2. 2 a and an internal thermal cutout that cannot be reset. there is one non - fused secondary for the main control board and 1 fused secondary for the rest of the circuitry both followed at the outputs by bridge rectifiers and filter capacitors. the value of this antisurge fuse is 1 a. the non - fused secondary, through the bridge rectifier and filter capacitor, produces an output voltage vdc1 which is approximately 24 v dc and unregulated. the fused secondary produces a soft regulated 12 v dc for use in the sidecom® communication system pc. board and bed exit system. the available power enables several functions to be controlled simultaneously. careassist® and careassist® es bed service manual ( man330 rev 4 ) page 3 - 5 electrical system ( a model beds ) chapter 3 : theory of operation function control the motors are controlled by relays. in order to protect these relays from electrical sparks during switching, the second fet transistor checks the delay power when a function is activated, thus allowing the switches to close with no voltage. in the same way, when a function control is released, the fet transistor cuts the power before the contact opens. overload limiting is provided by permanently controlling the current at the output of each motor. in the event of overload or mechanical blockage, the supply of the relevant motor is cut off by a control transistor without affecting the operation of the other motors, except for specific situations. in this case, the overload is indicated by an audible alarm. this overload indication also enables failure of a motor end of travel to be detected, except for the hilow columns, which have a backup end of travel. battery circuit the bed is equipped with battery backup. this battery backup is internal to the power supply unit. the charging circuit, @topic:9


# # # issue - bed adjustment failure # # # impact - inability to adjust the patient's position, which can lead to discomfort or medical complications. # # # possible causes - faulty remote control - electrical issues within the bed - mechanical problems with the bed's adjustment mechanisms # # # recommendations - verify that the remote control is functioning correctly. - inspect electrical connections and wiring for damage or loose contacts. - test the bed's adjustment mechanisms for any mechanical issues. - consider replacing the remote control or repairing any electrical or mechanical faults. @topic:7


# # # issue - sat o2 cable failure # # # impact - inaccurate or intermittent sat o2 monitoring - reduced patient safety # # # possible causes - cable damage due to wear and tear - loose or damaged connectors - electrical interference # # # recommendations - replace the sat o2 cable with a new or spare one - inspect and ensure all connections are secure - check for and eliminate any potential sources of electrical interference @topic:4


the cam showing is towards the foot end of the bed. warning : make sure the caster is correctly positioned with respect to the movements of the hexagonal bar. failure to do so may cause the brake and steer system to malfunction. personal injury or equipment damage could occur. 3. install the new caster ( f ) and socket ( g ) into the bed ( b ) so the window with the cam showing is towards the foot end of the bed. careassist® and careassist® es bed service manual ( man330 rev 4 ) page 4 - 45 4. 13 caster chapter 4 : removal, replacement, and adjustment procedures 4. install the two alignment bolts ( c ). 5. install the bolt ( e ) and nut ( d ) through the socket ( g ). 6. make sure that the new caster is braked. 7. if the caster is not braked, do as follows : a. remove the caster. b. rotate the cam inside the caster clockwise until the caster is braked. c. install the caster. d. make sure the new caster is braked. e. repeat as needed. 8. install the applicable head or foot frame cover ( a ). 9. to make sure the careassist® bed operates correctly, do the “ function checks ” on page 2 - 5. page 4 - 46 careassist® and careassist® es bed service manual ( man330 rev 4 ) 4. 14 sidecom® communication system p. c. board ( a model beds ) chapter 4 : removal, replacement, and adjustment procedures 4. 14 sidecom® communication system p. c. board ( a model beds ) tools required : t15 torx® 1 screwdriver t25 torx® screwdriver removal 1. set the brakes. 2. raise the @topic:26


* * issue * * the equipment is unable to change units from ib ( pounds ) to kg ( kilograms ) on the scale mode. * * impact * * the equipment cannot accurately measure weight in kilograms, which can lead to errors in patient care. * * possible causes * * * faulty scale mode board * software malfunction * calibration error * * recommendations * * * replace the scale mode board. * check and update software as needed. * calibrate the equipment according to manufacturer's instructions. * perform regular maintenance and testing to ensure equipment accuracy. @topic:25


# # # issue - lcd monitor is not turning on. # # # impact - inability to use the monitor for medical purposes. # # # possible causes - * * faulty power supply : * * the power adapter may not be properly connected or may be malfunctioning. - * * damaged power cord : * * the power cord connecting the monitor to the power outlet may be damaged or loose. - * * defective monitor : * * the monitor itself may have a hardware failure that prevents it from turning on. # # # recommendations - * * check power connections : * * ensure that the power adapter is securely connected to both the monitor and the power outlet. - * * inspect power cord : * * examine the power cord for any damage or loose connections. replace the cord if necessary. - * * restart monitor : * * unplug the monitor from the power outlet and wait for a few seconds before plugging it back in. - * * contact manufacturer : * * if the above steps do not resolve the issue, contact the manufacturer of the monitor for further assistance. @topic:2


# # # issue - equipment : medical cart - band : bdms - model : bdms 01 - description : machine cannot be turned on. # # # impact - the medical cart cannot be used, which may impact patient care. # # # possible causes - loose power cord connection to the power board. # # # recommendations - ensure that the power cord is securely connected to the power board. - test the equipment to verify that it is functioning properly. @topic:13


# # # issue the bedside physiologic monitor lacks a port for connecting to a large display. # # # impact the inability to connect the monitor to a large display limits the visibility and accessibility of patient data for healthcare professionals. # # # possible causes * * * design flaw : * * the manufacturer may have omitted the port due to space constraints or cost considerations. * * * compatibility issue : * * the monitor may not be compatible with the available large displays. * * * human error : * * the monitor may have been incorrectly configured or calibrated, preventing the port from functioning properly. # # # recommendations * * * contact the manufacturer : * * reach out to philips to inquire about the availability of an adapter or software update that enables the use of a large display. * * * explore alternative solutions : * * consider using a third - party adapter or connecting the monitor to a computer with a larger screen. * * * request a replacement monitor : * * if the issue cannot be resolved, request a replacement monitor with the necessary port. @topic:2


# # # issue - the remote lock button on the hill - rom 900b electric bed is not functioning properly. # # # impact - nursing staff are unable to properly lock and unlock the bed, which could pose a safety hazard to patients. # # # possible causes 1. * * mechanical failure : * * the remote lock button may be physically damaged or misaligned. 2. * * electrical failure : * * there may be a short circuit or other electrical fault in the wiring or components of the remote lock system. 3. * * software bug : * * the bed's software may contain a bug that is preventing the remote lock button from functioning properly. # # # recommendations 1. * * inspect the remote lock button : * * check for any visible signs of damage or misalignment. 2. * * test the wiring and components : * * use a multimeter or other diagnostic tool to test the continuity and voltage of the wiring and components in the remote lock system. 3. * * update the bed's software : * * if a software bug is suspected, update the bed's software to the latest version. 4. * * contact the manufacturer : * * if the issue cannot be resolved by following the above steps, contact the manufacturer of the bed for further troubleshooting assistance. @topic:7


# # # issue - the equipment is not working correctly. - the specific issue is that the barcode scanner is not functioning. # # # impact - the inability to scan barcodes could impact patient care if barcode data is used for patient identification, medication administration, or other critical tasks. # # # possible causes - hardware malfunction in the barcode scanner. - loose or disconnected cables or connections. - incorrect configuration or settings. - software issues or bugs. - physical damage to the barcode scanner. - other system or environmental factors. # # # recommendations 1. * * restart the equipment. * * this can often resolve minor issues or software glitches. 2. * * check cables and connections. * * ensure that all cables and connections are secure and properly seated. 3. * * verify configuration and settings. * * review the equipment's configuration and settings to ensure they are correct. 4. * * update software. * * if applicable, check for software updates and install any available updates. 5. * * perform a physical inspection. * * inspect the barcode scanner for any visible damage or signs of wear and tear. 6. * * contact the manufacturer. * * if the issue persists after troubleshooting the above, contact the manufacturer for further assistance or repair. 7. * * consider a backup plan. * * have a backup plan in place to ensure patient care is not impacted if the barcode scanner is unavailable for an extended period. @topic:23


# # # issue - telemetry ecg monitor is not turning on # # # impact - inability to monitor patient's electrocardiogram - delay in diagnosis and treatment - patient discomfort # # # possible causes - battery is depleted - power cord is faulty - problem with the power supply - internal circuitry issue # # # recommendations - replace the battery - inspect the power cord and replace if necessary - check the power outlet to ensure it is functioning - contact the manufacturer for further troubleshooting and repair @topic:11


# # # issue the suresigns vs4 monitor is failing to calibrate properly. # # # impact the monitor is unable to provide accurate vital sign readings. # # # possible causes - the temp sensor is faulty. - the calibration procedure was not performed correctly. - there is a problem with the monitor's software. # # # recommendations - replace the temp sensor. - repeat the calibration procedure. - contact philips for technical support. @topic:20


# # # issue - the physiologic monitor ( bedside ) by philips, namely the intellivue mp40 ( m8003a ), is unable to measure spo2. # # # impact - accurate spo2 monitoring is crucial for patient care, especially during anesthesia, surgery, and critical care. inability to measure spo2 impairs the healthcare team's ability to assess a patient's oxygenation status. # # # possible causes - faulty spo2 sensor or cable - improper connection between the sensor and monitor - malfunctioning spo2 module within the monitor - software or firmware issues - interference from other medical devices or environmental factors # # # recommendations - * * verify the spo2 sensor and cable : * * inspect the sensor for damage or obstructions. ensure the cable is securely connected to both the sensor and monitor. replace the sensor or cable if necessary. - * * inspect connections : * * check all connections between the sensor, cable, and monitor. ensure they are secure and free from any loose contacts. - * * troubleshooting the monitor : * * perform self - diagnostics or consult the manufacturer's service manual to identify any internal issues with the spo2 module. - * * software / firmware update : * * check for any available software or firmware updates for the monitor. installing updates can address bugs or improve the performance of the spo2 module. - * * eliminate interference : * * identify any potential sources of electromagnetic interference ( emi ) nearby. move the monitor or patient away from these sources, or use shielded cables to minimize interference. - * * contact technical support : * * if the issue persists despite the above recommendations, contact the manufacturer's technical support for further assistance and troubleshooting. @topic:4


/ min ffl 8 apr 02 9 : 21 fi philips page 1 ( last ) | 341 27 printing patient reports sample report printouts ecg reports patient information numeric block wave area below the header on ecg reports, the numeric block shows the current hr, pvc, and st values. the wave area shows the printed waves for all available ecg leads. a 1 mv calibration bar is printed at the beginning of each wave. with the 3x4, 6x2, and 2x4 formats, a rhythm stripe prints a longer section of the ecg wave from the primary ecg lead for ecg rhythm evaluation. the ecg signal bandwidth, the patient ’ s paced status, the ecg gain, and the print speed are printed at the bottom of the wave area. pace pulse marks are automatically printed beside the wave for paced patients. beat labels can be set to print on the rhythm stripe. the 12x1 ( 2p ) format prints the report over two pages. ecg report type available formats available paper sizes 12 - lead ecg 3x4 landscape a4, letter, a3, ledger 6x2 landscape a4, letter, a3, ledger 12x1 portrait a4 and letter only 12x1 landscape a4, letter, a3, ledger 12x1 ( 2p ) landscape a4, letter ( 2 pages ) multi - lead ecg 2x4 landscape a4, letter, a3, ledger 8x1 portrait a4 and letter only 8x1 landscape a4, letter, a3, ledger other reports see the sections on trends and calculations and the chapter on event surveillance for other example reports. 342 28 using the drug calculator drug mixtures for intravenous ( iv ) drug infusions combine information on drug dose, rate, amount, volume, concentration, and standardized rate. the @topic:8


in good working order, potential falls of the patient,. patient in state of confusion, • patient's learning ability,. • children ( aged less than 12 or under 1. 46 m tall ), [UNK] persons lacking the mental capacity to recognize unsafe actionsunauthorized persons, [UNK] check the list of recommended mattresses on the label on the adjustable head section. [UNK] persons authorized to use the bed's functions must be capable of doing so in a safe and controlled manner. in case of doubt, the bed's functions must be locked. mattress * * for the hill - [UNK] 900 bed, hill - rom recommends the mattresses listed below, which are compatible with the safety recommendations ( see " risk prevention " matress label np150 np150 npso np100 folding mattress clamp when installing a mattress extension cushion, the clamp must be folded to avoid ower limbs. 2 adjustable mattress clamp the position of the clamps must be adjusted according to the width of the mattress in order to center and secure the mattress. wwwwa make sure that the mattress is correctly installed and well centered on the sleep surface using the adjustable clamps ( 2 positions : s and l ) and that the foot is secured by the clamp to avoid the creation of entrapment zones. other mattresses may be used, but the manufacturer must always be consulted to make sure that the bed / mattress / siderail combination does not affect the bed's erformance, its suitability for use or its safety characteristics. users must check the compatibility of the patient's weight and the accessories placed on the bed and the mattress system in view of the specifications of the medical bed anthe mattress system. references of recommended mattresses 90 cm wide mattresses are incompatible with egress handles *. position bride art number 1 name 2 l l s primo tm - ad085a ( @topic:9


# # # issue - monitor is not linking to the central system. # # # impact - lack of real - time monitoring and data collection, which could lead to delayed or inaccurate diagnosis and treatment decisions. # # # possible causes - incorrect configuration of the monitor's network settings - server issues or downtime - firewall or network security restrictions - damage to the monitor's hardware or software # # # recommendations - ensure the monitor's network settings are correct and match the hospital's network configuration. - check the status of the server and resolve any issues promptly. - update the monitor's firmware and software to the latest versions. - test the monitor's connection to the central system and troubleshoot any errors. - contact the manufacturer's technical support for further assistance if necessary. @topic:16


# # # issue - cause of failure is not specified. # # # impact - unable to determine the root cause of the failure, which could lead to recurring issues or potential safety hazards. # # # possible causes - incomplete or inaccurate reporting - lack of proper diagnostic tools or expertise - miscommunication or misunderstanding during repair process # # # recommendations - train technicians on proper diagnostic procedures and documentation practices. - provide access to adequate diagnostic tools and resources. - establish clear communication channels between technicians and engineers responsible for repairs. - implement a system for tracking and analyzing failure causes to identify potential trends or systemic issues. @topic:12


# # # issue - cuff failure on the bedside physiologic monitor. # # # impact - the device may not be able to accurately measure blood pressure. - this could lead to incorrect treatment decisions. # # # possible causes - cuff damage - cuff leakage - cuff age - cuff improper use # # # recommendations - replace the cuff with a new one. - inspect the cuff regularly for damage or leakage. - replace the cuff according to the manufacturer's recommendations. - train staff on proper cuff use. @topic:3


# # # issue - equipment failure : the telemetry ekg module ( philips m2601b ) is not functioning properly. # # # impact - inability to monitor patient's electrocardiogram ( ecg ) remotely. # # # possible causes - * * hardware failure : * * the equipment may have experienced a hardware malfunction, such as a damaged sensor or circuit board. - * * software issues : * * the device's software may be outdated or corrupted, leading to improper operation. - * * electrical problems : * * power surges or unstable voltage may have damaged the equipment. - * * environmental factors : * * extreme temperatures, humidity, or dust may have compromised the device's performance. - * * incorrect handling : * * the equipment may have been mishandled during use or storage, causing damage. - * * end of life : * * the philips m2601b model is discontinued and parts are no longer available, making repair impossible. # # # recommendations - * * troubleshooting : * * check for loose connections, damaged cables, or other obvious issues. - * * software updates : * * ensure that the device is running the latest software version. - * * electrical inspection : * * inspect the power source and ensure that the equipment is receiving proper voltage. - * * environmental control : * * store and use the equipment in a controlled environment with appropriate temperature, humidity, and cleanliness. - * * proper handling : * * train staff on proper handling and maintenance procedures for the equipment. - * * warranty assessment : * * review the equipment's warranty status and contact the manufacturer for potential repair or replacement options. - * * consideration of alternative devices : * * explore alternative telemetry ekg modules that are available and compatible with the hospital's systems. @topic:18


# # # issue * blood pressure ( bp ) sensor is malfunctioning, resulting in intermittent readings or no readings at all. # # # impact * accurate bp monitoring is crucial for patient assessment and management. * intermittent or inaccurate bp readings can lead to incorrect medical decisions and potential harm to the patient. # # # possible causes * faulty pressure transducer in the sensor * loose or damaged connections between the sensor and the monitor * software issues in the monitor * electrical interference from other medical devices * air bubbles in the sensor tubing # # # recommendations 1. * * inspect the sensor and tubing : * * check for any visible damage, loose connections, or air bubbles in the tubing. 2. * * test the sensor : * * perform a manual bp measurement using a sphygmomanometer to compare the readings with the monitor. 3. * * update software : * * contact the manufacturer to check if any software updates are available for the monitor. 4. * * check for interference : * * move the monitor away from any potential sources of electrical interference, such as other medical devices or electrical equipment. 5. * * replace the sensor : * * if other troubleshooting steps fail, consider replacing the faulty bp sensor with a new one. @topic:14


# # # issue the philips suresigns vm4 vital signs monitor is unstable due to a loose locking base. # # # impact the instability of the monitor could lead to inaccurate readings or even injury to the patient. # # # possible causes - the locking base may have become loose over time due to frequent use or movement of the monitor. - the locking mechanism may be damaged or malfunctioning. # # # recommendations - tighten the locking base securely to ensure stability. - inspect the locking mechanism for any damage or malfunction and replace if necessary. - regularly inspect and maintain the monitor to prevent similar issues in the future. @topic:2


# # # issue - not connecting to the central monitor # # # impact - the device cannot send patient data to the central monitoring system. # # # possible causes - loose ethernet cable - faulty ethernet cable - incorrectly configured network settings - software issues - hardware issues with the device or central monitor # # # recommendations - check the ethernet cable connections and ensure they are secure. - replace the ethernet cable if necessary. - verify the network settings on the device and central monitor. - contact the manufacturer for technical support or repair. @topic:16


# # # issue - battery failure # # # impact - inability to use the device for patient transport - potential loss of patient data # # # possible causes - battery life cycle exceeded - battery not properly maintained - battery damaged # # # recommendations - replace the battery with a new one - follow the manufacturer's instructions for battery maintenance - inspect the battery regularly for damage @topic:11


# # # cause of failure - cl address of monitor changed, causing it to lose connection with central monitor # # # recommendations - ensure that cl address of the monitor is set correctly during installation or troubleshooting to prevent future connectivity issues. @topic:16


# # # issue the electric bed ( huntleigh enterprise 8000 ) is unable to adjust. # # # impact patients may not be able to be positioned properly or comfortably, which can lead to discomfort, pain, or other health issues. # # # possible causes - hardware issue : * broken or loose connections in the electrical system * malfunctioning motor or actuator - software issue : * corrupted or outdated firmware * faulty programming - user error : * improper use or overloading of the bed # # # recommendations 1. * * hardware inspection * * : * check for any loose or damaged wires, connectors, or components. * inspect the motor and actuator for any signs of wear or damage. 2. * * software troubleshooting * * : * check for any firmware updates or software patches. * reinstall or update the software if necessary. 3. * * user training * * : * educate users on the proper use and maintenance of the bed. * ensure that the bed is not overloaded or used improperly. 4. * * replacement of components * * : * if hardware inspection or software troubleshooting does not resolve the issue, replace the faulty components ( e. g., motor, actuator, or electrical components ). @topic:1


# # # issue - alarm alert error while powering on the medical device ( monitors, patient transport ) # # # impact - inability to use the device for patient monitoring and transport. # # # possible causes - defective alarm system within the device - faulty or loose connections between the device and accessories ( charging dock, cables ) - software or firmware issues that trigger false alarms # # # recommendations * * technical troubleshooting : * * - * * check alarm settings : * * ensure that alarms are properly configured and not overly sensitive. - * * inspect connections : * * verify that the charging dock and cables are securely connected to both the device and the power source. tighten any loose connections or replace defective components. - * * run diagnostics : * * utilize built - in diagnostic tools or software updates to identify and address any system errors. - * * perform factory reset : * * reset the device to its original settings to clear potential configuration issues. * * maintenance and monitoring : * * - * * regular maintenance : * * schedule routine inspections and maintenance checks to identify and resolve potential issues before they cause failures. - * * monitor alarm logs : * * regularly review alarm logs to track alarm events and identify any patterns that may indicate underlying problems. - * * software updates : * * ensure that the device is running the latest software updates to address known bugs or improve performance. * * communication and reporting : * * - * * inform user : * * clearly communicate the cause of failure and recommended actions to the user ( healthcare professional ) to avoid future misuse or confusion. - * * document and track : * * maintain detailed records of the issue, troubleshooting steps, and any repairs or replacements performed to facilitate future reference and trend analysis. * * additional tips : * * - * * inspect patient cables : * * ensure that patient cables are undamaged and properly connected. - * * @topic:11


as described in the previous section. if you are operating two or three displays, you can choose screens for all displays from one location : 1 select profiles in the monitor info line of the first display, 2 select display 1, display 2, or display 3 then select the screen you want to appear on that display from the list of available screens. when two displays are mounted next to each other or one above the other, a special screen can be assigned which spans across both displays. the screen content for these tall and wide screens can then use the increased area available with two displays. these screens appear in the screen list with a special tall screen or wide screen symbol. certain windows ( for example : cardiac output procedure ) can only be shown on one display at a time. if you try to open one of these windows when it is already shown on another display, you will see a blank grey window with a cross through it. using the xds remote display using the intellivue xds solution it is possible to view an independent monitor screen on an external display. the xds solution consists of a medical grade pc - based hardware platform, xds application software and the xds connectivity option on the monitor. depending on the configuration you can also operate the monitor from the external display. the xds must be connected to the same local area network ( lan ) as the monitor. it is also possible to use an existing pc, connected to the same lan, to host the xds application software. for more details, including limitations and restrictions, refer to the instructions for use for the xds application. using the visitor screen if a visitor screen is configured for your monitor, you can use it to clear the screen of all waves and numerics but continue to monitor the patient with active alarms and trend storage at the bedside and information center. you can change the name of the visitor screen in configuration mode. ♦ @topic:10


# # # issue the electric bed's head cannot be adjusted downwards. # # # impact the patient cannot be positioned comfortably for medical procedures or rest. # # # possible causes - faulty control box - problem with the motor - mechanical obstruction in the bed frame # # # recommendations * * immediate actions : * * - check for any visible damage or loose connections in the control box or motor. - try resetting the bed by unplugging it from power and plugging it back in. - maneuver the bed manually to remove any potential obstructions. * * further investigation and repair : * * - if the immediate actions do not resolve the issue, a more thorough inspection is necessary. - check the control box for any burn marks or blown components. - test the motor for continuity and resistance. - examine the bed frame for any bent or broken parts that may be hindering movement. - consult the manufacturer's manual for troubleshooting and repair instructions. * * preventive measures : * * - regular maintenance should be performed on the bed to prevent future issues. - ensure the bed is used according to the manufacturer's guidelines. - train staff on proper operation and handling of the equipment. @topic:1


# # # issue - equipment : physiologic bedside monitor - brand : philips - model : intellivue mp70 ( m8007a ) - description : the biomedical engineer ( bme ) found that the patient lead wire is damaged. a replacement lead wire was provided from another machine. # # # impact - the patient's physiological data may not be accurately monitored or displayed due to the damaged lead wire. # # # possible causes - the lead wire may have been damaged due to : - physical damage ( e. g., pulled, crushed, or cut ) - exposure to corrosive substances - normal wear and tear # # # recommendations - replace the damaged lead wire with a new one. - regularly inspect all lead wires for damage. - train staff on proper handling and care of lead wires. @topic:18


* * issue : * * unable to successfully transmit vital signs data from philips suresigns vs4 monitor to the vsm system. * * impact : * * * delay in patient monitoring and care * incomplete medical records * potential patient safety concerns * * possible causes : * * # # # 1. connectivity issues * loose connections between the monitor and the vsm system * faulty cables or connectors * interference from other devices on the network # # # 2. configuration errors * incorrect settings in the monitor or vsm system * mismatched communication protocols * inadequate network configuration # # # 3. software issues * outdated or corrupted software on the monitor or vsm system * bugs or glitches in the software * compatibility problems between different versions of software # # # 4. hardware malfunction * damaged components in the monitor or vsm system * power supply issues * internal hardware failure # # # 5. human error * incorrect operation of the equipment * inadequate training or experience * misinterpretation of error messages # # # recommendations : * * * * * verify connectivity : * * check all cables and connections to ensure they are secure and functioning properly. * * * review configuration : * * ensure that both the monitor and vsm system are configured correctly for data transmission. verify communication protocols and network settings. * * * update software : * * check for software updates and install them if necessary. * * * troubleshoot hardware : * * perform a physical examination of the equipment to identify any visible damage or faults. contact the manufacturer for further assistance if necessary. * * * user training : * * provide adequate training to all users to ensure proper operation and troubleshooting of the equipment. * * * error logs : * * review error logs on the monitor and vsm system to identify any specific error messages or clues to the cause of failure. * * * contact support : * * reach @topic:22


##hs compuant, dom > 2014 - 05 modules m1012 - 69701 453564453161 m _ co module c. o., eng m1012 - 69704 453564463411 m _ co module c. o., symbol multilang modules with picco m1012 - 69751 453564453191 m _ co module cio picco, eng m1012 - 69754 453564463431 m _ co module picco, symbol multilang replacement parts front housings m1012 - 60201 453563462021 m _ co frt housg co, eng m1012 - 60204 453564463451 m _ co frt housg co, symbol multilang front housings with picco m1012 - 60251 453563462031 m _ co frt housg cio picco, eng m1012 - 60254 453564463461 m _ co frt housg cio picco symbol multilang bezel m1012 - 42201 453563463241 m _ co plast bezel co module 237 [ 6 parts ] m1014a spirometry module part number 12nc description m1014 - 69601 451261014451 m _ spr module spiro, eng m1014 - 69602 451261014461 m _ spr module spiro, symbol front housing m1014 - 60201 451261014491 m _ spr frt housg assy spiro, eng m1014 - 60202 451261014501 m _ spr frt housg assy spiro, symbol bezel m1014 - 42201 45126 @topic:19


# # # issue - spring is loose. - the locking mechanism needs to be replaced. - the operation of the bed is affected. # # # impact - the bed cannot be used safely. - the patient may be at risk of falling out of bed. # # # possible causes - the spring may have become worn or damaged. - the locking mechanism may have become loose or broken. - the bed may have been overloaded. # # # recommendations - the spring should be replaced. - the locking mechanism should be replaced. - the bed should be inspected regularly to ensure that it is in good working order. @topic:6


# # # issue - equipment not functioning properly due to being borrowed by another department. # # # impact - disruption of patient care. - delays in patient treatment. - increased workload for staff. # # # possible causes - poor communication between departments. - lack of backup equipment. - failure to follow established protocols. # # # recommendations - improve communication between departments to ensure that equipment is not borrowed without authorization. - provide backup equipment to minimize disruption in the event of equipment failure. - establish clear protocols for the borrowing and return of equipment. @topic:12


* * issue : * * battery failure in vital signs monitor * * impact : * * inability to monitor patient's vital signs accurately and reliably, potentially compromising patient safety. * * possible causes : * * * exhaustion of battery life due to normal use and aging * manufacturing defect leading to premature battery failure * incorrect charging or handling of the battery * damage to the battery or charging system * * recommendations : * * * * * regular maintenance : * * regularly check battery performance and replace it when necessary. * * * proper charging protocol : * * follow manufacturer's instructions for charging the battery to avoid overcharging or undercharging. * * * training : * * ensure staff are trained on proper handling and maintenance of the equipment. * * * monitoring : * * implement a system to monitor battery performance and identify devices with declining batteries. * * * procurement : * * purchase high - quality batteries from reputable manufacturers to reduce the risk of premature failure. * * * consideration of alternative power sources : * * explore the use of alternative power sources, such as ac power adaptors, to minimize reliance on batteries. @topic:11


# # # issue the medical cart is displaying incorrect temperature readings. # # # impact this could lead to inaccurate diagnosis and treatment of patients. # # # possible causes - the temperature sensor is faulty. - the thermohygrometer is not calibrated correctly. - the medical cart is not being used in the correct environment. # # # recommendations - calibrate the thermohygrometer. - test the temperature sensor. - ensure that the medical cart is being used in the correct environment. @topic:20


# # # issue - hn scan is not accessible # # # impact - inability to monitor patient's vital signs # # # possible causes - software or hardware malfunction in the scanning device - connectivity issues between the scanning device and the patient monitor - incorrect configuration or settings on the patient monitor - damage to the scanning device or its components # # # recommendations - * * inspect the scanning device : * * check for any visible signs of damage or malfunction. - * * check connections : * * ensure that all cables and connections between the scanning device and the patient monitor are secure and functional. - * * restart the equipment : * * shut down both the scanning device and the patient monitor, then restart them to reset any temporary software issues. - * * configure the settings : * * verify that the settings on the patient monitor are correct for the type of scan being performed. - * * consult with the manufacturer : * * if the issue persists, contact the equipment manufacturer for technical support and guidance. - * * replace the scanning device : * * if necessary, replace the scanning device if it is found to be faulty or damaged beyond repair. @topic:22