# # # issue : equipment is not charging # # # impact : device cannot be used as intended # # # possible causes : 1. loose or disconnected ac power cord connection 2. faulty ac power cord 3. malfunctioning charging port on the device 4. internal electrical component malfunction # # # recommendations : - verify that the ac power cord is firmly connected to both the power outlet and the device. - inspect the ac power cord for any damage or loose connections. replace if necessary. - examine the charging port on the device for any loose connections or debris. clean if necessary. - contact the manufacturer or a qualified biomedical engineer if the issue persists after trying the above steps. @topic:11


- 12. 2 disconnect the antenna cable from the wireless board. 3 with a 12 mm wrench, remove the washer and locking nut from the antenna and then remove the antenna from the front panel. repairing the monitor 5 - 52 suresigns vs4 service guide 6 replacement parts and assembly drawings spare parts the following table lists the suresigns vs4 monitor spare parts and their philips part numbers. see figure 6 - 1 through figure 6 - 3 for the item numbers that correspond to each spare part in the list. note — for clarity, some screws and connector lines do not appear in the drawings. description part number figure item number 2d barcode scanner 4535 640 51331 6 - 2 32 antenna 4535 642 70421 6 - 1 37 battery ( lithium ion ) 9898 031 44631 6 - 2 24 battery connector board 4535 640 20271 6 - 2 12 cable — dc in 4535 640 20291 6 - 2 18 cable — lcd 4535 643 57381 6 - 1 5 cable — lcd signal - in 4535 640 53881 6 - 1 4 cable — navigation wheel board 4535 640 31241 6 - 1 8 cable — touchscreen interface 4535 643 57811 not shown cable — wireless board to navigation wheel board 4535 643 57401 6 - 1 39 cover without recorder 4535 642 09251 6 - 2 31 fan assembly 4535 640 24591 6 - 2 36 front end assembly ( nbp and spo > ) 4535 640 41441 6 - 2 16 front panel with touchscreen 4535 643 57411 6 - 1 1 front panel with touchscreen ( english labels ) 4535 643 57461 not shown lan ( i / o communications ) board 4535 640 41481 6 - 2 13 lcd assembly 45 @topic:19


# # # issue - the electrocardiogram ( ecg ) lead wire is damaged, resulting in exposed wires. # # # impact - the damaged lead wire may disrupt the proper transmission of ecg signals, leading to inaccurate or incomplete patient monitoring. - the exposed wires pose an electrical hazard to both patients and healthcare professionals. # # # possible causes - physical damage to the lead wire due to : - excessive bending or stretching - contact with sharp objects - exposure to excessive heat or moisture - electrical surge or power failure - manufacturing defect # # # recommendations - * * immediate actions : * * - disconnect the damaged lead wire from the patient and the monitor. - isolate the damaged lead wire in a safe location. - notify the healthcare team and the appropriate maintenance department. - * * troubleshooting and repair : * * - inspect the damaged lead wire to identify the source and extent of the damage. - if the damage is minor and can be repaired, follow the manufacturer's instructions for repair. - if the damage is significant, replace the entire lead wire. - * * preventive measures : * * - implement proper handling and storage guidelines for lead wires. - regularly inspect lead wires for signs of damage or wear. - ensure that lead wires are not subjected to excessive bending, stretching, or sharp objects. - protect lead wires from exposure to extreme heat or moisture. @topic:18


# # # issue - the vital signs monitor is not taking accurate blood pressure readings. # # # impact - this could lead to incorrect patient diagnosis and treatment. # # # possible causes * * 1. the blood pressure cuff is not the correct size for the patient. * * - the cuff should be snug, but not too tight. if the cuff is too small, it will not be able to inflate properly and will give an inaccurate reading. if the cuff is too large, it will not be able to create enough pressure to stop the blood flow, and will also give an inaccurate reading. * * 2. the blood pressure cuff is not placed correctly on the patient's arm. * * - the cuff should be placed on the upper arm, just above the elbow. the arrow on the cuff should be pointing towards the patient's heart. * * 3. the patient is not resting quietly during the blood pressure measurement. * * - the patient should be sitting or lying down, and should not be talking or moving around. # # # recommendations * * 1. check the size of the blood pressure cuff. * * - the cuff should be the correct size for the patient's arm. * * 2. place the blood pressure cuff correctly on the patient's arm. * * - the cuff should be placed on the upper arm, just above the elbow. the arrow on the cuff should be pointing towards the patient's heart. * * 3. ask the patient to rest quietly during the blood pressure measurement. * * - the patient should be sitting or lying down, and should not be talking or moving around. @topic:14


# # # cause of failure : insufficient power supply # # # # issue - the electric bed is not receiving power, resulting in failure to operate. # # # # impact - inability to use the electric bed, impacting patient care and hospital operations. # # # # possible causes - faulty power cord or plug : the power cord or plug may be damaged, loose, or disconnected, preventing the flow of electricity to the bed. - electrical outlet malfunction : the electrical outlet supplying power to the bed may be faulty, causing an interruption in the power supply. - loose connections within the bed : internal connections within the bed may have become loose, hindering the proper flow of electricity. - circuit breaker or fuse tripped : the circuit breaker or fuse protecting the electrical circuit of the bed may have tripped due to an electrical surge or overload. # # # # recommendations - check the power cord and plug connections, ensuring they are secure and undamaged. - inspect the electrical outlet and ensure it is functioning correctly. - open the bed's housing and inspect the internal connections for any loose wires or connectors. - reset the circuit breaker or replace the blown fuse if necessary. - if the problem persists, contact a qualified biomedical engineer for further troubleshooting and repair. @topic:9


# # # issue - the bed cannot be adjusted. # # # impact - the patient cannot be positioned comfortably or safely. - the bed cannot be used for its intended purpose. # # # possible causes - the bed's motor is not working. - the bed's electrical system is faulty. - the bed's mechanical system is jammed. - the bed's control panel is not functioning properly. # # # recommendations - check the bed's motor and electrical system for any faults. - check the bed's mechanical system for any jams. - check the bed's control panel for any faults. - if the bed cannot be repaired, it should be replaced. @topic:1


# # # issue the hospital bed is not able to be adjusted. # # # impact this issue could have a significant impact on patient care. if the bed cannot be adjusted, it could make it difficult for patients to get in and out of bed, which could lead to falls or other injuries. additionally, if the bed cannot be adjusted, it could make it difficult for nurses to provide care to patients, which could lead to delays in treatment or other problems. # # # possible causes there are several possible causes for this issue, including : - the bed is not plugged in. - the bed's power cord is damaged. - the bed's control panel is malfunctioning. - the bed's motor is malfunctioning. # # # recommendations to troubleshoot this issue, the following steps should be taken : 1. check to make sure that the bed is plugged in. 2. inspect the bed's power cord for any damage. 3. try resetting the bed's control panel. 4. if the above steps do not resolve the issue, the bed's motor may need to be replaced. @topic:1


# # # issue - monitor not linking with anesrecord system # # # impact - inability to monitor patient data and track vital signs # # # possible causes - loose lan cable - network connectivity issues - software compatibility issues - hardware malfunction # # # recommendations - ensure lan cable is securely connected - check network connectivity and ensure proper configuration - verify software compatibility between monitor and anesrecord system - contact biomedical engineer if issue persists @topic:16


* * issue : * * bed cannot be adjusted. * * impact : * * patient discomfort, inability to position the patient safely for medical procedures. * * possible causes : * * - motor failure - circuit board malfunction - power supply issue - mechanical obstruction * * recommendations : * * 1. check the power supply to the bed and ensure it is receiving adequate voltage. 2. inspect the motor and circuit board for any visible damage or loose connections. 3. check for any mechanical obstructions, such as bed linens or debris, that may be preventing the bed from adjusting. 4. if no obvious cause of failure is identified, further troubleshooting may be necessary, such as using a diagnostic tool or consulting with the manufacturer. 5. once the cause of failure is identified, the appropriate repair or replacement can be performed to restore the bed's functionality. @topic:1


# # # issue - broken ekg lead # # # impact - patient monitoring is disrupted - incorrect or incomplete patient data - potential for patient harm # # # possible causes - lead was damaged during use - lead was not properly stored or maintained - lead was inadvertently cut or torn # # # recommendations - replace the damaged lead with a new one - inspect all leads regularly for damage - store and maintain leads properly - train staff on proper handling of leads @topic:18


# # # issue - cause of failure : does not show values # # # impact - unable to monitor patient's vital signs - may lead to delayed or inaccurate diagnosis and treatment # # # possible causes - incompatible module - faulty monitor - loose connections # # # recommendations - ensure that the module is compatible with the monitor - replace the monitor if necessary - check and tighten connections @topic:22


# # # issue physiologic monitor cannot measure blood pressure. # # # impact unable to monitor patient's blood pressure, which is a vital sign essential for patient care. # # # possible causes - faulty connector on the monitor or patient cable - loose or disconnected cables - software malfunction - hardware malfunction # # # recommendations - inspect the connector on the monitor and patient cable for any damage or loose connections. - ensure that all cables are securely connected. - restart the monitor and check if the issue persists. - contact the manufacturer for technical support if the issue cannot be resolved with the above steps. @topic:15


# # # issue - oxygen saturation measurement failure due to faulty connector cable # # # impact - incorrect or unavailable oxygen saturation readings - delays in patient care and treatment decisions # # # possible causes - defective or damaged connector cable - misalignment or improper connection of the cable - corrosion or contamination of the connector pins - electrical interference or loose contacts # # # recommendations - replace the faulty connector cable with a new one - inspect the connector pins for damage or corrosion and clean as necessary - ensure proper alignment and secure connection of the cable - check for electrical interference or loose contacts and address any issues - perform regular maintenance and calibration of the equipment to prevent future issues @topic:4


# # # issue - equipment : bed, electric - band : huntleigh - model : contoura 360 - description : bme repaired the equipment and it is now working properly. - cause of failure : unable to adjust # # # impacts - the patient may not be able to get into or out of bed safely. - the patient may not be able to position themselves comfortably in bed. - the patient may be at risk of falls or other injuries. # # # possible causes - the adjustment mechanism may be broken or damaged. - the electrical components may be malfunctioning. - the bed may not be properly assembled. - the patient may not be able to operate the bed properly. # # # recommendations - inspect the adjustment mechanism for any damage. - test the electrical components to ensure they are functioning properly. - ensure the bed is properly assembled. - provide the patient with instructions on how to operate the bed. - if the problem cannot be resolved, contact a qualified technician for assistance. @topic:1


# # # issue the bed, electric is making a loud noise when adjusting its position. # # # impact the noise can be disruptive to patients and staff, and could potentially interfere with patient care. # # # possible causes * * * lack of lubrication : * * the bed may not have been lubricated properly, which could cause friction and noise. * * * loose or damaged parts : * * the bed may have loose or damaged parts, which could also cause noise. * * * improper assembly : * * the bed may not have been assembled properly, which could lead to noise. # # # recommendations * * * lubricate the bed : * * apply a lubricant to all moving parts of the bed to reduce friction and noise. * * * inspect and tighten parts : * * inspect the bed for any loose or damaged parts and tighten or replace them as needed. * * * reassemble the bed : * * if necessary, disassemble and reassemble the bed to ensure that all parts are properly aligned and tightened. @topic:9


* * issue * * : electric bed remote not working * * impact * * : patient cannot use bed * * possible causes * * : - loose remote control cord - loose or damaged electrical connections - faulty remote control - faulty bed motor * * recommendations * * : - check remote control cord and reconnect firmly - inspect electrical connections for loose or damaged wires ; reconnect or replace as necessary - replace remote control if it is faulty - contact vendor for assistance if bed motor is faulty @topic:7


# # # issue - equipment does not power on after replacing battery. # # # impact - patient monitoring is unavailable. # # # possible causes - battery is not properly installed. - battery is not making good contact with the terminals. - battery is damaged. - equipment is damaged. # # # recommendations - verify that the battery is properly installed. - inspect the battery terminals for damage or corrosion. - replace the battery. - if the issue persists, contact the manufacturer for repair. @topic:11


# # # issue medical cart program is unable to run normally. # # # impact the medical cart cannot be used properly, which can delay or affect patient care. # # # possible causes - the program files may be corrupted or missing. - the program may not be compatible with the operating system or other software on the medical cart. - there may be a problem with the hardware on the medical cart. - there may be a problem with the network connection. - the program may not be properly installed. # # # recommendations - check the program files to make sure they are not corrupted or missing. - make sure the program is compatible with the operating system and other software on the medical cart. - check the hardware on the medical cart to make sure it is working properly. - check the network connection to make sure it is stable. - reinstall the program. @topic:13


# # # issue : blank screen with frozen software # # # impact : interruption of medical care # # # possible causes : - faulty software update - hardware malfunction - corrupted data # # # recommendations : - restart the device - check for software updates - run diagnostics to identify hardware issues - restore from a backup if available - contact the manufacturer for support @topic:13


# # # issue - equipment : monitors, physiologic, electrocardiography, telemetric - band : philips - model : intellivue mx40 - description : bme found that the monitor could not be turned on because the brass terminals of the battery rails were broken. bme re - soldered the brass terminals. bme tested the device and it worked normally. # # # impact - the monitor could not be used to monitor patients'ecg. # # # possible causes - the brass terminals of the battery rails were broken due to : - normal wear and tear - accidental damage - manufacturing defect # # # recommendations - inspect the battery rails and terminals regularly for any damage. - replace the battery rails if they are damaged. - use a high - quality battery to avoid damage to the battery rails. @topic:11


# # # issue the range of the printed graph is too wide, making the graph difficult to read. # # # impact the user cannot read the graph properly, which can lead to misinterpretation of patient data. # # # possible causes - the range setting on the device is incorrect. - the device is not calibrated properly. - there is a problem with the sensor that is generating the graph data. # # # recommendations - check the range setting on the device and adjust it to the desired value. - calibrate the device according to the manufacturer's instructions. - inspect the sensor for any damage or contamination and clean it if necessary. - if the problem persists, contact the manufacturer for further assistance. @topic:20


* * issue : * * vital signs monitor displaying abnormal readings * * impact : * * incorrect or delayed diagnosis, improper treatment, patient safety risk * * possible causes : * * * * * loose cuff connection : * * the cuff connection may not have been securely attached, leading to inaccurate pressure readings. * * * damaged or leaking cuff : * * tears or holes in the cuff can cause air to escape, resulting in incorrect pressure measurements. * * * faulty tubing : * * kinks or obstructions in the tubing can affect pressure transmission, causing inaccurate readings. * * * transducer malfunction : * * the transducer, which converts the pressure signal into an electrical signal, may be malfunctioning, leading to distorted readings. * * * patient movement : * * movement during measurement can interfere with cuff placement and result in unreliable readings. * * recommendations : * * * * * check cuff connection : * * ensure the cuff is properly attached to the patient and the monitor. * * * inspect cuff : * * look for damage or leaks, and replace the cuff if necessary. * * * check tubing : * * inspect the tubing for kinks or obstructions, and replace it if necessary. * * * test transducer : * * perform a transducer calibration to ensure accurate function. * * * position patient correctly : * * instruct the patient to remain still and avoid movement during measurement. * * * consider other factors : * * assess the patient's condition and potential causes for abnormal readings, such as arrhythmias or hypertension. @topic:14


# # # issue bed's weighing function not working # # # impact patient's weight cannot be measured accurately, which may affect diagnosis and treatment. # # # possible causes - faulty weighing sensor - electrical connection issue - software glitch - battery failure # # # recommendations - inspect weighing sensor for damage or debris - check electrical connections for loose or damaged wires - restart the system to resolve any software glitches - replace batteries if necessary @topic:25


# # # issue - equipment : bed, electric - band : huntleigh - model : contoura 360 - cause of failure : being used as a temporary replacement during repairs # # # impact - reduced availability of the device for its intended purpose - potential delays in patient care - increased workload for staff # # # possible causes - lack of spare equipment - delays in repairs - inefficient maintenance scheduling # # # recommendations - maintain an adequate inventory of spare equipment to prevent shortages. - establish clear communication channels between maintenance staff and clinical departments to ensure timely repairs. - optimize maintenance scheduling to minimize equipment downtime. - explore alternative arrangements for providing patient care in the absence of critical equipment. - consider investing in equipment that is more durable and less prone to failure. @topic:12


# # # issue icp monitor is not turning on. # # # impact the inability to monitor intracranial pressure ( icp ) can lead to life - threatening complications, such as brain damage or death. # # # possible causes - power supply issues - hardware failure - software errors - user error # # # recommendations 1. * * check the power supply. * * ensure that the icp monitor is properly plugged into a working power outlet. if the power cord is damaged, it should be replaced. 2. * * inspect the hardware. * * look for any obvious signs of damage, such as loose connections or broken components. if any damage is found, the icp monitor should be taken out of service and repaired by a qualified technician. 3. * * update the software. * * if the icp monitor is running outdated software, it may be susceptible to bugs or glitches that could prevent it from turning on. check with the manufacturer for the latest software updates and install them as necessary. 4. * * check for user error. * * make sure that the icp monitor is being operated correctly. refer to the user manual for instructions on how to properly turn on the device. # # # additional notes if the icp monitor still does not turn on after following these recommendations, it should be taken out of service and sent to a qualified technician for repair. @topic:2


##ge sensor with an applicable spo2 tester. if the message persists eplace the module. demo mode active. 0 the spo2 parameter is in demo mode sensor not connected. check the spo2 sensor clear to retry. 041c00 connection check the sensor and cable connection. if the problem is still present, replace the spo2 sensor. if the problem is still present, replace the cable. if the problem is still present, verify the module functionality by replacing the sensor with an applicable spo2 tester. if the message persists replace the module. there is an spo2 raw queuspo2 rebooting. 041e00 a malfunction is detected. there very low re two possible causes of these overrun failures. first, the power provided to the board is out of specification. in this case, the failure may be cleared once the underlying cause is removed. second, the board has an actual hardware failure and recovery is not possible. it is recommended that the spo2 module be replaced, and if problem is still present replace the monitor's spo2 rebooting. 041f00 there is an spo2 hardware failure a malfunction is detected. there very low failures. first, the power provided to the board is out of specification. in this case, the failure may be cleared once the underlying cause is removed. second, the board has an actual hardware failure and recovery is not possible. it is recommended that the spo2 module be replaced, and if problem is still present replace the monitor's ain board. spo2 rebooting042000 a malfunction is detected. there very low are two possible causes of these there is an spo2 mcu failure failures. first, the power provided to the board is out of specification. in this case, the failure may be @topic:20


# # # issue - monitor is unable to measure patient's vital signs due to a faulty sensor extension cable. # # # impact - patient's vital signs cannot be monitored accurately. - this can lead to delayed diagnosis and treatment, which can be life - threatening. # # # possible causes - the sensor extension cable may be damaged due to wear and tear. - the cable may be crimped or kinked, which can prevent the signal from being transmitted properly. - the cable may be too long, which can cause signal loss. # # # recommendations - replace the faulty sensor extension cable with a new one. - inspect the cable regularly for damage and replace it if necessary. - use a cable that is the appropriate length for the application. - ensure that the cable is not crimped or kinked. @topic:22


# # # issue the basket of the vital signs monitor is too low. # # # impact the monitor may not be able to accurately measure the patient's vital signs. # # # possible causes - the basket may have been installed incorrectly. - the monitor may have been dropped or bumped. - the basket may be defective. # # # recommendations - check that the basket is installed correctly. - if the monitor has been dropped or bumped, have it inspected by a qualified technician. - if the basket is defective, replace it. @topic:15


* * issue : * * * remote control for electric bed is not functioning properly. * * impact : * * * patient may not be able to adjust the bed's position independently. * caregivers may have difficulty accessing the patient or providing care. * * possible causes : * * * * * remote control malfunction : * * the remote control may have a faulty electrical component or software issue. * * * battery depletion : * * the remote control's batteries may be low or dead. * * * connectivity issue : * * the remote control may not be properly paired with the bed or there may be signal interference. * * * bed receiver malfunction : * * the receiver in the bed that receives signals from the remote control may be defective. * * * physical damage : * * the remote control or bed receiver may have been damaged due to impact or exposure to liquids. * * recommendations : * * * * * check batteries : * * replace the batteries in the remote control. * * * reprogram remote control : * * if the remote control is not functioning properly after replacing the batteries, attempt to reprogram it according to the manufacturer's instructions. * * * inspect cables and connections : * * ensure that all cables and connections between the remote control, bed receiver, and bed are secure and undamaged. * * * test bed receiver : * * use a different remote control or a test signal to verify if the bed receiver is functioning properly. * * * contact manufacturer or biomedical engineer : * * if the issue persists, contact the manufacturer or a qualified biomedical engineer for further troubleshooting and repair. @topic:7


arrow mark over that column shows that the currently measured value is also in this range. these columns are displayed in the same color as the measurement data. the columns in the background show the cumulative percentage value : each of the foreground columns is added to the sum of those columns to the left of it. a question mark is displayed if less than two - thirds of the data are valid samples. the pop - up keys in the window can be used to change the measurement, the time period and the range. you can also print out a histogram report. the sp0 2 histograms can be trend histograms or realtime histograms with 1 second samples. for sp0 2 realtime histograms, the range cannot be changed. setting up trends trends are set up in configuration mode. you can make temporary changes to trends settings such as trend groups, priorities, or scales in monitoring mode. the general settings for all trends are under main setup - > trends. settings for a single segment in graphical trends or for the corresponding trend group can be made more easily in that segment menu. making segment settings the graphical trends window is divided into measurement segments, each of which has its own settings. 274 setting up trends 21 trends to enter the segment menu, ♦ select the left hand column of the segment, where the measurement label is displayed. expanded view to expand the segment to fill the graphical trends window, ♦ in the segment menu, select expand to have that segment enlarged to fill the window. in the expanded view, you can also highlight a measurement to make it more easily visible ( for example when multiple st trends are in one segment ). to highlight a measurement, ♦ in the segment menu, select highlight repeatedly until the highlight is on the required measurement. to return the segment to its original size, ♦ in the segment menu, select expand again. trend scales for segment @topic:8


# # # * * issue * * the printer on the central station physiological monitor is not printing evenly. # # # * * impact * * this could lead to inaccurate or incomplete medical records, which could have serious consequences for patient care. # # # * * possible causes * * * clogged or dirty ink roller * faulty print head * incorrect ink cartridge * software issue # # # * * recommendations * * * clean the ink roller * replace the print head * check the ink cartridge and replace it if necessary * update the software @topic:28


# # # issue the medical cart is not connecting to the hardware. # # # impact the medical cart is not able to be used, which can impact patient care. # # # possible causes * the connection between the medical cart and the hardware is not secure. * the hardware is not functioning properly. * there is a problem with the software on the medical cart. # # # recommendations * check the connection between the medical cart and the hardware. * if the connection is secure, check the hardware to make sure it is functioning properly. * if the hardware is functioning properly, check the software on the medical cart. @topic:13


# # # issue - remote control for adjusting bed height not working # # # impact - inability to adjust bed height, affecting patient comfort and safety # # # possible causes - faulty remote control - electrical issues with the bed - software glitches # # # recommendations - replace the remote control - inspect electrical connections and wiring of the bed - update software if available @topic:7


data window to display the pop - up keys that let you access the setup menu and make changes. alarms / inops from external devices the intellibridge module itself generates inops, but does not generate alarms. if the external device ’ s alarms are on, the module transmits these to the monitor. a message in the external device data window tells you either device alarms ignored or device alarms accepted or no alarms available. external device alarms status symbols precede some, but not all, measurement labels. the monitor is configured to accept external device alarms, but the alarms are switched off at the external device. alarms from external devices are : • all transmitted to the monitor. for all numerics configured in the setup menu, an alarm condition is announced at the monitor. for one or more measurements not configured in the setup menu, an alarm is announced as a text message for the highest priority alarm. priority is determined at the external device. refer to the documentation supplied with the intellibridge module for details of how alarms are announced on the monitor and the information center. 349 29 intellibridge ec10 module language conflict with external device drivers inops from the intellibridge module inop text inop condition!! /!!! no device data communication has been lost to a connected device when -!! a medium priority inop or patient alarm was present -!!! a high priority inop or patient alarm was present no device data communication has been lost to a connected device with no ( or only low priority ) inops / physical alarms. device unsupported device identification completed but no appropriate device driver installed device check conf. device identification completed but communication could not be established due to errors device check setup device identification completed but communication could not be established due to timeout language conflict with external device drivers you should avoid language conflicts between the intellibridge module device driver and the monitor. @topic:10


# # # issue - the cuff on the device is not inflating. # # # impact - the device cannot accurately measure blood pressure. # # # possible causes - the cuff is damaged. - the cuff is not properly connected to the device. - the air pump in the device is not working properly. - the pressure sensor in the device is not working properly. # # # recommendations - inspect the cuff for damage. - ensure that the cuff is properly connected to the device. - test the air pump in the device. - test the pressure sensor in the device. @topic:3


# # # issue - * * broken spo2 cable * * # # # impact - * * patient monitoring interruption * * # # # possible causes - * * normal wear and tear * * - * * accidental damage * * - * * defective cable * * # # # recommendations - * * replace the spo2 cable * * - * * inspect the cable regularly for any signs of damage * * - * * handle the cable with care to prevent accidental damage * * - * * consider using a cable management system to protect the cable from strain * * @topic:4


##min menu. for more information about using the system menu to configure the monitor, see the suresigns vs4 instructions for use or the suresigns vs4 installation and configuration guide. to access the system menu : e select the system button. the system menu appears. system menu recorder speed : waveform print : date format : display time : default patient type : monitor name : serial number : hardware id : software version : lan mac address : lan ip address : wlan mac address : wlan ip address : language : configuration : performance verification testing 3 - 4 suresigns vs4 service guide | — — rs ra 25. 0 mm / s ) ( save patient records ) @ 20 seconds : _ _ _ system admin ( _ mmiddiyyyy ) ( shutdown ( lock touch screen ) adult xn ( ustooo0001 \ usl0000001 5 - 00 - a6 a. 04. 52 00 - 09 - fb - 10 - 4b - 3c 0. 0. 0. 0 00 - 17 - 23 - a1 - 84 - b2 0. 0. 0. 0 english vs4 spo2 temp - t nbp - p recorder accessing the system admin menu accessing the system admin menu you can use the system admin menu to configure password - protected functions of the monitor, including demo mode, system diagnostics, and upgrading the software. for more information about using the system admin menu to configure the monitor, see the suresigns vs4 installation and configuration guide. to access the system admin menu : the system admin menu opens. system admin menu language : alarm tone : minimum alarm tone volume : alarm pause duration : default nbp interval : default continuous spo2 : default save spo2 to @topic:10


# # # issue - spo2 cable broken # # # impact - patient's vital signs cannot be monitored accurately. # # # possible causes - the cable may have been crimped or cut. - the cable may have been pulled too tightly. - the cable may have been exposed to excessive heat or moisture. # # # recommendations - replace the spo2 cable with a new one. - inspect the cable for any damage before use. - avoid pulling on the cable excessively. - protect the cable from heat and moisture. @topic:4


##4a front bezel ( c. o. / cco, c0 2, press, press / temp ), russian not shown m3014 - 44179 451261009721 m3014a front bezel ( c. o. / cco, c0 2, press, press / temp ), czech not shown m3014 - 44180 451261009731 m3014a front bezel ( c. o. / cco, c0 2, press, press / temp ), polish not shown m3015a # c06 sidestream co 2 with press / temp m3015 - 44131 453563332291 m3015a front bezel, english ( also for french, danish, traditional chinese and simplified chinese ) 1a m3015 - 44133 453563332301 m3015a front bezel, german not shown m3015 - 44134 453563332311 m3015a front bezel, dutch not shown m3015 - 44135 453563332321 m3015a front bezel, spanish not shown m3015 - 44136 453563332331 m3015a front bezel, italian not shown m3015 - 44137 453563332341 m3015a front bezel, norwegian not shown m3015 - 44138 453563332351 m3015a front bezel, swedish not shown m3015 - 44139 453563332361 m3015a front bezel, finnish not shown m3015 - 44140 453563332371 m3015a front bezel, japanese not shown m3015 - 44144 453563332381 m3015a front bezel, portuguese not @topic:19


# # # issue - inability to display patient vital signs on the patient transport monitor due to a faulty vga cable. # # # impact - compromised patient monitoring capabilities, potentially delaying or misinforming clinical decisions. # # # possible causes - damaged or loose vga cable connection. - incompatibility between the vga cable and the monitor. - malfunctioning vga port on the monitor. # # # recommendations - ensure the vga cable is securely connected to both the monitor and the patient transport equipment. - verify the compatibility of the vga cable with the monitor and the equipment. - if the issue persists, test the vga port on the monitor using another vga cable or device. - if the vga port is confirmed to be faulty, consider replacing the monitor. @topic:2


# # # issue - misplacement of equipment # # # impact - the equipment may not be available for use in its intended location, potentially affecting patient care. - wasted time and resources in moving the equipment back to its correct location. # # # possible causes - lack of clear instructions or communication regarding the equipment's intended destination. - misunderstanding or misinterpretation of the equipment's location. - inadequate labeling or identification of the equipment. # # # recommendations - establish clear communication channels to ensure that equipment is moved to the correct location. - provide detailed instructions and / or maps to guide the movement of equipment. - label and identify equipment clearly to prevent confusion. - train staff on proper equipment handling and movement protocols. - implement a system to track equipment movement and ensure its availability at the intended location. @topic:12


# # # issue : - central monitor display is not turning on. # # # impact : - inability to monitor patient's vital signs, such as heart rate, blood pressure, and oxygen levels. - delays in diagnosis and treatment. - increased risk of patient complications. # # # possible causes : - * * loose or disconnected power cable : * * ensure that the power cable is securely connected to both the monitor and the power outlet. - * * faulty power supply : * * inspect the power supply for any damage or loose connections. replace if necessary. - * * defective display panel : * * the display panel may have malfunctioned and needs to be replaced. - * * software glitch : * * a software malfunction can prevent the monitor from displaying properly. try restarting the monitor or updating the software. - * * hardware failure : * * internal components within the monitor may have failed, such as the motherboard or graphics card. # # # recommendations : - check the power cable and connections carefully. - unplug the monitor and wait for a few minutes before plugging it back in. - contact the manufacturer or a qualified technician for further diagnosis and repair if the issue persists. - regularly inspect and maintain medical equipment to prevent future failures. @topic:2


# # # issue - battery failure in a physiologic monitor. # # # impact - interruption of patient monitoring. - potential patient safety concerns if the monitor is not functioning properly. # # # possible causes - battery degradation over time. - improper charging or maintenance of the battery. - manufacturing defect in the battery. # # # recommendations - replace the battery with a new one of the same type and capacity. - ensure that the battery is properly charged and maintained according to the manufacturer's instructions. - regularly inspect the battery for signs of damage or degradation. - if the battery failure is recurring, consider investigating potential underlying issues with the monitor or the charging system. @topic:11


# # # issue - the bed automatically lowers itself even without the button being pressed. # # # impact - patient safety hazard - inconvenience to medical staff # # # possible causes - malfunctioning of the control panel - faulty wiring or connections - software glitch - loose or damaged components # # # recommendations 1. perform a thorough inspection of the control panel, wiring, and connections. 2. check for any loose or damaged components that may be causing the malfunction. 3. update the software if a newer version is available. 4. replace any faulty components as necessary. 5. conduct a functional test to ensure the bed is operating properly. 6. monitor the bed's performance over time to prevent recurrence of the issue. @topic:1


# # # issue - co2 call fail # # # impact - patient monitoring is compromised. # # # possible causes - connector not clean. - contect not properly cleaned. # # # recommendations - clean the connector. - clean the contect. @topic:4


# # # issue * physiologic monitor ( bedside ) philips suresigns vs3 is not functioning properly due to the absence of a probe o2 sat cable. # # # impact * without a probe o2 sat cable, the monitor cannot accurately measure a patient's oxygen saturation levels. * this can lead to delayed diagnosis and treatment of conditions such as hypoxia, which can be life - threatening. # # # possible causes * the probe o2 sat cable may have become damaged or disconnected. * the cable may not be compatible with the monitor. * the monitor may be malfunctioning. # # # recommendations * inspect the probe o2 sat cable for any damage or disconnections. * if the cable is damaged, replace it with a new one. * if the cable is not compatible with the monitor, obtain a compatible cable. * if the monitor is malfunctioning, contact the manufacturer for repair or replacement. @topic:4


# # # issue - central monitor ( hewlett packard, model : m3150a ) - battery failure # # # impact - inability to monitor patient vital signs accurately - potential for patient harm # # # possible causes - * * battery degradation : * * over time, batteries lose their capacity to hold a charge. this can be due to factors such as age, temperature, and usage patterns. - * * improper charging : * * not charging the battery properly can also lead to premature failure. for example, using the wrong charger or charging the battery for too long can damage the battery cells. - * * physical damage : * * if the battery is physically damaged, such as being dropped or punctured, this can also lead to failure. # # # recommendations - * * regular maintenance : * * regularly check the battery's charge capacity and replace it as needed. - * * proper charging : * * use the correct charger and follow the recommended charging procedures. - * * handle with care : * * avoid dropping or damaging the battery. - * * consider using a backup battery : * * this can provide a temporary power source in the event of a battery failure. - * * proper disposal : * * batteries should be disposed of properly to prevent environmental hazards. @topic:11


# # # issue the physiologic monitor 3150 mri from invivo research displays unstable oxygen saturation ( o2 sat ) and electrocardiogram ( ekg ) readings. # # # impact unstable o2 sat and ekg readings can affect the accuracy of patient monitoring and lead to delayed or incorrect diagnosis and treatment. # # # possible causes * * 1. sensor malfunction : * * - faulty or loose o2 sensor or ekg electrode - damaged or poorly connected cables * * 2. patient factors : * * - motion or shivering of the patient - vasoconstriction or vasodilation affecting o2 sat readings - arrhythmias or other cardiac abnormalities affecting ekg readings * * 3. equipment issues : * * - internal software or hardware malfunction - interfering electromagnetic signals from other medical devices - irregular power supply or battery depletion # # # recommendations * * 1. inspect sensors and cables : * * - check for secure connections and damaged sensors / electrodes. - replace or clean faulty components as necessary. * * 2. monitor patient conditions : * * - assess the patient's movements and ensure proper positioning. - rule out any underlying medical conditions that may contribute to unstable readings. * * 3. calibrate and test equipment : * * - perform regular calibration procedures to ensure accurate readings. - test the equipment with known values or a simulation device to verify functionality. * * 4. troubleshooting software and hardware : * * - update software if available. - contact the manufacturer for technical support or replacement parts if necessary. * * 5. electromagnetic interference mitigation : * * - identify and remove sources of electromagnetic interference, such as mobile phones or other medical devices. - consider shielding the monitor or using shielded cables. * * 6. power supply check : * * - ensure the monitor is connected to @topic:4


* * issue : * * philips intellivue mp70 ( m8007a ) bedside physiologic monitor cuff is faulty. * * impact : * * the faulty cuff prevents accurate blood pressure measurements, potentially leading to incorrect patient care decisions. * * possible causes : * * - damaged or worn - out cuff fabric, rendering it unable to inflate or deflate properly and provide accurate readings. - loose or disconnected tubing connecting the cuff to the monitor, causing communication issues and incorrect readings. - malfunction or calibration issues with the monitor's pressure transducer, resulting in inaccurate measurements. * * recommendations : * * - replace the damaged cuff with a new one to ensure proper inflation and deflation. - inspect and tighten any loose or disconnected tubing to restore communication between the cuff and the monitor. - calibrate the monitor's pressure transducer to ensure accurate readings. @topic:14


# # # issue the cause of equipment failure is not clearly specified. # # # impact the impact of the equipment failure is not mentioned. # # # possible causes - the equipment may have been damaged during transport from ward 6d1 to ward 5w. - the equipment may have been improperly installed or used. - the equipment may have been defective. # # # recommendations - the equipment should be inspected for damage before and after transport. - the equipment should be installed and used according to the manufacturer's instructions. - the equipment should be regularly maintained and serviced. @topic:12


select the recorder test button to start the test. the recorder prints a test pattern similar to the following illustration : 2 use the pattern to identify the recorder problem, such as a faulty print head or motor. troubleshooting suresigns vs4 service guide 4 - 31 testing the navigation wheel and keys testing the navigation wheel and keys the keys / rotary wheel test tests the functions of the navigation ( rotary ) wheel and the keys on the front panel. if you do not get the expected results, see “ display problems ” on page 4 - 4. to run the keys / rotary wheel test : step 1 in the system diagnostics menu, select the keys test button. the keys / rotary wheel test window appears. each fixed key on the monitor ( except on / standby ) is represented by a button in the right side of the window. keys / rotary wheel test push rotary wheel to exit. 2 press a fixed key on the front of the monitor to test it. if the key is working properly, the corresponding icon in the test window lights with a grey border for five seconds. repeat for each key on the front of the monitor. 3 to test the navigation wheel, turn the wheel clockwise or counterclockwise. if the wheel is working properly, the wheel icon on the screen draws red dots in the direction that you turn the wheel. 4 to end the test, press the wheel. troubleshooting 4 - 32 suresigns vs4 service guide testing the display testing the display to test the display : step 1 in the system diagnostics menu, select the display test button to start the test. the display test draws a sequence of patterns on the monitor screen. each pattern displays for five seconds. when the last pattern is drawn, the sequence starts again from the first pattern. the following describes the test patterns displayed : pattern description of screen vertical bars alternating black and white vertical bars, starting @topic:22


# # # issue - monitor not displaying ekg waveforms. # # # impact - inability to monitor patient's heart activity. # # # possible causes - * * defective ekg cable : * * the cable connecting the patient to the monitor may be damaged or faulty. - * * loose cable connection : * * the cable may not be securely connected to the patient or the monitor. - * * faulty monitor : * * the monitor itself may be malfunctioning and unable to process the ekg signals. - * * software error : * * a software bug or glitch may be preventing the monitor from displaying the waveforms. - * * improper configuration : * * the monitor may not be properly configured to receive and display ekg signals. # # # recommendations - * * inspect the ekg cable : * * check the cable for signs of damage or wear. ensure that the connectors are secure. - * * replace the ekg cable : * * if the cable appears damaged or is suspected to be faulty, replace it with a new one. - * * verify cable connections : * * make sure that the ekg cable is securely connected to the patient and the monitor. - * * restart the monitor : * * power down the monitor and restart it. this may clear any temporary software glitches. - * * check monitor configuration : * * ensure that the monitor is set to receive and display ekg signals. consult the manufacturer's manual for specific instructions. - * * contact technical support : * * if the issue persists after trying the above recommendations, contact the manufacturer's technical support for further assistance. @topic:18


# # # issue sphygmomanometer, electronic, automatic ( ge healthcare, dinmap procare 100 ) is unable to measure blood pressure. # # # impact patients are unable to receive accurate blood pressure readings. # # # possible causes - cuff leak # # # recommendations - replace the cuff. @topic:14


# # # issue - cuff bp torn # # # impact - inaccurate blood pressure readings - patient discomfort # # # possible causes - cuff bp is old and worn - cuff bp was not properly placed or inflated - animal chewed on cuff bp - cuff bp was damaged during cleaning - cuff bp was not properly stored # # # recommendations - inspect cuff bp regularly for signs of wear and tear - replace cuff bp if it is torn or damaged - ensure that cuff bp is properly placed and inflated - do not allow animals to chew on cuff bp - clean cuff bp according to manufacturer's instructions - store cuff bp in a dry, clean place @topic:3


# # # issue - monitor is freezing # # # impact - inability to monitor patient's vital signs - delays in patient care # # # possible causes - software glitch - battery issue - hardware failure # # # recommendations - restart the monitor - check the battery connections - consult the manufacturer's service manual for additional troubleshooting steps - contact technical support if the issue persists @topic:2


# # # issue - cuff of the vital signs monitor is damaged. # # # impact - the monitor cannot accurately measure vital signs, such as blood pressure. # # # possible causes - the cuff may be punctured or torn. - the cuff may be too old and worn out. - the cuff may not be the correct size for the patient. # # # recommendations - replace the cuff with a new one. - inspect the cuff regularly for signs of damage. - use the correct size cuff for the patient. @topic:5


##ist® es bed service manual ( man330 rev 4 ) page 4 - 17 4. 5 head section motor chapter 4 : removal, replacement, and adjustment procedures figure 4 - 8. head motor 14. loosen the jam nut ( j ) on the cpr cable ( f ). 15. remove the cpr cable ( f ) from the cable mount ( h ). note : the cpr cable is threaded into the cable mount. 16. remove the e - ring ( k ) from the pin ( l ) at the rod end of the motor ( i ). 17. remove the e - ring ( m ) from the pin ( n ) at the body end of the motor ( i ). 18. while supporting the motor ( i ), remove the pin ( l ). 19. while supporting the motor ( i ), remove the pin ( n ). 20. remove the motor ( i ) from the bed. page 4 - 18 careassist® and careassist® es bed service manual ( man330 rev 4 ) 4. 5 head section motor chapter 4 : removal, replacement, and adjustment procedures replacement caution : make sure that the elastic found on the new motors is not wound around the motor body before removing it ( cut ). if necessary, turn the motor end rod in the appropriate direction to unwind this elastic. it is used to indicate the correct position of the switches and internal end of travels of the motor. the incorrect position of these elements may damage the motor or the structure of the bed. 1. perform the removal procedure in reverse order. 2. remove the elastic on the motor. 3. activate the cpr function. the cpr function should activate when the cpr handle is approximately half way through its travel. 4. adjust the cpr cable as needed ( refer to procedure 4. 6 ). 5. to make sure the careassist @topic:26


# # # issue * broken casing on philips intellivue x2 ( m3002a ) patient transport monitor # # # impact * monitor may not be fully functional * safety hazard if broken casing exposes electrical components # # # possible causes * physical damage ( e. g., dropping, impact ) * excessive wear and tear * manufacturing defect # # # recommendations * inspect monitor for any visible damage * test monitor to ensure proper functionality * contact philips for repair or replacement if necessary * implement measures to prevent future damage ( e. g., using a protective case ) @topic:2


# # # issue - the electric bed cannot be adjusted up or down. # # # impact - patient care may be impacted as the bed cannot be adjusted to the desired height for patient comfort or medical procedures. # # # possible causes - power supply issue - electrical component failure - mechanical component failure - control panel malfunction - software glitch # # # recommendations - check the power supply to the bed to ensure it is connected and providing power. - inspect the electrical components, such as wires, connectors, and motors, for any signs of damage or loose connections. - examine the mechanical components, such as gears, pulleys, and hydraulic systems, for any signs of wear, damage, or obstruction. - check the control panel for any loose buttons or switches, and ensure that the display is functioning correctly. - update the software if available, as a software glitch could be causing the issue. # # # additional notes - the technician should follow the manufacturer's instructions for troubleshooting and repairing the bed. - if the issue persists after following the recommendations, it is advisable to contact the manufacturer for further assistance or replacement of the bed. @topic:1


* * issue * * - failure to lock the guardrail on the bed. * * impact * * - the guardrail may not be able to effectively prevent patients from falling out of bed, posing a safety hazard. * * possible causes * * - the lock mechanism on the guardrail is faulty. - the guardrail itself may be damaged or bent, preventing the lock from engaging properly. * * recommendations * * - inspect the lock mechanism on the guardrail and replace it if damaged. - check the guardrail for any damage or bending and repair or replace it if necessary. - ensure that the guardrail is properly installed and locked in place before using it for patient care. @topic:6


# # # issue - battery failure in welch allyn 53000 vital signs monitor. # # # impact - the device may not be able to accurately monitor patient vital signs. - this could lead to delayed or incorrect treatment decisions. # # # possible causes - the battery may have reached the end of its lifespan. - the battery may have been damaged by improper handling or storage. - there may be a problem with the device's charging system. # # # recommendations - replace the battery with a new one. - check the device's charging system to ensure that it is functioning properly. - follow the manufacturer's instructions for handling and storing the device and battery. @topic:11


# # # issue - equipment failure # # # impact - inability to monitor patient's vital signs accurately # # # possible causes - * * defective spo2 sensor cable * * # # # recommendations - replace the defective cable - ensure that the cable is properly connected to both the patient and the monitor - perform a calibration check after replacing the cable @topic:4


output on the intellivue patient monitors is delayed by a maximum of 30 ms. your biomedical engineer should verify that your ecg / defibrillator combination does not exceed the recommended maximum delay of 60 ms. configuration tasks you must configure these settings during installation in configuration mode. • line frequency • printer • altitude • equipment label ( for wireless networked monitors, or when the information center is in flexible monitoring mode ). 245 7 installation instructions configuration tasks setting altitude and line frequency you require a local barometric pressure rating from a reliable source ( such as airport, regional weather station, or hospital weather station ) that is located at the same altitude as the institution. 1 from the main setup menu, select global setting. select altitude and enter the altitude. 2 from the main setup menu, select global setting. select line frequency and choose the line frequency. configuring the equipment label if the information center is in fixed monitoring mode, it controls the equipment label. you do not need to follow this procedure. however, if you are on a wireless network, or your information center is configured for flexible monitoring mode, you must set the equipment label. this associates the monitor with a central monitoring sector. an identical monitor label must also be configured in the information center. 1 select the bed label screen element to call up the bed info menu. 2 select equipment label to call up the onscreen keyboard. 3 enter the system identifier. this needs to be set up in either the monitor or the information center. if the information center is in flexible monitoring mode, the monitor must be setup to match the information center ’ s monitor label. configuring the printer 1 from the main setup menu select reports. 2 select printer settings and configure local to enabled if the printer is connected directly to the monitor. see configuration guide for further details. 246 8 1 site preparation introduction this @topic:10


# # issue - the cause of failure is incorrect. # # # impact - this can lead to confusion and incorrect repairs. # # # possible causes - the technician may have misunderstood the user's explanation of the issue. - the user may have provided incorrect information. # # # recommendations - the technician should verify the cause of failure with the user. - the hospital should provide training to users on how to accurately report issues. @topic:12


# # # issue - broken temperature probe clip # # # impact - inability to accurately measure body temperature # # # possible causes - prolonged use and wear of the clip - mishandling of the equipment - manufacturing defect # # # recommendations - regular inspection and maintenance of the device - proper handling and storage of the equipment - contact the manufacturer for further assistance if necessary @topic:20


# # # issue - time displayed on the monitor is incorrect # # # impact - inaccurate monitoring of patient's vital signs - misleading information for healthcare professionals - potential delays or errors in patient care # # # possible causes - incorrect time setting - malfunctioning internal clock - interference with gps or network time synchronization # # # recommendations - verify the time setting and adjust if necessary - check the internal clock for proper functionality - ensure a stable gps or network connection for time synchronization - consider replacing the monitor if the issue persists @topic:29


# # # issue - broken locking mechanism on bed rails # # # impact - patient safety hazard : patient may fall out of bed due to unsecured rails. - inconvenience : caregivers may have difficulty accessing the patient or adjusting the bed position. # # # possible causes - manufacturing defect in the locking mechanism - excessive force applied to the rails - wear and tear over time # # # recommendations - inspect the bed rails regularly for any signs of damage or wear. - ensure that caregivers are properly trained on how to operate the bed safely. - replace any damaged locking mechanisms promptly. - consider using alternative bed rails or restraints if the patient is at high risk of falling. @topic:6


# # # issue - ekg signal not displayed on monitor # # # impact - clinicians cannot monitor patient's ekg activity, which is essential for diagnosing and treating cardiac conditions. # # # possible causes - loose connection between the ekg cable and the monitor - faulty ekg cable - malfunctioning monitor # # # recommendations - check the connection between the ekg cable and the monitor and ensure it is secure. - replace the ekg cable if it is damaged or faulty. - contact the manufacturer or a qualified technician to troubleshoot and repair the monitor if necessary. @topic:18


, the ethernet over usb interface with the wast is auto - configured with the specifications noted in the table below. when the connex spot monitor reaches out to the wast for the firmware package, it uses tcp port xxxx. be prepared to modify any firewall ( s ) that prevent incoming connections to allow this traffic. the wast attempts to configure the windows firewall by opening this specific port on the rndis interface. if administrative controls such as a group policy object ( gpo ) or third - party software have been implemented to prevent this change, the wast attempts to diagnose the problem and then inform you of possible remediation steps. work with your it staff or network administrator as needed to enable this functionality. interface specifications ip address network mask device ethernet over usb interface 255. 255. 255. 255 169. 254. 10. 10 69. 254. 10. 1 wast rndis interface 255. 255. 255. 255 at the start of a software upgrade, the welch allyn service tool creates a local http server and uses the usb connection to transfer necessary file package to the connex spot monitor. once the file package is on the device, it launches the installation of host controller firmware and reboots the device immediately following the installation. when the device auto - reboots, the software upgrade of individual modules and sensors ( nibp, spo2, and / or temperature ) is complete. software upgrade availability for the connex spot monitor can be checked directly within the welch allyn service tool ( wast ). availability of software upgrades is also communicated via the customer service bulletin on baxter's website at https : / / www. welchallyn. com / content / welchallyn / emeai / in / products / categories / patient - monitoring / vital - signs - devices / connex @topic:10


# # # issue the blood pressure monitor displays error code e40. # # # impact the blood pressure monitor cannot be used to measure blood pressure. # # # possible causes * firmware update not performed correctly * faulty sensor * problem with the display * malfunctioning valve * damaged or loose connections # # # recommendations * check if the firmware update was performed correctly and reinstall if necessary. * inspect the sensor for any damage or contamination and clean or replace if needed. * examine the display for any visible defects or damage. * inspect the valve for proper function and replace if faulty. * ensure that all connections are secure and properly seated. @topic:15


# # # issue - leaky lock saw # # # impact - the bed rails cannot be locked, which poses a safety hazard to the patient. # # # possible causes - the lock saw may be damaged or broken. - the lock saw may not be properly installed. # # # recommendations - inspect the lock saw for damage or breakage. - replace the lock saw if it is damaged or broken. - ensure that the lock saw is properly installed. - test the lock saw to ensure that it is functioning properly. @topic:6


* * issue : * * * loose storage compartment on the medical cart. * * impact : * * * potential for medical equipment or supplies to fall out of the cart, posing a safety hazard to patients and staff. * reduced efficiency and organization in storage and transportation of medical supplies. * * possible causes : * * * wear and tear over time, leading to loosening of screws or bolts. * improper assembly or maintenance of the medical cart. * overloading of the storage compartment, exceeding its weight capacity. * external impact or mishandling of the medical cart. * * recommendations : * * * * * regular inspection and maintenance : * * conduct routine inspections to identify loose components and tighten them accordingly. * * * proper assembly and installation : * * ensure that the medical cart is assembled and installed according to the manufacturer's instructions. * * * weight management : * * avoid overloading the storage compartment to prevent excessive strain on the screws or bolts. * * * careful handling : * * avoid rough handling or impacts to the medical cart to prevent loosening of components. * * * training for staff : * * educate staff on the proper handling and maintenance of the medical cart, including the importance of reporting any loose or damaged components promptly. @topic:31


# # # issue - remote control failure # # # impact - unable to adjust bed # # # possible causes - faulty push buttons - broken remote control cable # # # recommendations - replaced faulty push buttons ( 2 ) - repaired broken remote control cable - device is now fully operational @topic:7


# # # issue - equipment failure : the physiological monitor central station is not functioning properly. # # # impact - patient monitoring : the device is unable to accurately monitor the patient's vital signs, which can lead to delayed diagnosis and treatment. - clinical workflow : the healthcare team is unable to access real - time patient data, which can impact decision - making and patient care. # # # possible causes - loose or disconnected cables : the signal cable connecting the monitor to the central station may have become loose or disconnected, leading to a loss of signal and display. - faulty hardware : the monitor or central station may have malfunctioning hardware components, such as a faulty display or power supply, causing the equipment to fail. - software issues : software bugs or glitches within the monitor or central station can result in unexpected behavior, including a blank display. # # # recommendations - * * check connections * * : ensure that all cables are securely connected and that the power supply is providing adequate voltage to the equipment. - * * inspect hardware * * : examine the monitor and central station for any visible signs of damage, such as loose components or burnt wires. - * * update software * * : check for any available software updates and apply them to the equipment to address potential bugs and improve performance. - * * contact the manufacturer * * : if the issue persists after troubleshooting and self - repair attempts, contact the manufacturer for further assistance or a replacement device. @topic:2


# # # issue - vital signs monitor is pumping air continuously # # # impact - patient may experience discomfort or injury due to excessive pressure from the blood pressure cuff. - inaccurate vital signs readings due to incorrect cuff inflation. # # # possible causes - * * defective air pump : * * the pump that inflates the blood pressure cuff may have failed, causing it to pump air continuously. - * * leak in the air hose : * * a leak in the hose that connects the pump to the cuff can cause air to escape, resulting in continuous pumping. - * * faulty pressure sensor : * * the sensor that measures blood pressure may be faulty, providing incorrect pressure readings to the pump, causing it to continue pumping. - * * software issue : * * a software error may have caused the pump to malfunction and pump air continuously. # # # recommendations - * * inspect the air pump : * * check if the pump is functioning properly and producing the correct amount of pressure. - * * examine the air hose : * * inspect the hose for any leaks or damage that could cause air to escape. - * * test the pressure sensor : * * verify the accuracy of the pressure sensor by comparing the readings with another calibrated device. - * * update software : * * if a software issue is suspected, update the monitor's firmware to the latest version. - * * contact the manufacturer : * * if the above troubleshooting steps do not resolve the issue, contact the manufacturer for further assistance or repair. @topic:14


# # # issue - malfunctioning bed barrier ( left side ) # # # impact - patient safety hazard, as the barrier is not operational and could potentially lower unexpectedly, causing injury to the patient. # # # possible causes - broken wire in the left side barrier electrical system # # # recommendations - replace the broken wire and ensure the integrity of the electrical system to prevent further failures. consider regular maintenance and inspection of the bed, especially the barriers, to identify and address potential issues early on. @topic:9


# # # issue - broken bed electric remote # # # impact - malfunctioning bed, hindering patient care and safety # # # possible causes - physical damage due to mishandling, accidental drop, or excessive force - electrical damage caused by power surges, short circuits, or liquid spills - wear and tear from prolonged use # # # recommendations - proper handling and use of the remote - regular inspection and maintenance of the bed and its components - training of staff on proper handling and usage - consider using more durable or impact - resistant remotes if prone to damage - ensure electrical safety measures are in place to prevent power surges and electrical faults @topic:7


##se the grey front cover ( the console covering the measurement connector hardware ) gently from the bottom of the extension. position the screwdriver in the small slits provided for this purpose. the front cover then clicks away from the extension. 3 remove the front cover. removing the extension bottom cover to remove the extension bottom cover, do the following : 1 position the extension on the dual link bar with the measurement connector hardware facing upwards and the arm of the dual link bar towards you. there are four long mounting pins threaded into the extension in each of the four corners under the cover. locate the heads of the two long mounting pins on the side away from you 2 use tweezers to prise the pins gently out enough to be removed by hand. 3 remove the two pins and set them aside for refitting. note 4 using your hands, gently pry the bottom cover away from the extension at the link bar end first. the bottom cover is press - latched at the link bar end. remove it gently making sure not to bang or touch the inside of the extension. note if you accidentally try to remove the wrong side of the bottom cover, you will notice that it is attached to the inside of the extension with a ribbon connector and that the dual link bar prevents you from removing it completely. do not try to forcibly remove the wrong side of the m3015a cover ; you cannot access replaceable parts from this side. do not lose these long since the extension will not function unless they are in place.. 169 5 repair and disassemblydisassembly procedures for the m3015a measurement server extension ( hw rev. a ) the following illustration shows the location of the replaceable parts in the m3015a measurement server extension. brackets holding the pump c0 2 scrubber red band for removing the ir lamp when the assembly has been opened removing the c0 2 scrubber @topic:19


# # # issue - motor failure in knee section of electric hospital bed # # # impact - patient discomfort or immobility due to inability to adjust bed position # # # possible causes - overload or excessive force on knee motor - electrical fault or short circuit - mechanical wear or damage to motor components - improper maintenance or lack of lubrication # # # recommendations - inspect and test motor for any signs of damage or wear - check electrical connections for loose wires or corrosion - lubricate moving parts as per manufacturer's instructions - ensure proper user training to prevent overloading or misuse - consider replacing motor if necessary @topic:9


# # # issue monitor does not display any output # # # impact the bedside monitor is used to display vital signs, such as heart rate, blood pressure, and oxygen saturation. if the monitor is not displaying any output, it may be difficult or impossible to monitor the patient's condition. this could lead to a delay in diagnosis and treatment, which could have serious consequences for the patient's health. # # # possible causes - the monitor is not plugged in - the monitor is not turned on - the monitor's power cord is damaged - the monitor's display is damaged - the monitor's software is malfunctioning # # # recommendations - check that the monitor is plugged in and turned on. - check the monitor's power cord for damage. replace the power cord if it is damaged. - check the monitor's display for damage. if the display is damaged, replace the monitor. - contact the manufacturer of the monitor to troubleshoot the software malfunction. @topic:2


# # issue - cause of failure is not specified # # impact - it will be difficult for the technician to know what caused the failure and how to fix it. # # possible causes - the technician may have forgotten to specify the cause of failure. - the technician may not know the cause of failure. - the technician may not have had time to investigate the cause of failure. # # recommendations - the technician should specify the cause of failure in the description of the medical device. - the technician should investigate the cause of failure before attempting to fix the medical device. - the technician should consult with a more experienced technician if they are unable to determine the cause of failure. @topic:12


* * issue : * * electric bed's elevating side rail not functioning * * impact : * * patient safety and comfort is compromised, as the side rail cannot be raised to prevent falls. * * possible causes : * * * * * damaged wire : * * the technician identified that the elevating side rail was not functioning due to a damaged wire. a damaged wire can cause electrical interruptions and prevent the side rail from receiving power. * * * faulty electrical connection : * * a loose or faulty electrical connection between the side rail and the bed's control system can also cause the side rail to stop functioning. * * * control system malfunction : * * the bed's control system may have developed a fault that is preventing it from sending the correct commands to the side rail. * * recommendations : * * * * * inspect the wire : * * the technician should thoroughly inspect the wire for any visible damage or breaks. if the wire is damaged, it should be replaced. * * * check electrical connections : * * all electrical connections between the side rail and the bed's control system should be checked and tightened if necessary. * * * test the control system : * * the bed's control system should be tested to ensure that it is sending the correct commands to the side rail. if the control system is faulty, it may need to be repaired or replaced. @topic:6


cable ( 3. 0 m ) adapter cable for masimo lncs sensors yes no 989803148221 the philips m3001a / m3002a option ao 1 and the m1020b option ao 1 sp0 2 module use masimo certified pulse oximetry for reduced noise and low perfusion performance with masimo sensors under the masimo nr & lp protocol available from masimo. sensors for use with the masimo set® sp0 2 module and orderable from philips sensors philips order no lnop adult adhesive sensors ( box of 20 ) 989803140231 adult sensor sample pack ( 4 sensors ) 989803140241 neo sensor sample pack ( 4 sensors ) 989803140251 pediatric adhesive sensors ( box of 20 ) 989803140261 lnop neonatal adhesive sensors ( box of 20 ) 989803140271 neo pre - term sensitive skin adhesive sensors ( box of 20 ) 989803140281 neo adhesive sensors ( box of 20 ) 989803140291 neo pre - term sensitive skin adhesive sensors ( box of 20 ) 989803140301 neo / infant adhesive sensors ( box of 20 ) 989803140311 dc - i adult reusable sensor 989803140321 dcip pediatric reusable sensor 989803140331 lnop tc - i tip clip reusable sensor 989803140341 neo sensitive replacement posey wrap ( 12 ) 989803140351 neo replacement tapes ( 100 ) 989803140361 392 temperature accessories 37 accessories temperature accessories temperature probes part no. minimum measurement time for accurate readings reusable general purpose probe 21075a 90 sec small flexible vinyl probe ( infant / @topic:8


# # # issue - the physiologic monitor ( bedside ) from philips ( model : intellivue mp70 ( m8007a ) ) is not displaying any lights when connected. # # # impact - the monitor is essential for patient monitoring, and its failure can delay or impact proper patient care. # # # possible causes - * * faulty power supply : * * ensure that the power cord and wall outlet are functioning correctly and the monitor is receiving power. - * * loose connections : * * check all cables and connections for any loose or disconnected ends. tighten them securely. - * * damaged or defective sensor : * * the new spo2 sensor might be faulty or damaged. try connecting a different sensor to confirm if the issue lies with the sensor. - * * monitor settings : * * verify that the monitor settings are configured correctly, including the appropriate alarm limits and display parameters. - * * internal hardware failure : * * there could be an underlying hardware issue within the monitor that requires further troubleshooting or repair. # # # recommendations 1. * * power supply check : * * inspect the power cord, wall outlet, and monitor power connection for any visible damage or loose connections. replace or repair any faulty components. 2. * * cable and connection inspection : * * ensure all cables and connections are secure and properly inserted. tighten any loose connections and replace any damaged cables. 3. * * sensor verification : * * connect a different spo2 sensor to the monitor to isolate the issue. if the new sensor works properly, the original sensor may be defective and needs replacement. 4. * * monitor settings review : * * refer to the manufacturer's instructions or consult a biomedical engineer to verify and adjust the monitor's settings appropriately. 5. * * hardware troubleshooting : * * if the above steps don't resolve the issue, there may @topic:22


# # # issue : - monitor shows unrecoverable bisx error 24 # # # impact : - inability to monitor patient's brain activity accurately - delayed or incorrect medical decision - making - increased risk of patient complications # # # possible causes : - hardware malfunction in the monitor - faulty sensor connection / cable - software bug / glitch - power supply issue - battery depletion # # # recommendations : - * * check sensor connection * * : ensure that the sensor cables are securely connected to the patient and the monitor. - * * inspect cables * * : examine the cables for any visible damage or breaks. replace if necessary. - * * restart monitor * * : turn off the monitor and turn it back on to reset the software. - * * contact biomed tech * * : if the issue persists after troubleshooting, contact the biomed tech team for further assistance. - * * consider using backup monitor * * : if possible, utilize a backup monitor to continue monitoring the patient's brain activity while the primary monitor is being repaired. - * * document error * * : log the error message, date, and time for future reference and troubleshooting purposes. @topic:22


# # # issue - battery not charging # # # impact - physiological monitor cannot be used. # # # possible causes - faulty battery - faulty charger - problem with the physiological monitor's charging port # # # recommendations - check the battery and charger for any damage. - try using a different battery and charger. - reset the physiological monitor. - if the problem persists, contact the manufacturer for further assistance. @topic:11


* * issue : * * equipment malfunction after maintenance by vendor * * impact : * * equipment unavailable for patient monitoring * * possible causes : * * * improper maintenance procedure by vendor * vendor used defective parts * damage to equipment during maintenance * * recommendations : * * * review maintenance logs and contact the vendor to determine the specific cause of failure. * inspect the equipment carefully for any signs of damage or improper maintenance. * consider obtaining a second opinion from another qualified vendor to verify the diagnosis. * establish clear maintenance protocols and ensure that all vendors adhere to them. * consider investing in specialized training or equipment for the in - house maintenance of biomedical equipment. @topic:12


# # # cause of failure : power cord malfunction # # # possible causes the power cord may be damaged, resulting in a loss of power to the monitor. # # # recommendations - replace the power cord with a new one. - inspect the power cord for any damage, such as cuts or breaks. - ensure that the power cord is properly connected to both the monitor and the power outlet. @topic:2


* * issue : * * * cuff bp ( blood pressure ) readings are abnormally high for all patients. * * impact : * * * inaccurate blood pressure readings can lead to incorrect diagnoses and treatment plans. * patient safety is compromised due to the potential for over - or undertreatment. * * possible causes : * * * * * defective cuff bp : * * the cuff bp may be maladjusted or damaged, leading to inaccurate readings. * * * improper cuff placement : * * the cuff bp may not be placed correctly on the patient's arm, resulting in false high readings. * * * patient positioning : * * the patient may not be positioned appropriately during the blood pressure measurement, influencing the results. * * * instrument error : * * the vital signs monitor itself may have an issue causing the inflated cuff pressure reading to be abnormally high, independent of the cuff. * * * human error : * * the technician performing the measurement may have made a mistake, such as using the wrong cuff size or inflating the cuff incorrectly. * * recommendations : * * * * * check the cuff bp : * * replace the cuff bp with a new one to eliminate any issues with the existing cuff. * * * verify cuff placement : * * ensure that the cuff bp is placed correctly on the patient's arm, following the manufacturer's guidelines. * * * instruct the patient : * * inform the patient about the importance of proper positioning and remaining still during the measurement. * * * calibrate the monitor : * * perform a calibration check on the vital signs monitor to ensure its accuracy. * * * retrain staff : * * if human error is suspected, provide additional training to technicians on proper blood pressure measurement techniques. @topic:14


# # # issue - printer is not functioning properly and unable to print. # # # impact - inability to obtain printed documentation, reports, or patient information. this can hinder the efficient operation of the hospital and delay patient care. # # # possible causes - printer hardware malfunction or mechanical failure - printer driver or software issues on the central monitor - connectivity problems between the printer and the central monitor - paper or toner shortage or misalignment - printer settings or configuration errors # # # recommendations - * * verify printer connections : * * ensure that the printer is properly connected to the central monitor and that all cables are securely plugged in. - * * check paper and toner : * * inspect the printer for any paper jams or toner shortages. ensure that the correct type of paper is being used. - * * restart printer : * * turn off the printer, wait a few minutes, and then restart it. this can sometimes resolve minor software glitches. - * * check printer settings : * * access the printer's settings and ensure that the correct print options are selected. check for any error messages or warnings. - * * update printer driver : * * if the printer is connected to a computer or network, verify that the latest printer driver is installed and updated. - * * inspect printer hardware : * * if the issue persists, consider inspecting the printer's hardware for any visible damage or loose connections. - * * contact manufacturer support : * * if the problem cannot be resolved through the above steps, contact the manufacturer of the printer for technical assistance or repair. @topic:28


4. 0 kpa ) 50 mmhg ( 6. 6 kpa ) 30 mmhg ( 4. 0 kpa ) imc0 2 4 mmhg ( 0. 5 kpa ) na 4 mmhg ( 0. 5 kpa ) na 4 mmhg ( 0. 5 kpa ) na nbp systolic 160 mmhg ( 21. 3 kpa ) 90 mmhg ( 12. 0 kpa ) 120 mmhg ( 16. 0 kpa ) 70 mmhg ( 9. 3 kpa ) 90 mmhg ( 12. 0 kpa ) 40mmhg ( 5. 3 kpa ) nbp diastolic 90 mmhg ( 12. 0 kpa ) 50 mmhg ( 6. 6 kpa ) 70 mmhg ( 9. 3 kpa ) 40 mmhg ( 5. 3 kpa ) 60 mmhg ( 8. 0 kpa ) 20 mmhg ( 2. 6 kpa ) nbp map ( mean ) 110 mmhg ( 14. 6 kpa ) 60 mmhg ( 8. 0 kpa ) 90 mmhg ( 12. 0 kpa ) 50 mmhg ( 6. 6 kpa ) 70 mmhg ( 9. 3 kpa ) 24 mmhg ( 3. 2 kpa ) configuring the monitor 5 - 13 setting alarm and audio disables adult pediatric neonatal ibp systolic 160 mmhg ( 21. 3 kpa ) 90 mmhg ( 12. 0 kpa ) 120 mmhg ( 16. 0 kpa ) 70 mmhg ( 9. 3 kpa ) 90 mmhg ( 12. 0 kpa ) 40 mmhg ( 5. 3 kpa ) ibp diastolic 90 mmhg ( 12. 0 kpa ) 50 mmhg ( 6. 6 kpa ) 70 mmhg ( 9. 3 kpa ) 40 mm @topic:8


# # # issue - loose plug # # # impact - equipment not receiving power # # # possible causes - physical damage to the plug or power cord - faulty electrical connection - loose connection between the plug and the equipment # # # recommendations - inspect the plug and power cord for any visible damage and replace if necessary. - ensure that the electrical connection is secure and free of corrosion. - tighten the connection between the plug and the equipment. - consider using a surge protector or power conditioner to protect the equipment from voltage fluctuations. @topic:12


# # # issue - missing cuff connector # # # impact - inability to connect the cuff to the monitor, preventing vital signs monitoring # # # possible causes - connector was not properly secured and became detached - connector was damaged or broken # # # recommendations - inspect the connector regularly to ensure it is secure - replace the connector if it is damaged or broken - train staff on the proper use and maintenance of the equipment @topic:5


# # # issue failed to replace battery and charge it for 8 hours. # # # impact the medical device may not function properly due to insufficient battery power. # # # possible causes - the battery may not have been properly installed. - the battery may be defective. - the battery charger may be defective. # # # recommendations - ensure that the battery is properly installed according to the manufacturer's instructions. - test the battery using a battery tester to verify its functionality. - inspect the battery charger for any damage or defects. * * additional notes : * * - the technician should also check the device's power supply to ensure that it is providing the correct voltage and current. - the technician should document the battery replacement and charging process in the device's maintenance log. @topic:11


# # # issue spo2 cable not illuminating # # # impact unable to obtain accurate spo2 readings, potentially leading to delayed or incorrect medical decisions. # # # possible causes - damaged or disconnected cable - faulty connectors - hardware or software issue within the monitor # # # recommendations - inspect the cable for any visible damage or loose connections. - ensure that the connectors are securely attached to the monitor and the patient's finger. - perform a troubleshooting procedure as outlined by the manufacturer to determine if the issue is with the cable, the monitor, or the patient's finger. - replace the cable if necessary. - contact the manufacturer for further assistance if the issue persists after following the troubleshooting procedure. @topic:4


# # # issue the remote control for the electric hospital bed is not functioning properly. # # # impact the patient is unable to adjust the bed's position, which can lead to discomfort and difficulty performing daily activities. # # # possible causes * the remote control's batteries are dead. * the remote control is not properly paired with the bed. * there is a loose connection between the remote control and the bed. * the remote control's buttons are not working properly. * the remote control's circuit board is damaged. # # # recommendations * check the batteries in the remote control and replace them if necessary. * pair the remote control with the bed according to the manufacturer's instructions. * check the connections between the remote control and the bed and tighten any loose wires. * clean the remote control's buttons with a cotton swab and rubbing alcohol. * if the above steps do not resolve the issue, contact the manufacturer of the bed for further assistance. @topic:7


# # # issue - ups not charging battery - ups alarm # # # impact - medical equipment may not function properly during a power outage. # # # possible causes - battery may be faulty - ups may be faulty # # # recommendations - replace the battery. - check the ups for any faults. @topic:11


feeding tube, orange french ) 989803183561 innersense esophageal temperature / feeding tube, orange ( 6. 5 french ) 989803183571 innersense esophageal temperature / feeding tube, orangfrench ) 989803183581 cardiac output ( c. o. ) accessories see pressure accessories for pulsion continuous cardiac output accessories. description art no. accessories common to both methods s set of ice buckets 4455a remote handswitch 15244a picco inline temperature probe m1646a right heart thermom1642a. 4 m 2. 4 m cable m1643a right heart catheter for thermodilution edwards co - set + ( only with 23001a / b ) ( from edwards lifesciences ) contact edwards lifesciences for information 831hf75 2. 4 m injectate temp. probe ( reusable0. 5 m injectate temp. probe ( reusableice bath temp. probe 23001a 23001b 23002a. o. interface cable. 4 m 2. 4 m cablm1643a pulsion pressure transducer kits ( pressure line contact pulsion for informatioavailable in various lengths ) pressure interface cable for disposable pressure pulsion pmk 206 transducer 436 39 accessories mainstream co2 accessories description sensor part no. m2501a adult / pediatric airway adapter ( reusable ) m2513a infant airway adapter ( reusable ) m2516a adult airway adapter ( single - patient use ) m2533a infant airway adapter ( single - patient use ) 2536a sidestream co2 accessories description art no. m27 @topic:8


# # # issue - the electric bed is making a loud noise. # # # impact - the noise may disturb patients and staff, and it could indicate a problem with the bed that could lead to a failure. # # # possible causes - the bed frame or joints may be loose or damaged. - the motor or gears may be worn or damaged. - the bed may not be properly lubricated. - the bed may have been overloaded. - the bed may have been used improperly. # # # recommendations - inspect the bed frame and joints for any loose or damaged parts. - inspect the motor and gears for any wear or damage. - lubricate the bed according to the manufacturer's instructions. - ensure that the bed is not overloaded. - ensure that the bed is used properly. - if the problem persists, contact the manufacturer or a qualified technician for further assistance. @topic:9


# # # issue - the provided input does not specify the cause of failure for the medical device. # # # impact - as the cause of failure is unknown, it is difficult to assess the impact of the failure on the device or its operation. # # # possible causes - the technician may have failed to properly identify the cause of failure during their inspection. - the device may have an intermittent failure that was not detected during testing. - the device may have been damaged during transport or installation. - the device may have been misconfigured or improperly maintained. # # # recommendations - the technician should conduct further troubleshooting to identify the root cause of the failure. - the technician should consult with the device manufacturer for technical support. - the device should be recalibrated and tested to ensure proper operation. - the hospital should implement a regular maintenance schedule for the device to prevent future failures. @topic:12


# # # issue - equipment : monitors, physiologic, central station - band : philips - model : m3150a - description : none - cause of failure : equipment is damaged and cannot be used # # # impact - the central station monitor is not functioning, which could lead to delays in patient care or incorrect patient data being collected. # # # possible causes - the equipment may have been dropped or damaged during transport. - the equipment may have been exposed to moisture or other environmental factors that have caused damage. - the equipment may have been overloaded or used in a way that it was not designed for. # # # recommendations - inspect the equipment for any visible damage. - check the equipment's power supply and connections. - check the equipment's software and firmware for any updates. - contact the manufacturer or a qualified technician for further assistance. @topic:2


# # # issue - cuff bp leak # # # impact - inaccurate blood pressure readings - delays in patient care - increased risk of adverse events # # # possible causes - damaged blood pressure cuff - loose connections - faulty sensor # # # recommendations - replace the blood pressure cuff - check connections and ensure they are secure - calibrate the sensor according to the manufacturer's instructions @topic:3


# # # issue - the bed makes a loud noise when it is raised or lowered. # # # impact - the noise can be annoying to patients and staff. - the noise can make it difficult to use the bed. # # # possible causes - loose or worn joints - dry or damaged bearings - misaligned components # # # recommendations - inspect the bed for loose or worn joints. - lubricate the bed's joints and bearings. - realign the bed's components. @topic:9


# # # issue - cuff inflates and does not deflate # # # impact - patient's blood pressure cannot be measured accurately - patient may experience discomfort or pain # # # possible causes - cuff is too tight - air hose is leaking - pump is malfunctioning - cuff bladder is ruptured # # # recommendations - check the cuff size and adjust if necessary - replace the air hose if it is leaking - inspect the pump for any signs of damage or malfunction - replace the cuff bladder if it is ruptured @topic:3


# # # issue : - equipment : electric bed - model : uq2000b - malfunction : unable to adjust the headrest down # # # impact : - patient discomfort and inconvenience - prolonged hospital stay - increased workload for healthcare staff # # # possible causes : - * * faulty remote control : * * the remote control may be malfunctioning or damaged, preventing it from sending signals to the bed. - * * loose or damaged wiring : * * the wiring connecting the remote control to the bed may be loose or damaged, disrupting the signal transmission. - * * mechanical issue : * * the bed mechanism responsible for adjusting the headrest may be malfunctioning or obstructed. - * * software glitch : * * a software issue in the bed's control system may be preventing the headrest adjustment function from working properly. # # # recommendations : 1. * * inspect and replace the remote control : * * examine the remote control for any visible damage or wear and tear. if necessary, replace it with a new or spare remote. 2. * * check wiring connections : * * ensure that all wiring connections between the remote control and the bed are secure and free of damage. replace any loose or damaged cables. 3. * * inspect and troubleshoot the bed mechanism : * * check the bed mechanism for any obstructions or mechanical faults. clean and lubricate any moving parts as needed. 4. * * reset the bed's control system : * * if the issue persists, reset the bed's control system by following the manufacturer's instructions. this may resolve any software glitches. 5. * * consult with a qualified technician : * * if the issue cannot be resolved by following these recommendations, contact a certified biomedical engineer or technician for further diagnosis and repair. @topic:1


# # # issue - cuff is damaged. # # # impact - vital signs monitor cannot take accurate measurements. # # # possible causes - cuff is old and worn. - cuff is punctured or torn. - hook - and - loop closure is not working properly. # # # recommendations - replace the cuff. - inspect the cuff for damage regularly. - clean the cuff regularly according to the manufacturer's instructions. @topic:5


# # # issue - the patient's headrest cannot be adjusted. this can be a problem for both the patient and the healthcare providers. # # # impact - the patient may be uncomfortable or in pain if they are unable to adjust their headrest. this can make it difficult for them to rest or sleep. - the healthcare providers may be unable to properly care for the patient if they are unable to adjust the headrest. this can include providing proper medical treatment or monitoring the patient's condition. # # # possible causes - the headrest may be obstructed by something, such as a blanket or pillow. - the headrest may be damaged or broken. - the headrest may be incorrectly installed. # # # recommendations - check the headrest for obstructions and remove them. - inspect the headrest for damage or breakage. if the headrest is damaged or broken, it will need to be repaired or replaced. - reinstall the headrest according to the manufacturer's instructions. @topic:1


# # # issue - cuff failure in philips suresigns vs4 vital signs monitor # # # impact - inaccurate vital signs readings - patient safety compromised # # # possible causes - cuff degradation due to age and wear and tear - improper cuff maintenance or storage - cuff damage during use # # # recommendations - replace cuff with a new one - conduct regular cuff inspections and maintenance - ensure proper cuff storage and handling - train staff on correct cuff usage and troubleshooting @topic:15


# # # issue - spo2 sensor failure # # # impact - inability to accurately monitor patient's oxygen saturation levels # # # possible causes - damaged or faulty spo2 sensor - loose or disconnected sensor cable - interference from other electrical devices # # # recommendations - replace the spo2 sensor with a new one - ensure that the sensor cable is properly connected and secured - check for any potential sources of electrical interference and eliminate them if possible @topic:4


# # # issue - vital signs monitor is displaying an incorrect date. # # # impact - the inaccurate date may affect the accuracy of patient data and monitoring. # # # possible causes - the date may not have been set correctly during initial setup. - the battery may have died, causing the date to reset to default. - there may be a software glitch or hardware issue that is preventing the date from being set correctly. # # # recommendations - check the date and time settings on the monitor and ensure they are correct. - if the date and time cannot be set correctly, contact the manufacturer for technical support. - regularly inspect and maintain the vital signs monitor to prevent future issues. @topic:29