assigned locally at the bedside. this is the my central care group which includes all beds ( up to 16 ) from the information center your bed is connected to. this care group setting is typically used in facilities with only one information center. the selection of beds is automatic and cannot be changed. the care group monitors'and equipment status is shown in symbol form in the alarm status overview bar. flashing symbols indicate active alarms, symbols that are not flashing indicate alarms that have been acknowledged. selecting a bed symbol calls up the window for that bed or a setup window to select it. because multiple equipment can be used for one patient, it can occur that the alarm status is different at different devices. the alarm status will then be consolidated and the highest priority alarm information will be displayed. the alarm status overview bar must be configured to display on the monitor screen. if it is not visible on your monitor, select a screen which has been configured to show the bar. some screens may show the care group monitors grouped together in a tabular format on the right hand side of the screen. piic ix depending on the information center configuration, the alarm status @topic:10


* * issue : * * remote control malfunction on huntleigh enterprise 5000 electric bed * * impact : * * inability to adjust bed height, compromising patient care and comfort. * * possible causes : * * * * * remote control hardware failure : * * malfunctioning buttons or circuitry within the remote control itself. * * * communication issues : * * poor signal or interference between the remote control and the bed's receiver. * * * power supply issues : * * insufficient battery power or faulty power supply to the remote control. * * * software malfunction : * * bugs or errors in the software controlling the remote control's functionality. * * * damage to the bed's receiver : * * broken or disconnected wires or sensors that prevent the bed from responding to remote commands. * * recommendations : * * * * * inspect the remote control : * * check for any visible damage or signs of wear and tear. test the buttons and ensure they are functioning properly. * * * verify communication : * * ensure there are no obstructions between the remote control and the bed's receiver. try using the remote control in different areas to rule out interference. * * * check power supply : * * replace the batteries in the remote control and verify that the power adaptor is securely connected. * * * restart the bed : * * unplug the bed from the power outlet for 10 seconds and then plug it back in. this may reset the software and resolve any glitches. * * * consult the manufacturer : * * if the above steps do not rectify the issue, contact the manufacturer for further troubleshooting assistance or to schedule a repair. * * * document the failure : * * accurately record the date, time, and details of the failure, including any error messages or observations made during troubleshooting. @topic:7


# # # issue - equipment : electric bed for use in obstetrics and gynecology - manufacturer : huntleigh - model : gynae 1 - description : loose headboard bolt # # # impact - inability to use the bed safely and effectively - potential for injury to patients or staff # # # possible causes - bolt was not tightened properly during assembly - bolt was loosened over time due to wear and tear - bolt was damaged # # # recommendations - inspect the bed regularly and tighten any loose bolts. - replace any damaged bolts. - train staff on the proper assembly and maintenance of the bed. @topic:6


# # # issue - battery not charging # # # impact - inability to use the device for extended periods of time - reduced patient safety and comfort # # # possible causes - faulty battery - faulty charger - problem with the battery charging circuitry in the device - firmware or software issue # # # recommendations - replace the battery - replace the charger - contact the manufacturer for repair or replacement of the device - update the firmware or software if an update is available @topic:11


# # # issue arthroscope system - monitor lcd is not turning on. # # # impact the surgical team is unable to use the arthroscope system to perform the surgery. this could delay or cancel the surgery, which could have a negative impact on the patient's outcome. # # # possible causes - the power supply is not connected or is faulty. - the lcd screen is damaged. - the motherboard is damaged. - there is a software issue. # # # recommendations - check the power supply and connections. - inspect the lcd screen for any damage. - contact the manufacturer for technical support. - replace the faulty component. @topic:2


# # # issue - modul has a problem. # # # impact - device may not function properly. # # # possible causes - manufacturing defect - wear and tear - improper use - power surge # # # recommendations - replace the modul. - inspect the device for any other damage. - train staff on proper use of the device. - install a surge protector. @topic:12